 Research
 Open access
 Published:
Characterization of the headtotrunk orientation with handheld optical 3D apparatus based on the fringe projection technique
BioMedical Engineering OnLine volume 12, Article number: 96 (2013)
Abstract
Background
Knowing the orientation of the head is important in many fields, including medicine. Many methods and measuring systems exist, but usually they use different markers or sensors attached to the subject’s head for head orientation determination. In certain applications these attachments may represent a burden or a distraction to the subject under study which may have an unfavourable impact on the measurement. We propose a noncontact optical method for headtotrunk orientation measurement that does not require any attachments to the subject under study.
Methods
An innovative handheld 3D apparatus has been developed for noninvasive and fast 3D shape measurements. It is based on the triangulation principle in combination with fringe projection. The shape of the subject’s upper trunk and head is reconstructed from a single image using the Fourier transform profilometry method. Two shape measurements are required to determine the headtotrunk orientation angles: one in the reference (neutral) position and the other one in the position of interest. The algorithm for the headtotrunk orientation angle extraction is based on the separate alignment of the shape of the subject’s upper trunk and head against the corresponding shape in the reference pose. Single factor analysis of variance (ANOVA) was used for statistical characterisation of the method precision.
Results
The method and the 3D apparatus were verified invitro using a mannequin and a reference orientation tracker. The uncertainty of the calculated orientation was 2°. During the invivo test with a human subject diagnosed with cervical dystonia (aged 60), the repeatability of the measurements was 3°. Invitro and invivo comparison was done on the basis of an experiment with the mannequin and a healthy male (aged 29). These results show that only the difference between flexion/extension measured angles was statistically significant. The differences between means were less than 1° for all ranges.
Conclusions
The new noncontact method enables the compensation of the movement of the measuring instrument or the subject’s body as a whole, is noninvasive, requires little additional equipment and causes little stress for the subject and operator. We find that it is appropriate for measurements of the head orientation with respect to the trunk for the characterization of the cervical dystonia.
Background
Knowing the orientation of the head is important in many different fields such as humancomputer interfaces, video compression, face recognition systems, biological experiments and medicine [1–3]. Head orientation measurement techniques are mainly based on a combination of different gyroscopes, accelerometers and electronic compasses. One such system is based on a threeaxis accelerometer and a resonator gyroscope mounted on a hard hat [4]. The data acquired by the tilt sensor reached an accuracy of 0.63° and the gyroscope reached an accuracy of 0.52°. Another measuring system was composed of four sensors placed on the person and the stationary electromagnetic transmitter of a Fastrack system [5]. Its static orientation accuracy is 0.15° [6]. Another technique uses two inclinometers attached to each arm of a pair of eyeglasses [7]. Yet another uses two intersecting plastic protractors, a grid pattern and a laser mounted on a helmet [8]. The rotations in the sagittal and transverse plane were calculated employing the location of the laser illumination. The rotation in the coronal plane was calculated from protractors. Nintendo Wii remote controllers fixed to a mechanical frame and an IR beacon, mounted on the head were also used to measure the head posture. The standard deviation of 15 repeated measurements ranged from 0.56° to 2.70° [3]. The head orientation can be also determined by using 2D camera. One technique is based on finding the best translation and rotation of the premeasured 3D shape of the subject’s face according to the acquired 2D image [2]. The reported standard deviation is 2.3 mm in position and 0.47° in orientation. The second 2D camera based technique uses a template tracking algorithm that automatically extracts neck angles from sagittal videos [9].
Although the presented systems achieve high levels of measuring uncertainty, our concerns are related mostly to the fixation of the different markers on the subject’s body. In our opinion it is very hard to ensure that the junction between the marker and the human body is rigid. The systems are also usually quite large and difficult to transport. Other weakness of the above mentioned methods is impossibility to distinguish between the head rotations caused by the neck and the rotations caused by the torso twisting, because the torso rotations are not monitored.
This paper presents a novel method that enables the measurement of the head posture relative to the trunk. The method has been developed with the aim of characterizing the effects of cervical dystonia, a disease which affects the patient’s head posture and the range of movement as well as their quality of life in many other ways [10, 11]. The main effects on the head posture are the head lateral tilt in the body’s coronal plane (laterocollis), head rotation in transversal plane (torticollis) and flexion (anterocollis) or extension (retrocollis) in the body’s sagittal plane. Most patients suffer from a combination of two or more effects [5]. The presented method utilizes optical 3D measurements of the upper trunk and head to measure these effects. The relative rotation of the head with respect to the trunk is calculated from the alignment of the trunk and head measurements to the reference one.
Methods
Handheld 3D camera
The handheld 3D camera (HHC) is composed of a digital singlelens reflex (DSLR) camera body, lens and grating projection system (GrPS). In our experimental setup, we used a Nikon D90 with a Nikkor 50 mm f/1.8 G AFS lens. The GrPS is composed of an industrial Cmount lens (Computar M2514MP, 25 mm, F1.4, 2/3″), a grating mask and a connection arm (see Figure 1). The builtin camera flash was used as a light source. The grating mask was placed on the back of the holder, between the projector lens and the camera flash. The mask is rectangular in shape, approximately 8.8×6.6 mm in size and composed of a 67stripe Ronchi pattern which was made using a laser micromarking technique (LPKF MarkLine8V).
For acquiring measurements a picture of the measured surface was captured using the camera flash. The mask was illuminated by the camera flash and the generated light pattern was projected onto the surface by the GrPS lens (see Figure 2). The deformed grating pattern is seen on the captured image. For the surface reconstruction, we transferred the images to a personal computer and processed them with developed surface reconstruction software. It is based on the principle of the Fourier transform profilometry method [12, 13], the phase unwrapped by a qualityguided unwrapping algorithm [14] and reconstruction method [15].
The measuring range of the system is 700×520×400 mm at a distance of 2 m. Calibration was done by the reference surface [16]. Ten measurements of the reference surface (plate with semicircular grooves, of which the dimensions are exactly known) were captured across the measuring range and imported into the software for the optimization of the surface reconstruction parameters. These parameters include the focal length of the lens, the rotations of the camera and the projector, the triangulation angle, the angular separation between the light planes, the camera and lens distortion, the camera sensor dimensions and the image center location. The parameters were numerically optimized until the minimum standard deviation of the displacements between the measured and reference surface was found. After the calibration the standard deviation of the displacement was 1.6 mm.
In Figure 3 examples of the three reconstructed surfaces of the same human subject, captured by the presented 3D measuring system are shown. The surface seen in Figure 3a was acquired in the reference position and is later aligned in the coordinate system; measurements in Figure 3b and Figure 3c, are registered following the procedure presented in the next section.
Headtotrunk rotation extraction from 3D measurements
The rotations of the human body and head can be described as rotations in three body planes; the sagittal, coronal and transverse plane (see Figure 4). Because the software can measure only the rotations in the software’s coordinate system, it is essential that the body planes are parallel to the coordinate system planes of the software. The sagittal body plane is found using the mirroring and aligning procedure [17]. We determined the coronal plane assuming that the body’s coronal plane is parallel to the wall the person was leaning on. The wall was measured at the same time as the person and can be seen in Figure 4. The directions of the X, Y and Z axis are also seen in Figure 4. Any translation of the coronal plane in the Z direction and the transverse plane in the Y direction has no effect on the calculated Euler angles.
Hereinafter, the flexion and extension in the sagittal plane (around the X axis) will be noted as the positive and negative φ angle, respectively; the right and left rotation in the transverse plane (around the Y axis) as the positive and negative angles θ, respectively, and the left and right lateral tilt (around the Z axis) as the positive and negative angles ψ, respectively.
Each measured surface was split on the subject’s head surface and the subject’s trunk surface, where only the parts of the head or trunk on which we expected a small degree of deformation during the movement are kept for further analysis. This means that the neck and upper arm parts are excluded from the head and trunk partial measurements to enable a reliable determination of the head orientation. The divided measurements are then aligned to the reference measurement by using the Geomagic Studio software (GMS) [18]. An example of such alignment procedure is shown in Figure 5, where the colours indicate the deviations of the currently analysed body position with respect to the reference measurement (shown in grey).
In this manner we obtained a pair of rotation matrices from the GMS for each measurement. We denoted the matrix of the head rotation R_{ h } and matrix of the trunk rotation R_{ t }. Due to the possibility of the person moving between consecutive measurements and the movement of the measuring instrument, we calculated the relative head rotation with respect to the trunk R as:
Since the matrix R_{ t } is orthogonal, the inverse operation can be substituted with transposition, to reduce the computational demands.
The Euler angles, which represent the headtotorso orientation, are then calculated by the rotation matrix decomposition technique as follows [19]:
Invitro verification
The uncertainty of the measuring method was verified by measuring the upper part of a female mannequin with a movable head. An inertial measurement sensor (Xsens 3DOF orientation Tracker, hereinafter referred to as “XOT”, [20]) was attached to the top of the mannequin’s head to perform reference measurements of the head’s orientation. Since the trunk of the mannequin was immovably fixed, the changes in the orientation XOT measured were entirely caused by the head rotation. That is why we can interpret measured head orientation as headtotrunk orientation. Its specified resolution is 0.235° at the acquisition rate of 100 Hz. In addition, simultaneous 3D measurements with a 3D laser scanner (LS) with a higher resolution compared to the HHC (one standard deviation after the calibration was 0.3 mm) were carried out. However, it is important to note, that the LS is inappropriate for invivo measurements, due to the long data acquisition times (about 5 s) and its lesser portability. With the presented experimental setup we evaluated the impact of the apparatus’ resolution on the final–head rotation–precision.
A series of measurements of the fixed mannequin with a moveable head were acquired by all three measuring systems. The orientation of the head was measured in eight positions which imitated the standard positions for the determination of the cervical range of motion. These standard positions are:

1
Neutral head position, where the body planes of the head are parallel to the body planes of the trunk (this is also the reference position).

2
Natural head position, where the head is relaxed; in healthy subjects it should be the same as the reference one. In the case of the mannequin, the position was simulated as a random rotation around all axes.

3
Position of the maximal angle φ (flexion).

4
Position of the minimal angle φ (extension).

5
Position of the maximal angle θ (right rotation).

6
Position of the minimal angle θ (left rotation).

7
Position of the maximal angle ψ (left lateral tilt).

8
Position of the minimal angle ψ (right lateral tilt).
Since the coordinate systems of the utilized measuring instruments (CS_{1} and CS_{2} on Figure 6) are misaligned, the rotation matrix R_{ g } between CS_{1} and CS_{2} has to be determined in order to measure the rotation angles about the same axes and be able to compare the rotation values obtained by different measuring instruments. If the first instrument measures the rotation of the object as R_{1}, the second as R_{2} and R_{1→2} is the rotation R_{1} transformed to the CS_{2}, the R_{ g } can be calculated using the following procedure:
First, the rotation matrices are transformed into the axisangle representation [19], where the rotation angle α is:
The vector k which represents the rotation axis is:
where R_{i,j} is the element of a rotation matrix in the ith row and the jth column. In this way, we calculate α_{1}, α_{2}, k_{1} and k_{2} from the known rotation matrices R_{1} and R_{2}. Further we calculate the rotation axis k_{ g } and the angle α_{ g } between CS_{1} and CS_{2} as follows:
In general the rotation matrix R from the known angle α and axis k = (k_{1},k_{2},k_{3})^{T} is calculated as:
where v = 1 − cos(α).
Ideally the angles α_{1} and α_{2} should be the same if both instruments measure the same object movement. But due to the limited measurement uncertainty some variation occurs. Therefore, the measurements were taken in all standard positions and {\mathbf{R}}_{g}^{i} for each ith position was calculated. After that, the average rotation matrix {\overline{\mathbf{R}}}_{g} was calculated by averaging the corresponding Euler angles. So to transform R_{1} to R_{1→2}, we calculate k_{1} and α_{1} as shown in Equations (7) and (8). Vector k_{1→2} and angle α_{1→2} in CS_{2} are then defined as:
Once k_{1→2} and α_{1→2} are obtained, we calculate R_{1→2} as shown in the Equation (9). In the case of ideal measuring instruments, R_{1→2} and R_{2} should be exactly the same.
The differences of measured angles between the reference instrument (XOT) and 3D measuring instruments (LS and HHC) were analysed using single factor analysis of variance (ANOVA). The factor was measuring instrument. We analysed the main angles of all standard positions with the exception of the reference one. That results into seven ANOVA analyses between three sets of data. Differences were considered significant at p < 0.05. Bonferroni corrected ttest was used to pairwise analyse the differences XOT vs. LS and XOT vs. HHC. Here, differences were considered significant at p < 0.0250.
Invivo verification
All invivo measurements were performed using the HHC apparatus. The human subject (diagnosed with cervical dystonia, aged 60) was dressed in a white Tshirt and a white headband was used to prevent the hair obstructing the subject’s face and extending the measurable surface. During the measuring procedure, the subject was resting against a wall. He was asked to rotate his head left and right as far as possible. Then he stepped away from the wall, relaxed for one minute and repeated the procedure for 13 times.
The presented research has received approval by the National Medical Ethics Committee of the Republic of Slovenia, which assesses the compliance with the Helsinki Declaration. The approval (No.: 133/04/13) is dated on 8th May 2013.
Invitro vs. invivo comparison
Invitro and invivo comparison was done on the basis of an experiment where the HHC apparatus was used to measure headtotrunk orientation of the mannequin and a healthy male (aged 29), respectively. To assure the repeatable head positioning, a laser projector, firmly attached to the head (in the direction of axis –Z on Figure 4), was used for illuminating the markers on the opposite wall. In the case of rotation and flexion/extension laser beam projector and one marker for each position were used. In the case of lateral tilt we used laser plane projector and two markers for each position, placed on the opposite wall approximately four meters apart. The neutral, moderate left and right rotation, moderate left and right lateral tilt and moderate flexion and extension were measured for 15 times. Only the main angles of each position were considered, since only these angles were monitored using the laser projector. Absolute values of the main angles (θ for left and right position, φ for flexion and extension and ψ for left and right lateral tilt) were summed to eliminate the effect of the mismatch of the laser placements on the both heads. The ranges of the rotation for both subjects were analysed using ANOVA single factor analysis. The factor was measured subject. Differences were considered significant at p < 0.05.
Results and discussion
The uncertainty of the proposed method was assessed by three experiments. Invitro verification was performed using the mannequin to evaluate the uncertainty of the method by comparing the results with the reference instrument. Invivo verification was performed in order to evaluate the human influence on the measurement’s uncertainty, and invitro vs. invivo comparison to prove the relevance of invitro results. The gathered data was analysed using standard statistics’ approaches; unless noted otherwise, the precision of the measurements is defined as one standard deviation.
The results of the invitro mannequin measurement are shown in Table 1. The reference instrument was the XOT, which recorded the data at each head position for one minute. 3D measurements were repeated twenty times using LS and forty times using HHC. The average Euler angles and standard deviations are shown for each instrument and for each head position. From these results we see that the XOT and LS have approximately the same precision (around 0.12°). This means that the proposed method of the head orientation extraction from the 3D shape works well. Additionally, we noticed that the HHC has somehow a lower precision (around 1.6°), which is attributed to the lower 3D measuring precision. The uncertainty of the calculated Euler angles compared to the XOT is presented by the BlandAltman plots in Figure 7. The angle differences Δφ, Δθ and Δψ are defined as:
where angles with the index XOT were measured with XOT and angles with the index 3D were measured with LS in Figure 7a and HHC in Figure 7b. The average angles of the position were determined as the average of the angles measured with XOT for each position.
Figure 7 indicates that the angle differences measured by LS are approximately half of the ones measured by HHC. This is a consequence of higher resolution of LS relative to HHC as they both use the same 3D surface reconstruction method and algorithm for the head orientation characterization. We expected less accurate results at a higher rotation of the head around the Y axis, due to the smaller area of overlapping between the reference and the current head surface. A healthy person can rotate the head up to 80° left or right [21]. In Table 1 we see that the standard deviation of the measured angles is indeed related to its amplitude. The left rotation (see Table 1, row “left rotation”, column “HHC”) has an amplitude of −50° and a standard deviation of 1.9° in case of using the HHC apparatus. On the other side, the right rotation has an amplitude of 40° and a standard deviation of 0.8° (see Table 1, row “right rotation”, column “HHC”), which confirmed our expectations. This effect can be minimized by adjusting the position of the measuring system in a way, that the impact angles between the optical axis of the measuring system and the surface of the head and trunk are as close as possible; if the head is rotated by 80° in relation to the trunk, the measuring system has to be placed in the direction rotated by 40° in relation to the trunk. The same problem can occur during flexion and extension, but those angles are usually smaller, up to 60° [21]. The least problematic is lateral tilt. Even in the case of large tilting angles, the impact angle between the measured surface and the optical axis of the measuring system does not change much. The problem of reduced overlapping is also insignificant.
The ANOVA analysis of the differences between main measured angles for all measuring instruments showed, that differences were significant at all positions. The critical F(2,57) was 3.1. Calculated F values were 6.7, 10.0, 271.1, 430.6, 35.2, 33.0, 74.6 for natural position, left rotation, right rotation, left lateral tilt, right lateral tilt, flexion and extension, respectively. Bonferroni corrected ttests also showed that all differences were significant, with the exception of the natural position between XOT and LS (p = 0.12), left rotation between XOT and HHC (p = 0.56), right rotation between XOT and HHC (p = 0.11) and flexion between XOT and LS (p = 0.14). We think that the differences were caused by the nonideal alignment between CS of the both measuring instruments and by the limited measuring accuracy.
The average angles and standard deviations of the 13 invivo measurements are seen in the Table 2. The standard deviation of the measured angles is increased compared to the results of the mannequin analysis. However, it is important to note that this deviation also includes scatter caused by the subject’s nervousness, misinterpretation of the command, tiredness, deformations of his face during head movement, measurement noise caused by the skin, etc.
The results of the invitro and invivo measurements are shown in Table 3. The critical F(1,28) for the statistical analysis was 4.2. From Table 3, column ANOVA, we see that there is no statistically significant difference between measured angles θ for the rotation of the human subject and the mannequin (F = 3.1). The difference between measured angles φ for the flexion/extension is marginally significant (F = 4.2). And the difference between measured angles ψ for the lateral tilt is statistically significant (F = 20.7). We explain this significance as the result of the limited human ability of accurate positioning. The illumination of the markers proved to be quite challenging task for the human subject. Tilting the head for the required angle was particularly difficult, since the subject had to ensure the overlapping of the laser line with two markers.
The proposed method enables a satisfactory level of measurement precision (2° for invitro and 3° for invivo measurement). It is a bit lower compared to the methods, presented in the Background section (from 0.15° to 2.7°) [2–9], but it has other advantages, like compensation of the whole body rotations, higher system portability, and finally no need for any device attached to the patient’s body.
Since the daily clinical practice needs a short, simple and precise tool for rating the improvement or deterioration of the cervical dystonia [22, 23], we believe that the presented method has great potential to replace the currently used measuring systems.
Conclusions
The new noncontact method for measuring headtotrunk orientation is based on 3D surface acquisition. It enables the compensation of the movement of the measuring instrument or the subject’s body as a whole, is noninvasive, requires little additional equipment and causes little stress for the subject and operator. A subject is measured using the handheld 3D apparatus which is based on the fringe projection technique. The apparatus is composed of a commercial DSLR camera and custom projection optics, which uses the light from the camera’s builtin flash. It is also highly portable and very userfriendly. The algorithm of the headtotrunk orientation extraction is based on the separate alignment of the trunk and head on the reference measurement. The final results are the angles of rotation around each coordinate axis. Evaluation of the proposed method shows that the invitro measuring uncertainty is no more than 2° and invivo accuracy is less than 3°. The compassion of invitro vs. invivo measurements showed, that only the difference between flexion/extension measured angles was statistically significant. The differences between means were less than 1° for all ranges. The main contribution factors to the measurement’s uncertainty are the accuracy of the 3D apparatus and the subject’s timevarying deformation uncertainty. The measuring uncertainty is acceptable for the clinical use of the system, which can greatly improve the assessment of cervical dystonia in daily clinical practice.
Abbreviations
 FTP:

Fourier transform profilometry
 DSLR:

Digital singlelens reflex
 GPrS:

Grating projection system
 HHC:

Handheld 3D camera
 LS:

Laser scanner
 XOT:

Xsens 3DOF orientation tracker
 GMS:

Geomagic studio
 CS:

Coordinate system
 ANOVA:

Analysis of variance.
References
Ohayon S, Rivlin E: Robust 3D head tracking using camera pose estimation. In 18th International Conference on Pattern Recognition, ICPR 2006. IEEE; 2006:1063–1066.
Malassiotis S, Strintzis MG: Robust realtime 3D head pose estimation from range data. Pattern Recogn 2005, 38: 1153–1165. 10.1016/j.patcog.2004.11.020
Kim J, Nam KW, Jang IG, Yang HK, Kim KG, Hwang JM: Nintendo Wii remote controllers for head posture measurement: accuracy, validity, and reliability of the infrared optical head tracker. Invest Ophthalmol Vis Sci 2012, 53: 1388–1396. 10.1167/iovs.118329
Lian E, Hachadorian J, Hoan NT, Toi V: A Novel Electronic Cervical Range of Motion Measurement System. In The Third International Conference on the Development of Biomedical Engineering in Vietnam. Berlin: Springer Berlin Heidelber; 2010:140–143.
Boccagni C, Carpaneto J, Micera S, Bagnato S, Galardi G: Motion analysis in cervical dystonia. Neurol Sci 2008, 29: 375–381.
Fastrack Brochure http://www.polhemus.com/polhemus_editor/assets/FASTRAK%20Brochure.pdf
Usumez A, Usumez S, Orhan M: Effect of complete dentures on dynamic measurement of changing head position: a pilot study. J Prosthet Dent 2003, 90: 394–400. 10.1016/S00223913(03)004141
Hald ES, Hertle RW, Yang D: Development and validation of a digital head posture measuring system. Am J Ophthalmol 2009, 147(1092–1100):e1093.
Delbert H, Ervin S, Catriona S, Tom C: Extraction of average neck flexion angle during swallowing in neutral and chintuck positions. Biomed Eng Online 2009, 8: 25. 10.1186/1475925X825
Gregori B, Agostino R, Bologna M, Dinapoli L, Colosimo C, Accornero N, Berardelli A: Fast voluntary neck movements in patients with cervical dystonia: a kinematic study before and after therapy with botulinum toxin type A. Clin Neurophysiol 2008, 119: 273–280. 10.1016/j.clinph.2007.10.007
Geyer HL, Bressman SB: The diagnosis of dystonia. Lancet Neurol 2006, 5: 780–790. 10.1016/S14744422(06)705476
Takeda M, Mutoh K: Fourier transform profilometry for the automatic measurement of 3D object shape. Appl Opt 1983, 22: 3977–3982. 10.1364/AO.22.003977
Su X, Chen W: Fourier transform profilometry: a review. Opt Lasers Eng 2001, 35: 263–284. 10.1016/S01438166(01)000239
Herráez MA, Burton DR, Lalor MJ, Gdeisat MA: Fast twodimensional phaseunwrapping algorithm based on sorting by reliability following a noncontinuous path. Appl Opt 2002, 41: 7437–7444. 10.1364/AO.41.007437
Jain R, Kasturi R, Schunck BG: Machine vision. New York: McGrawHill; 1995.
Jezeršek M, Možina J: Highspeed measurement of foot shape based on multiplelaserplane triangulation. Opt Eng 2009, 48(113604–113604):113608.
Nkenke E, Benz M, Maier T, Wiltfang J, Holbach LM, Kramer M, Hausler G, Neukam FW: Relative en and exophthalmometry in zygomatic fractures comparing optical noncontact, nonionizing 3D imaging to the Hertel instrument and computed tomography. J Craniomaxillofac Surg 2003, 31: 362–368. 10.1016/j.jcms.2003.07.001
Geomagic Studio Presentation http://www.geomagic.com/en/products/studio/overview/
Spong MW, Hutchinson S, Vidyasagar M: Robot modeling and control. New York: John Wiley & Sons; 2006.
Xsens MTi presentation http://www.xsens.com/en/general/mti
Youdas JW, Garrett TR, Suman VJ, Bogard CL, Hallman HO, Carey JR: Normal range of motion of the cervical spine: an initial goniometric study. Phys Ther 1992, 72: 770–780.
Jost WH, Hefter H, Stenner A, Reichel G: Rating scales for cervical dystonia: a critical evaluation of tools for outcome assessment of botulinum toxin therapy. J Neural Transm 2012, 120: 1–10.
Cano SJ, Hobart JC, Fitzpatrick R, Bhatia K, Thompson AJ, Warner TT: Patient‒based outcomes of cervical dystonia: a review of rating scales. Mov Disord 2004, 19: 1054–1059. 10.1002/mds.20055
Acknowledgements
Research was conducted as a part of the project Laser triangulation in medicine (LASTRIM–L74274), financed by the Slovenian Research Agency. The authors would also like to thank Professor Zvezdan Pirtošek, MD, PhD, and all other personnel of the Department of Neurology for all the help and suggestions during invivo measurement execution.
Author information
Authors and Affiliations
Corresponding author
Additional information
Competing interests
The authors declare that they have no competing interests.
Authors’ contributions
UP carried out the measurements, analysis, and drafted the manuscript. JD concepted and designed the experiment and revised the manuscript critically. JM proposed and conceived the research. MJ wrote the manuscript and designed the experimental system. All authors have read and approved the final version of the manuscript.
Authors’ original submitted files for images
Below are the links to the authors’ original submitted files for images.
Rights and permissions
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
About this article
Cite this article
Pavlovčič, U., Diaci, J., Možina, J. et al. Characterization of the headtotrunk orientation with handheld optical 3D apparatus based on the fringe projection technique. BioMed Eng OnLine 12, 96 (2013). https://doi.org/10.1186/1475925X1296
Received:
Accepted:
Published:
DOI: https://doi.org/10.1186/1475925X1296