- Open Access
A medium invasiveness multi-level patient’s specific template for pedicle screw placement in the scoliosis surgery
© The Author(s) 2017
- Received: 25 July 2017
- Accepted: 7 November 2017
- Published: 14 November 2017
Several methods including free-hand technique, fluoroscopic guidance, image-guided navigation, computer-assisted surgery system, robotic platform and patient’s specific templates are being used for pedicle screw placement. These methods have screw misplacements and are not always easy to be applied. Furthermore, it is necessary to expose completely a large portions of the spine in order to access fit entirely around the vertebrae.
In this study, a multi-level patient’s specific template with medium invasiveness was proposed for pedicle screw placement in the scoliosis surgery. It helps to solve the problems related to the soft tissues removal. After a computer tomography (CT) scan of the spine, the templates were designed based on surgical considerations. Each template was manufactured using three-dimensional printing technology under a semi-flexible post processing. The templates were placed on vertebras at four points—at the base of the superior-inferior articular processes on both left–right sides. This helps to obtain less invasive and more accurate procedure as well as true-stable and easy placement in a unique position. The accuracy of screw positions was confirmed by CT scan after screw placement.
The result showed the correct alignment in pedicle screw placement. In addition, the template has been initially tested on a metal wire series Moulage (height 70 cm and material is PVC). The results demonstrated that it could be possible to implement it on a real patient.
The proposed template significantly reduced screw misplacements, increased stability, and decreased the sliding & the intervention invasiveness.
The pedicle screw placement in scoliosis of the lumbar and thoracic spine has been extensively used in the surgical community . The Pedicle screw fixation is used in spinal fusion surgery to fuse vertebrae together securely in a fixed position . These devices provide stability and secure the spine after surgery and keep bone grafts in position during the spine recovery .
Correct placement of pedicle screws, in the lumbar and thoracic spine, needs to have a well 3D sensation and perception of the pedicle morphology for accurate identification of the ideal screw axis [4, 5]. It has been a problem, due to variations in anatomical shapes, dimension and orientation, which can cause the inefficiencies of treatment or severe injury to neurological structure [6, 7].
At the present time, the pedicle screw placements are applied by free-hand technique [8–13] or be performed under fluoroscopic guidance [14–19]. Other methods have also been conducted on a limited basis, such as image-guided navigation [20–25], computer-assisted surgery system [26–28], robotic platform [1, 29–32] and patient’s specific templates [33–45].
Free-hand procedure usage avoids any complexity but has error in rang 10–40% [46–48]. Image-guided navigation method reduces screw placement errors [49–52] but requires correct positioning and orientation of the drill to keep stability, which is not always easy to do. Furthermore, this technique is expensive and is used by a limited number of surgeons, due to the difficult challenges.
The robotic platform effectively reduces screw misplacement [53, 54] but it may not be a practical technique for small hospitals, which perform a limited number of spine fixation, due to the high cost and a long term learning process. According to studies conducted by Ludwig et al. , 18% of the pedicles of human cadaveric cervical spines, which were implanted with computer-assisted image-guided surgical system, had a critical violations.
Another different approach, which is cheaper with less complexity, could be the use of patient’s specific templates . These templates are very similar to dental implants surgical-guides which help to guide the drill in right direction. Nevertheless, in most designs, it is necessary to remove a huge region of the soft tissues and expose completely a large portions of the spine in order to access fit entirely around the vertebrae [37, 41].
A comprehensive literature review has been performed on various aspects of Patient’s specific templates. It is evident from the literature review that there is approximately no report on the design of templates by considering three factors including stability, medium invasiveness and easy verification, simultaneously. The outcome of our study would be extremely useful as the technology charts for designing medium invasiveness patient’s specific templates which are false-stable and not easily verifiable for pedicle screw placement in the spine.
The exact anatomical size of supra-spinous ligament, that covered the bone, is not clear.
The use of transverse processes requires a larger bone exposure.
Failure to use these points causes instability. The result of this study can fix this defect using designed multi-level template. Therefore the remaining required supporting points have been considered on another spinal level at the base of the articular processes.
Literature review (tabular form)
Location of the supporting points
Tominc et al. 
Fit and lock on the lamina
1st to 7th thoracic vertebrae
On the lamina at the base of the superior articular process on both sides and at the tip of the spinous process
Ferrari et al. 
Complementary likeness of the bone surface
Thoracic and lumbar vertebrae
On the lamina, articular and spinous processes
Ma et al. 
Inverse of the vertebral posterior surface
1st to 12th thoracic vertebrae
On the lamina, superior articular, spinous and transverse processes
Lu et al. 
Inverse of the spinous, lamina, and transverse processes
2nd to 12th thoracic vertebrae
On the spinous process, lamina, and transverse processes
Lu et al. 
Inverse of C2 spinous process and lamina
2nd cervical vertebrae
On the spinous process, lamina, and lateral masses of C2
Lu et al. 
Inverse of the vertebral posterior surface
2nd to 7th cervical vertebrae
On the spinous process and lamina
Lu et al. 
Inverse of the vertebral posterior surface
12th thoracic to 5th lumbar vertebrae
On the posterior surface of the lumbar vertebra
Ryken et al. 
Inverse of the vertebral surface
5th to 6th cervical vertebrae
On the lamina, spinous and transverse processes
Ryken et al. 
Inverse of the vertebral surface
2nd cervical to 1st thoracic vertebrae
On the spinous process and lamina
Berry et al. 
Cervical, thoracic and lumbar vertebrae
On the posterior surface of the lamina plus a posterior support to fit the spinous process
Goffin et al. 
Inverse of the spinous process and lamina
1st and 2nd cervical vertebrae
On the spinous process and the left and right side of the lamina
Porada et al. 
1st to 4th lumbar vertebrae
On the surfaces of a vertebra’s transverse and spinous processes
Van Brussel et al. 
2nd to 4th lumbar vertebrae
A large contact area on the top of the spinous process and a small knife-edge area on the two transverse processes
3D model and 3D printer parameters related to the production
3D model parameters
3D printer parameters
Solid layer thickness
Print bed temperature
Support dual extrusion
In order to consider the strength of the designed template during surgery, the optimal conditions in fabrication parameters of FDM machine have been obtained to improve the strength of shafts. Two variables have remarkable impact on the fabricated product’s strength in the FDM machine. These two variables are Layer thickness and Fill angle .
Therefore, before fabricating drill guide templates, different samples regarding various conditions, mentioned in , were fabricated via FDM machine. Consequently, the optimal parameters which can achieve maximum strength were specified. Note that the values of 0.06 and 60 have been considered for layer thickness and fill angle respectively.
Regarding the considered values, the maximum (UTS) of each shaft is 38.67 MPa and the diameter of each shaft has been considered 5 mm. Therefore, it is obvious that every shaft can tolerate 75.8 N. Note that, 4 shafts were used for every drill guide template.
Moreover, based on information associated with force applied by surgeon on the drill guide template during surgery, the applied force would be variable between 30 and 50 N. Therefore, it could be concluded that the designed shafts would not be broken during surgery.
In addition, the orientation of shafts has been designed to guarantee drill guide template during surgery. Finally, the experimental condition demonstrated that the drill guide template is stable during surgery and can tolerate surgeon force properly.
As mentioned in the introduction section, using free-hand technique for pedicle screw placement is followed by high risk of screws miss positioning [46–48]. Image-guided navigation method is expensive and not always easy task to be performed [49–52]. The robotic platform is not often used because of its high cost and difficulty in learning [53, 54].
Note that, 7 out of 103 screws, were implanted correctly but with less accurately (between 1 and 2 mm). This error can be ignored using free-hand procedure that has error in range 10–40% [46–48]. Therefore, it could be claimed that the obtained results, compared to previous studies, are acceptable. The errors are caused due to variations in anatomical shapes, dimension, and orientation in scoliosis cases. In addition, manufacturing process by 3D printing has dimensional errors because of the shrinkage during fabrication. Moreover, complex geometry, in the spine due to scoliosis, could lead to make error during process.
Results of surgery testing for 110 screws
Number of screws
Trajectory error < 1 mm
Trajectory error between 1 and 1.5 mm
Trajectory error between 1.5 and 2 mm
Distribution by vertebra of the screws that violated the pedicular cortex
Total number of screws
Misplaced screws (n)
Misplaced screws (%)a
All the multilevel templates, manufactured until now, have been associated to a high error rate due to the changes of vertebral bodies from the CT acquisition to the surgical table [42, 57, 58]. In this study, to provide a good accuracy for multi-level templates, the CT scan of spine was performed on patients lying in prone position to simulate equal facet joint relations during the operating. The bridges, between different levels of designed template, have also been processed under a semi-flexible post processing in order to achieve fit ideally on two or more vertebrae in vivo. This technique were significantly reduced the problems related to the changes occurred in each vertebral body.
To achieve easy verification of correct positioning, templates has been processed under a transparent post processing. During the in vivo trials, the surgeon found that template alignment was easy, due to the transparency of templates. We have found two previous research which have relatively similar results to us [62, 63]. Ferrari et al.  designed patient-specific templates for pedicle spine screws placement but their method has some limitations: the method is suitable only for single guide but ours used multi-level guides which have much superiority to single ones regarding less invasiveness, less surgery time, and less bone exposures. We also see that their proposed method suggests to fit the guide on all main areas of vertebra including spinous processes, lamina, and articular process. This leads to eliminate of considerable soft tissues. Our method has used only anterior and posterior articular process symmetrically which prevents to eliminate much soft tissues. These symmetrical areas could prevent the sliding of the guides and neutralize its freedom degrees without elimination of lamina and spinous soft tissues. Therefore, we think that our method is less invasive than Ferrari’s method. The most important point is that Ferrari’s proposal is not suitable for multi-level guides and scoliosis. The reason is that they have used drilled bushings, as supports on the vertebras, to keep stability. In our system, there is no any physical contact between bushing and vertebra surface. Therefore, it is possible to disregard the damaged vertebras, or the vertebras with abnormal geometries, as supporting areas (reference).
- 1.They have used spinous process, both lamina, and transverse processes as contact points but these points are not suggested due to following reasons:
The anatomical data of spinous process can not be specified due to supra-spinous. Therefore, we think that spinous process can not be used as contact point.
It is correct that usage of transverse processes and lamina increases the stability but a considerable amount of soft tissue shall be eliminated. Therefore, these could not be suggested as contact points.
We have used inferior articular process as contact point which avoids the above mentioned problems.
One of the weak points of that work is that the size of the guide is not suitable for multi-level in vivo surgeries. They have mentioned in result section that the guide has hit to L4 screw head due to its big size and they have used fluoroscopy method instead to fix the screws. In general, the big guides normally hit to the screws and other equipment and verification of the process is not easy. Furthermore, the method is not user friendly for surgeons. The big guide leads to elimination of much soft tissues as well.
The accuracy of their method for fixing the screws was 84% but our method’s accuracy is 96%.
The most important point in design of the guides, which are used in in vivo surgeries, is the geometry and size of the guide. In in vivo surgeries, the expose of soft tissue is an important factor concerning the time of surgery and related infection. Therefore, big size guides are only suitable for single level method not multi-level one. The multi-level guides could lead to minimum tissue expose.
With respect to the design principles, under the surgical considerations and noting to small and thin-fitting areas, it is feasible to decrease intervention invasiveness.
Using additional supporting points on another spinal level, at the multi-level templates, decreases the possibility to have a false-stable template positions.
Semi-flexible bridges, between different levels in multi-level templates, significantly reduce error rate due to the changes in each vertebral bodies.
Transparent multi-level patient’s specific template is easy to apply and give hint to the surgeon how to use the template correctly.
The outcome of this study would be extremely useful as the technology charts for designing medium invasiveness patient’s specific templates which are false-stable and not easily verifiable for pedicle screw placement in the scoliosis surgery.
FA performed the simulations and experimental methods and wrote the manuscript. KH and AS reviewed the results and edited the manuscript. FT advised the method of performing the research and commented on final draft of manuscript. All authors read and approved the final manuscript.
The authors declare that they have no competing interests.
Availability of data and materials
Consent for publication
Authors grant BioMed Central a license to publish the article and identify itself as the original publisher.
Ethics approval and consent to participate
The subjects have been complied by the World Medical Association Declaration of Helsinki regarding ethical conduct of research involving human subjects.
This research was not supported by any fund.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Van Dijk JD, Van Den Ende RP, Stramigioli S, Köchling M, Höss N. Clinical pedicle screw accuracy and deviation from planning in robot-guided spine surgery: robot-guided pedicle screw accuracy. Spine. 2015;40(17):E986–91.View ArticleGoogle Scholar
- Shi YM, Zhu FZ, Wei X, Chen BY. Study of transpedicular screw fixation on spine development in a piglet model. J Orthop Surg Res. 2016;11(1):1.View ArticleGoogle Scholar
- Boileau P, Gendre P, Baba M, Thélu CÉ, Baring T, Gonzalez JF, Trojani C. A guided surgical approach and novel fixation method for arthroscopic Latarjet. J Shoulder Elbow Surg. 2016;25(1):78–89.View ArticleGoogle Scholar
- Grigoriou E, Dormans JP. Pediatric spine trauma. in the growing spine. Berlin: Springer; 2016. p. 359–81.Google Scholar
- Legaye J. Three-dimensional assessment of the scoliosis. London: INTECH Open Access Publisher; 2012.View ArticleGoogle Scholar
- Parchi PD, Evangelisti G, Cervi V, Andreani L, Carbone M, Condino S, Ferrari V, Lisanti M. Patient’s specific template for spine surgery. In: Ritacco L, Milano F, Chao E, editors. Computer-assisted musculoskeletal surgery. Cham: Springer; 2016. p. 199–215.Google Scholar
- Oxland TR. Fundamental biomechanics of the spine—what we have learned in the past 25 years and future directions. J Biomech. 2016;46(6):817–32.View ArticleGoogle Scholar
- Polly DW, Yaszemski AK, Jones KE. Placement of thoracic pedicle screws. JBJS Essent Surg Tech. 2016;6(1):e9.View ArticleGoogle Scholar
- Parker SL, McGirt MJ, Farber SH, Amin AG, Rick AM, Suk I, Witham TF. Accuracy of free-hand pedicle screws in the thoracic and lumbar spine: analysis of 6816 consecutive screws. Neurosurgery. 2011;68(1):170–8.View ArticleGoogle Scholar
- Modi HN, Suh SW, Fernandez H, Yang JH, Song HR. Accuracy and safety of pedicle screw placement in neuromuscular scoliosis with free-hand technique. Eur Spine J. 2008;17(12):1686–96.View ArticleGoogle Scholar
- Karapinar L, Erel N, Ozturk H, Altay T, Kaya A. Pedicle screw placement with a free hand technique in thoracolumbar spine: is it safe? J Spinal Disord Tech. 2008;21(1):63–7.View ArticleGoogle Scholar
- Kim YJ, Lenke LG. Thoracic pedicle screw placement: free-hand technique. Neurology India. 2005;53(4):512.View ArticleGoogle Scholar
- Kim YJ, Lenke LG, Bridwell KH, Cho YS, Riew KD. Free hand pedicle screw placement in the thoracic spine: is it safe? Spine. 2004;29(3):333–42.View ArticleGoogle Scholar
- Su AW, McIntosh AL, Schueler BA, Milbrandt TA, Winkler JA, Stans AA, Larson AN. How does patient radiation exposure compare with low-dose O-arm versus fluoroscopy for pedicle screw placement in idiopathic scoliosis? J Pediatric Orthop. 2017;37(3):171–77.View ArticleGoogle Scholar
- Lang Z, Tian W, Liu Y, Liu B, Yuan Q, Sun Y. Minimally invasive pedicle screw fixation using intraoperative 3-dimensional fluoroscopy-based navigation (CAMISS Technique) for hangman fracture. Spine. 2016;41(1):39–45.View ArticleGoogle Scholar
- Yoshii T, Hirai T, Sakai K, Inose H, Kato T, Okawa A. Cervical pedicle screw placement using intraoperative computed tomography imaging with a mobile scanner gantry. Eur Spine J. 2016;25(6):1690–7.View ArticleGoogle Scholar
- Ohba T, Ebata S, Fujita K, Sato H, Haro H. Percutaneous pedicle screw placements: accuracy and rates of cranial facet joint violation using conventional fluoroscopy compared with intraoperative three-dimensional computed tomography computer navigation. Eur Spine J. 2016;25(6):1775–80.View ArticleGoogle Scholar
- Gelalis ID, Paschos NK, Pakos EE, Politis AN, Arnaoutoglou CM, Karageorgos AC, Xenakis TA. Accuracy of pedicle screw placement: a systematic review of prospective in vivo studies comparing free hand, fluoroscopy guidance and navigation techniques. Eur Spine J. 2012;21(2):247–55.View ArticleGoogle Scholar
- Hyun SJ, Kim YJ, Rhim SC, Cheh G, Cho SK. Pedicle screw placement in the thoracolumbar spine using a novel, simple, safe, and effective guide-pin: a computerized tomography analysis. J Korean Neurosurg Soc. 2015;58(1):9–13.View ArticleGoogle Scholar
- Linte CA, Moore JT, Chen EC, Peters TM. Image-guided procedures: tools, techniques, and clinical applications. In: Bioengineering for surgery. Elsevier; 2016. p. 59–90.Google Scholar
- Ryang YM, Villard J, Obermüller T, Friedrich B, Wolf P, Gempt J, Meyer B. Learning curve of 3D fluoroscopy image–guided pedicle screw placement in the thoracolumbar spine. Spine J. 2015;15(3):467–76.View ArticleGoogle Scholar
- Kantelhardt SR, Neulen A, Keric N, Gutenberg A, Conrad J, Giese A. Alternative radiation-free registration technique for image-guided pedicle screw placement in deformed cervico-thoracic segments. J Neurosurg Sci. 2015;61(5):464–72.Google Scholar
- Theologis AA, Burch S. Safety and efficacy of reconstruction of complex cervical spine pathology using pedicle screws inserted with stealth navigation and 3D image-guided (O-Arm) technology. Spine. 2015;40(18):1397–406.View ArticleGoogle Scholar
- Silbermann J, Riese F, Allam Y, Reichert T, Koeppert H, Gutberlet M. Computer tomography assessment of pedicle screw placement in lumbar and sacral spine: comparison between free-hand and O-arm based navigation techniques. Eur Spine J. 2011;20(6):875–81.View ArticleGoogle Scholar
- Metz LN, Burch S. Computer-assisted surgical planning and image-guided surgical navigation in refractory adult scoliosis surgery: case report and review of the literature. Spine. 2008;33(9):E287–92.View ArticleGoogle Scholar
- Tian W, Zeng C, An Y, Wang C, Liu Y, Li J. Accuracy and postoperative assessment of pedicle screw placement during scoliosis surgery with computer-assisted navigation: a meta-analysis. Int J Med Robot Comput Ass Surg. 2017.doi:10.1002/rcs.1732
- Williams J, Sandhu F, Betz R, George K. The use of dynamic surgical guidance [DSG] shortens the learning curve for accurate placement of pedicle screws: a cadaveric study. Bone Joint J. 2016;98(SUPP 5):1.Google Scholar
- Solitro GF, Amirouche F. Innovative approach in the development of computer assisted algorithm for spine pedicle screw placement. Med Eng Phys. 2016;38(4):354–65.View ArticleGoogle Scholar
- Hockel K, Maier G, Tatagiba M, Roser F. Spinal robotics: present indications and trends. In: Pinheiro-Franco J, Vaccaro A, Benzel E, Mayer H, editors. Advanced concepts in lumbar degenerative disk disease. Berlin: Springer; 2016. p. 591–600.Google Scholar
- Joung S, Park I. Medical robotics for musculoskeletal surgery. In: Zheng G, Li S, editors. Computational radiology for orthopaedic interventions. Cham: Springer; 2016. p. 299–332.Google Scholar
- Fujishiro T, Nakaya Y, Fukumoto S, Adachi S, Nakano A, Fujiwara K, Neo M. Accuracy of pedicle screw placement with robotic guidance system: a cadaveric study. Spine. 2015;40(24):1882–9.View ArticleGoogle Scholar
- Kim HJ, Lee SH, Chang BS, Lee CK, Lim TO, Hoo LP, Yeom JS. Monitoring the quality of robot-assisted pedicle screw fixation in the lumbar spine by using a cumulative summation test. Spine. 2015;40(2):87–94.View ArticleGoogle Scholar
- Tominc U, Vesel M, Al Mawed S, Dobravec M, Jug M, Herman S, Kreuh D. Personalized guiding templates for pedicle screw placement. In: Information and communication technology, electronics and microelectronics (MIPRO), 2014 37th international convention on IEEE; 2014. p. 249–251.Google Scholar
- Ferrari V, Parchi P, Condino S, Carbone M, Baluganti A, Ferrari M, Lisanti M. An optimal design for patient-specific templates for pedicle spine screws placement. Int J Med Robot Comput Ass Surg. 2013;9(3):298–304.View ArticleGoogle Scholar
- Ma T, Xu YQ, Cheng YB, Jiang MY, Xu XM, Xie L, Lu S. A novel computer-assisted drill guide template for thoracic pedicle screw placement: a cadaveric study. Arch Orthop Trauma Surg. 2012;132(1):65–72.View ArticleGoogle Scholar
- Lu S, Zhang YZ, Wang Z, Shi JH, Chen YB, Xu XM, Xu YQ. Accuracy and efficacy of thoracic pedicle screws in scoliosis with patient-specific drill template. Med Biol Eng Comput. 2012;50(7):751–8.View ArticleGoogle Scholar
- Lu S, Xu YQ, Zhang YZ, Xie L, Guo H, Li DP. A novel computer-assisted drill guide template for placement of C2 laminar screws. Eur Spine J. 2009;18(9):1379–85.View ArticleGoogle Scholar
- Lu S, Xu YQ, Lu WW, Ni GX, Li YB, Shi JH, Zhang YZ. A novel patient-specific navigational template for cervical pedicle screw placement. Spine. 2009;34(26):E959–66.View ArticleGoogle Scholar
- Lu S, Xu YQ, Zhang YZ, Li YB, Xie LE, Shi JH, Chen YB. A novel computer-assisted drill guide template for lumbar pedicle screw placement: a cadaveric and clinical study. Int J Med Robot Comput Ass Surg. 2009;5(2):184–91.View ArticleGoogle Scholar
- Ryken TC, Kim J, Owen BD, Christensen GE, Reinhardt JM. Engineering patient-specific drill templates and bioabsorbable posterior cervical plates: a feasibility study: technical note. J Neurosurg Spine. 2009;10(2):129–32.View ArticleGoogle Scholar
- Ryken TC, Owen BD, Christensen GE, Reinhardt JM. Image-based drill templates for cervical pedicle screw placement: laboratory investigation. J Neurosurg Spine. 2009;10(1):21–6.View ArticleGoogle Scholar
- Berry E, Cuppone M, Porada S, Millner PA, Rao A, Chiverton N, Seedhom BB. Personalised image-based templates for intra-operative guidance. Proc Inst Mech Eng. 2005;219(2):111–8.View ArticleGoogle Scholar
- Goffin J, Van Brussel K, Martens K, Vander Sloten J, Van Audekercke R, Smet MH. Three-dimensional computed tomography-based, personalized drill guide for posterior cervical stabilization at C1–C2. Spine. 2001;26(12):1343–7.View ArticleGoogle Scholar
- Porada S, Millner PA, Chibverton N, Berry E, Seedhom BB. Computer aided surgery with lumbar vertebra drill-guides, using computer aided planning, design and visualization. In: Poster session in proceedings medical image understanding and analysis; 2001.Google Scholar
- Van Brussel K, Vander Sloten J, Van Audekercke R, Berghe LV, Lauwereyns P, Fabry G, Swaelens B. A medical image-based drill guide for pedicle screw insertion: a cadaver study. J Biomech. 1998;31:39.View ArticleGoogle Scholar
- Wang VY, Chin CT, Lu DC, Smith JS, Chou D. Free-hand thoracic pedicle screws placed by neurosurgery residents: a CT analysis. Eur Spine J. 2010;19(5):821–7.View ArticleGoogle Scholar
- Merloz P, Tonetti J, Eid A, Faure C, Lavallee S, Troccaz J, Cinquin P. Computer assisted spine surgery. Clin Orthop Relat Res. 1997;337:86–96.View ArticleGoogle Scholar
- Farber GL, Place HM, Mazur RA, Jones DC, Damiano TR. Accuracy of pedicle screw placement in lumbar fusions by plain radiographs and computed tomography. Spine. 1995;20(13):1494–9.View ArticleGoogle Scholar
- Ishikawa Y, Kanemura T, Yoshida G, Ito Z, Muramoto A, Ohno S. Clinical accuracy of three-dimensional fluoroscopy-based computer-assisted cervical pedicle screw placement: a retrospective comparative study of conventional versus computer-assisted cervical pedicle screw placement: clinical article. J Neurosurg Spine. 2010;13(5):606–11.View ArticleGoogle Scholar
- Amiot LP, Lang K, Putzier M, Zippel H, Labelle H. Comparative results between conventional and computer-assisted pedicle screw installation in the thoracic, lumbar, and sacral spine. Spine. 2000;25(5):606–14.View ArticleGoogle Scholar
- Laine T, Lund T, Ylikoski M, Lohikoski J, Schlenzka D. Accuracy of pedicle screw insertion with and without computer assistance: a randomised controlled clinical study in 100 consecutive patients. Eur Spine J. 2000;9(3):235–40.View ArticleGoogle Scholar
- Schwarzenbach O, Berlemann U, Jost B, Visarius H, Arm E, Langlotz F, Ozdoba C. Accuracy of computer-assisted pedicle screw placement: an in vivo computed tomography analysis. Spine. 1997;22(4):452–8.View ArticleGoogle Scholar
- Kantelhardt SR, Martinez R, Baerwinkel S, Burger R, Giese A, Rohde V. Perioperative course and accuracy of screw positioning in conventional, open robotic-guided and percutaneous robotic-guided, pedicle screw placement. Eur Spine J. 2011;20(6):860–8.View ArticleGoogle Scholar
- Lieberman IH, Togawa D, Kayanja MM, Reinhardt MK, Friedlander A, Knoller N, Benzel EC. Bone-mounted miniature robotic guidance for pedicle screw and translaminar facet screw placement: part I—Technical development and a test case result. Neurosurgery. 2006;59(3):641–50.View ArticleGoogle Scholar
- Ludwig SC, Kowalski JM, Edwards CC, Heller JG. Cervical pedicle screws: comparative accuracy of two insertion techniques. Spine. 2000;25(20):2675–81.View ArticleGoogle Scholar
- Radermacher K, Portheine F, Anton M, Zimolong A, Kaspers G, Rau G, Staudte HW. Computer assisted orthopaedic surgery with image based individual templates. Clin Orthop Relat Res. 1998;354:28–38.View ArticleGoogle Scholar
- Merc M, Drstvensek I, Vogrin M, Brajlih T, Recnik G. A multi-level rapid prototyping drill guide template reduces the perforation risk of pedicle screw placement in the lumbar and sacral spine. Arch Orthop Trauma Surg. 2013;133(7):893–9.View ArticleGoogle Scholar
- Merc M, Vogrin M, Recnik G, Drstvensek I, Brajlih T, Friedrich T. Error rate of multi-level rapid prototyping trajectories for pedicle screw placement in lumbar and sacral spine. Chin J Traumatol. 2014;17(5):261–6.Google Scholar
- Ahn Sung-Hoon, Montero Michael, Odell Dan, Roundy Shad, Wright Paul K. Anisotropic material properties of fused deposition modeling ABS. Rapid Prototyp J. 2002;8(4):248–57.View ArticleGoogle Scholar
- Anitha R, Arunachalam S, Radhakrishnan P. Critical parameters influencing the quality of prototypes in fused deposition modelling. J Mater Process Technol. 2001;118(1):385–8.View ArticleGoogle Scholar
- Nidagundi VB, Keshavamurthy R, Prakash CPS. Studies on parametric optimization for fused deposition modeling process. In: 4th International conference on materials processing and characterization, 2(4–5); 2015. p. 1691–99.Google Scholar
- Ferrari V, Parchi P, Condino S, Carbone M, Baluganti A, Ferrari M, Mosca F, Lisanti M. An optimal design for patient-specific templates for pedicle spine screws placement. Int J Med Robot. 2012;9(3):298–304.View ArticleGoogle Scholar
- Putzier M, Strube P, Cecchinato R, Lamartina C, Hoff EK. A new navigational tool for pedicle screw placement in patients with severe scoliosis: a pilot study to prove feasibility, accuracy, and identify operative challenges. Clin Spine Surg. 2017;30(4):E430–9.Google Scholar