- Open Access
Simulation of in vivo dynamics during robot assisted joint movement
© Bobrowitsch et al.; licensee BioMed Central. 2014
- Received: 27 August 2014
- Accepted: 5 December 2014
- Published: 16 December 2014
Robots are very useful tools in orthopedic research. They can provide force/torque controlled specimen motion with high repeatability and precision. A method to analyze dissipative energy outcome in an entire joint was developed in our group. In a previous study, a sheep knee was flexed while axial load remained constant during the measurement of dissipated energy. We intend to apply this method for the investigation of osteoarthritis. Additionally, the method should be improved by simulation of in vivo knee dynamics. Thus, a new biomechanical testing tool will be developed for analyzing in vitro joint properties after different treatments.
Discretization of passive knee flexion was used to construct a complex flexion movement by a robot and simulate altering axial load similar to in vivo sheep knee dynamics described in a previous experimental study.
The robot applied an in vivo like axial force profile with high reproducibility during the corresponding knee flexion (total standard deviation of 0.025 body weight (BW)). A total residual error between the in vivo and simulated axial force was 0.16 BW. Posterior-anterior and medio-lateral forces were detected by the robot as a backlash of joint structures. Their curve forms were similar to curve forms of corresponding in vivo measured forces, but in contrast to the axial force, they showed higher total standard deviation of 0.118 and 0.203 BW and higher total residual error of 0.79 and 0.21 BW for posterior-anterior and medio-lateral forces respectively.
We developed and evaluated an algorithm for the robotic simulation of complex in vivo joint dynamics using a joint specimen. This should be a new biomechanical testing tool for analyzing joint properties after different treatments.
Robots are very useful tools in orthopedic research. High repeatability and precision of the robotic movement as well as the control of applied forces/torques allow assessment of joint stability after different surgical reconstruction techniques , investigation of instability mechanisms of different implants  and so on. Additionally, a robot can be applied for measurements of dissipated energy as a frictional characteristic in an entire joint .
Osteoarthritis is a common degenerative joint disease, which leads to a loss of the excellent frictional properties of synovial joints such as the knee. The abrasive processes over a lifetime or arthritic inflammatory diseases can lead to cartilage surface degeneration. The result is increased wear and energy dissipation during daily movements .
As a result of the rising number of patients suffering from osteoarthritis, there is a need for further development of effective therapeutic approaches. Clinical, histological and imaging tests indirectly described the mechanical improvement of joint cartilage after osteoarthritis treatment. Therefore, tribological tests for mechanical characterization of the cartilage are also required.
Currently, three related tribological methods are prevalent. The first method was the so-called ‘pin-on-disc’ , where small cartilage/bone cylinders were cyclically rotated or reciprocated under constant axial load in order to determine the coefficient of friction. The second method was the pendulum. Here, cadaver knees were used as the fulcrum of a pendulum and the amplitude decay was used to calculate the friction coefficient [6, 7]. In the third method, a robot was used to measure friction coefficients in rabbit stifle joints by applying a linear movement .
A new method to measure dissipated energy as a friction characteristic in an entire joint was developed in our group [3, 9, 10]. Using a robot, ovine knee joints were physiologically, non-linearly moved under angle and force/torque control . The physiological joint flexion was determined by a passive path. To simplify the measurement method, we used a constant axial load and flexion velocity during the knee flexion , but under real conditions, the axial force and also the flexion velocity altered during stance and swing phases in a gait cycle. Taylor et al. described both the flexion motion and contact forces of the ovine knee joint throughout a gait cycle .
The aim of this study was to develop and evaluate an algorithm for a robot  in order to reproduce the complex in vivo joint dynamics in ovine knee specimens using the data of Taylor et al. . In vitro simulation of in vivo dynamics will provide a new biomechanical testing tool for analyzing joint properties after different treatments.
A robotic 6-degree-of-freedom setup (KUKA KR 60–3 robot, Augsburg, Germany; reproducibility: ±0.06 mm) including a universal force/torque sensor (ATI UFS: Theta SI1000-120; resolution: 0.25 N and 0.025 Nm) was used to perform axially loaded knee flexion.
The femoral epicondyles of the knee joint were palpated and marked with a black felt pen. The initial origin of the robotic coordinate systems of the base (tibia, fixed) and tool (femur, moving) were placed in the midpoint between the lateral and medial epicondyle. For the tibial and femoral local coordinate systems, the x-axes were defined from medial to lateral, the y-axes from posterior to anterior and the longitudinal z-axes from distal to proximal. The x-axis for each of the two coordinate systems was a flexion axis specified perpendicular to the respective femoral and tibial longitudinal z-axis, approximately coinciding with the epicondylar axis (Figure 1). Tibiofemoral movement was recorded in the tibia (base) coordinate system.
Recording of passive knee flexion
Similar to the previous study , passive knee flexion was recorded with seven different increments (0.1, 0.2, 0.3, 0.4, 0.5, 0.75 and 1°) for one specimen in order to determine the optimal angular resolution for the knee flexion motion. Passive knee flexion of another three specimens was recorded with the optimal increment. The optimal increment was determined as a minimal increment with which the desired flexion velocity could be performed by the robot.
Passive knee flexion was characterized in a way that it follows the path of minimal resistance. It has been shown that this path of passive flexion was unique for each knee joint and therefore describes the individual unloaded motion of the joint [12, 13]. Passive knee flexion was recorded with the robotic system using a combination of angle controlled and force/torque controlled motion . While the flexion movement around the local femur flexion medio-lateral axis was carried out in angle controlled mode, the remaining five degree of freedom were force/torque controlled at the same time. All forces and torques were specified to be zero (acceptable tolerance for the residual magnitude less than 2 N and 0.2 Nm respectively), except for the axial force (along the tibia longitudinal axis), where 10 N were applied to guarantee a contact of the two joint surfaces . The knee flexion motion was started by approximately 50° of the knee flexion angle which corresponds to the minimal knee flexion angle during the heel-strike recorded by Taylor et al. , then the knee was flexed for 30° with the specified increment (until the flexion angle of 80°). Each point of the passive flexion path was recorded after all force/torque conditions were met. Thus, for instance, after recording passive knee flexion with an increment of 0.5°, the path of the passive knee flexion consisted of 61 points (an initial point plus 60 points after rotations with an increment of 0.5°).
In order to optimize the robot movement around the knee flexion axis, the origins of the tibial (base) and femoral (tool) coordinate systems were newly defined in the knee rotation center. Therefore, passive knee flexion was performed twice: the first time using a manually determined knee rotation center and the second time using the calculated knee rotation center. Thereafter, Matlab software (Matlab R2013a, The Math-Works, Inc., Natick, MA, USA) was used for the data processing described below.
The knee rotation center was calculated as a minimal amplitude point  during tibiofemoral movement of the first recorded passive knee flexion. Ehrig et al. showed that the minimal amplitude method was less prone to noise  than the least-squares algorithm which was used for the same purpose in the previous study . Thus, the robotic flexion axis was corrected to the calculated knee rotation center which corresponded to that optimized axis and the passive knee flexion was recorded again.
Preparation of dynamics data for a robot-assisted knee flexion simulation
The axial contact force values in body weight (BW) for each of the seven increments were interpolated from the knee contact force values published by Taylor et al.  to corresponding gait cycle values of the flexion angle curves (see the example shown in Figure 2B). Thus, to calculate a set of axial forces for a gait cycle with a desired body weight in Newton (N), the set of the axial contact force data in BW had to be multiplied with the sheep’s body weight in N.
In vivo loaded knee flexion simulation
The flexion angle, angular velocity and axial force data sets were vectors of the same length. These three vectors of data comprised dynamics information for the simulation of knee flexion over one gait cycle. The loaded knee flexion path of the robot movement consisted of points (spatial positions of the robot) from the previously recorded passive flexion path. In order to reproduce the knee flexion during the gait cycle, we substituted the values of the interpolated flexion angle data for indexes of the corresponding points of the passive flexion path. Thus, the loaded knee flexion path was constructed from points of the passive flexion path to simulate the sheep’s knee flexion.
The robot performed the loaded knee flexion path in a superimposed force torque control mode. Only the axial force was actively simulated (controlled) by the robot. Medio-lateral and posterior-anterior forces resulted as a backlash of knee joint structures. The simulated body weights were 300 N, 300 N, 300 N, and 350 N corresponding to the sheep’s body weights. The in vivo loads were simulated for 200 gait cycles on each of the four knee specimens. The 60 initial gait cycles were considered as pre-conditional cycles, while the last 140 gait cycles were used for further data analysis.
Experiment with flexion velocity over 10°/s limit
As a precaution of our occupational health and safety department, the angular velocity of the robot was limited to 10°/s. This allowed work with the robot without a protective fence. In order to show the reaction of the robot control to target angular velocities above this 10°/s limit, the knee flexion was performed on one specimen with three different flexion velocity profiles. Three flexion velocity profiles were calculated for maximal flexion velocity of 10, 15 and 20°/s.
Comparison of simulated forces to in vivo forces
where m denoted four specimens, n denoted 140 gait cycles. F ij was the simulated force vector and F invivo was the in vivo force vector corresponding to Taylor et al. . Both force vectors consisted of 100 elements in body weight units.
where p denoted 100 elements of a force vector. F kij was an element of the simulated force and F k _ invivo was an element of the in vivo force.
Standard deviation of the simulated forces
where , m denoted four specimens, n denoted 140 gait cycles and p denoted 100 elements of a force vector.
The experiments on all knee specimens were successfully completed. Visual examinations of the specimens before, during and after joint dissection revealed that the bones, ligaments, menisci, joint cartilage and capsule were intact.
Determination of an optimal flexion increment
Influence of flexion velocities above the 10°/s limit
Comparison of simulated forces to in vivo forces
Standard deviation of the simulated forces
The total standard deviation σ total resulted in 0.025 BW for the axial contact force. In contrast, the posterior-anterior and medio-lateral forces showed higher total standard deviations σ total of 0.118 and 0.203 BW respectively.
In our study, we presented an algorithm for the KUKA KR 60–3 robot to simulate the complex flexion movement of an ovine knee including the altering axial load similar to the in vivo dynamics of a sheep’s knee described in the study of Taylor et al. . Knee flexion simulations with different body weights and flexion velocities were performed.The robot movement was based on points (spatial positions of the robot) recorded during passive knee flexion. Discretization of the flexion path allowed construction of the gait cycle path from points of the passive flexion path. The optimal increment (0.5°) of the flexion movement was determined for the robot. The robot motion control automatically reduced flexion velocity if the points of the flexion path were too close to each other. This effect was observed for increments of 0.4° and smaller (Figure 3).
During the stance phase (from 0 to 54% of the gait cycle ), the knee was flexed by about 7°, therefore, the smaller the flexion increment the more points could be used to apply different forces, making the simulation force curve smoother and more detailed. The increment of 0.5° was considered as optimal because the maximally allowed flexion velocity of 10°/s could be achieved. According to Taylor et al. , an estimated maximal knee flexion velocity amounted to 270°/s during a 1 m/s walking speed of a sheep. In contrast, our robot flexed the sheep knee approximately 27 fold slower than was measured in vivo . A slow knee joint flexion velocity was well suited to measure changes of the joint-dissipated energy, as was shown during measurements with the knee flexion velocity of 3°/s in the previous study .The use of the velocity profile with maximal target velocity of 20°/s showed the possibility to perform the gait cycle period almost two times faster than the period for the velocity profile with a maximal target velocity of 10°/s (Figure 4). It was done without a strong effect on the axial force simulation (Figure 5B). If the flexion velocity during the swing phase is considered of minor importance, the use of the velocity profile with a maximal target velocity of 20°/s may be appropriate for the faster performance of experiments. This could be important during the simulation of thousands of gait cycles because of specimen autolysis.
There were only two target points (Figure 5) between 39 and 52% of the gait cycle with the biggest axial force difference of almost 1 BW. Therefore, the stepwise force alteration between these points was accompanied by the biggest residual error (Figure 6B). Due to the correction of inconsistencies between the start and end of the in vivo measured knee flexion (see section “Preparation of dynamics data for a robot-assisted knee flexion simulation”), the simulated flexion and therefore the simulated axial force could differ from the in vivo curves of Taylor et al. . This could explain the increased residual error at the beginning of the gait cycle (Figure 6B).
Only the axial force was actively simulated by the robot. The medio-lateral and posterior-anterior forces resulted as a backlash of knee joint structures. The highest residual error was observed for the posterior-anterior force (ϵ total = 0.79 BW). We assume that in the absence of active muscles, the patellar or hamstring tendons could not contribute to the posterior-anterior force. Therefore, in our experimental set-up, the measured posterior-anterior force differed from in vivo values (Figure 6A). For future applications of our robot system, it could be very interesting to analyze the contribution of passive joint structures like ligaments, capsule, etc. to the development of posterior-anterior and medio-lateral forces, which are very important for joint stability. The Orthoload database (http://www.orthoload.com) can be used to transfer in vivo measured forces to our robot system. Then, with our robot system, it could be possible to analyze, for instance, soft tissue balancing after total joint replacement.
The sheep model is a well-established model for analyzing biological responses during healing [18, 19]. In the future, we intend to use the robot system presented for the in vitro establishment of an enzymatically induced osteoarthritis model. Hollande et al. found a significant reduction of collagen II content in cartilage from osteoarthritis knees in comparison to healthy knees. They explained this phenomenon through increased hydration which took place more on the cartilage surface than in the deep zone of the cartilage . We will simulate the in vivo knee dynamics in order to reveal the relationship between the number of gait cycles and joint cartilage damage (dissipated energy outcome) after injection of a certain quantity of enzyme. Thus, we will study the possibility of transferring the in vitro osteoarthritis model to an in vivo healing model of osteoarthritis.
The study presented an algorithm to simulate in vivo dynamics on ovine knees by the KR 60–3 robot. We used discrete points of the passive flexion path to construct a complex knee flexion motion similar to the motion measured on sheep in vivo. The robot applied an in vivo-like axial force profile with high reproducibility during the corresponding knee flexion. Posterior-anterior and medio-lateral forces were detected by the robot as a backlash of joint structures. Their curve forms were similar to the curve forms of corresponding in vivo measured forces, but in contrast to the axial force, they showed higher standard deviation and total residual error.
We gratefully acknowledge the financial support of Aesculap AG, Tuttlingen, Germany. We acknowledge the support from Deutsche Forschungsgemeinschaft and the Open Access Publishing Fund of Tübingen University. We thank Aline Naumann from the Institute of Clinical Epidemiology and Applied Biometry for the consultation in statistical methodology.
- Wellmann M, Bobrowitsch E, Khan N, Patzer T, Windhagen H, Petersen W, Bohnsack M: Biomechanical effectiveness of an arthroscopic posterior bankart repair versus an open bone block procedure for posterior shoulder instability. Am J Sports Med 2011,39(4):796–803. 10.1177/0363546510389991View ArticleGoogle Scholar
- Herrmann S, Kaehler M, Souffrant R, Rachholz R, Zierath J, Kluess D, Mittelmeier W, Woernle C, Bader R: HiL simulation in biomechanics: a new approach for testing total joint replacements. Comput Methods Programs Biomed 2012,105(2):109–119. 10.1016/j.cmpb.2011.07.012View ArticleGoogle Scholar
- Lorenz A, Rothstock S, Bobrowitsch E, Beck A, Gruhler G, Ipach I, Leichtle UG, Wulker N, Walter C: Cartilage surface characterization by frictional dissipated energy during axially loaded knee flexion–an in vitro sheep model. J Biomech 2013,46(8):1427–1432. 10.1016/j.jbiomech.2013.03.009View ArticleGoogle Scholar
- Neu CP, Reddi AH, Komvopoulos K, Schmid TM, Di Cesare PE: Increased friction coefficient and superficial zone protein expression in patients with advanced osteoarthritis. Arthritis Rheum 2010,62(9):2680–2687. 10.1002/art.27577View ArticleGoogle Scholar
- Shi L, Sikavitsas VI, Striolo A: Experimental friction coefficients for bovine cartilage measured with a pin-on-disk tribometer: testing configuration and lubricant effects. Ann Biomed Eng 2011,39(1):132–146. 10.1007/s10439-010-0167-3View ArticleGoogle Scholar
- Drewniak EI, Jay GD, Fleming BC, Crisco JJ: Comparison of two methods for calculating the frictional properties of articular cartilage using a simple pendulum and intact mouse knee joints. J Biomech 2009,42(12):1996–1999. 10.1016/j.jbiomech.2009.05.024View ArticleGoogle Scholar
- Teeple E, Elsaid KA, Fleming BC, Jay GD, Aslani K, Crisco JJ, Mechrefe AP: Coefficients of friction, lubricin, and cartilage damage in the anterior cruciate ligament-deficient guinea pig knee. J Orthop Res 2008,26(2):231–237. 10.1002/jor.20492View ArticleGoogle Scholar
- Mabuchi K, Fujie H: Use of robotics technology to measure friction in animal joints. Clin Biomech (Bristol, Avon) 1996,11(3):121–125. 10.1016/0268-0033(95)00062-3View ArticleGoogle Scholar
- Bobrowitsch E, Lorenz A, Jorg J, Leichtle UG, Wulker N, Walter C: Changes in dissipated energy and contact pressure after osteochondral graft transplantation. Med Eng Phys 2014,36(9):1156–1161. 10.1016/j.medengphy.2014.06.015View ArticleGoogle Scholar
- Walter C, Leichtle U, Lorenz A, Mittag F, Wulker N, Muller O, Bobrowitsch E, Rothstock S: Dissipated energy as a method to characterize the cartilage damage in large animal joints: an in vitro testing model. Med Eng Phys 2013,35(9):1251–1255. 10.1016/j.medengphy.2013.01.002View ArticleGoogle Scholar
- Taylor WR, Ehrig RM, Heller MO, Schell H, Seebeck P, Duda GN: Tibio-femoral joint contact forces in sheep. J Biomech 2006,39(5):791–798. 10.1016/j.jbiomech.2005.02.006View ArticleGoogle Scholar
- Wilson DR, Feikes JD, O'Connor JJ: Ligaments and articular contact guide passive knee flexion. J Biomech 1998,31(12):1127–1136. 10.1016/S0021-9290(98)00119-5View ArticleGoogle Scholar
- Wilson DR, Feikes JD, Zavatsky AB, O'Connor JJ: The components of passive knee movement are coupled to flexion angle. J Biomech 2000,33(4):465–473. 10.1016/S0021-9290(99)00206-7View ArticleGoogle Scholar
- Fujie H, Mabuchi K, Woo SL, Livesay GA, Arai S, Tsukamoto Y: The use of robotics technology to study human joint kinematics: a new methodology. J Biomech Eng 1993,115(3):211–217. 10.1115/1.2895477View ArticleGoogle Scholar
- Rudy TW, Livesay GA, Woo SL, Fu FH: A combined robotic/universal force sensor approach to determine in situ forces of knee ligaments. J Biomech 1996,29(10):1357–1360. 10.1016/0021-9290(96)00056-5View ArticleGoogle Scholar
- Marin F, Mannel H, Claes L, Durselen L: Accurate determination of a joint rotation center based on the minimal amplitude point method. Comput Aided Surg 2003,8(1):30–34. 10.3109/10929080309146100View ArticleGoogle Scholar
- Ehrig RM, Taylor WR, Duda GN, Heller MO: A survey of formal methods for determining the centre of rotation of ball joints. J Biomech 2006,39(15):2798–2809. 10.1016/j.jbiomech.2005.10.002View ArticleGoogle Scholar
- Duda GN, Eckert-Hubner K, Sokiranski R, Kreutner A, Miller R, Claes L: Analysis of inter-fragmentary movement as a function of musculoskeletal loading conditions in sheep. J Biomech 1998,31(3):201–210.View ArticleGoogle Scholar
- von Rechenberg B, Akens MK, Nadler D, Bittmann P, Zlinszky K, Neges K, Auer JA: The use of photooxidized, mushroom-structured osteochondral grafts for cartilage resurfacing–a comparison to photooxidized cylindrical grafts in an experimental study in sheep. Osteoarthritis Cartilage 2004,12(3):201–216. 10.1016/j.joca.2003.11.001View ArticleGoogle Scholar
- Hollander AP, Heathfield TF, Webber C, Iwata Y, Bourne R, Rorabeck C, Poole AR: Increased damage to type II collagen in osteoarthritic articular cartilage detected by a new immunoassay. J Clin Invest 1994,93(4):1722–1732. 10.1172/JCI117156View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.