- Open Access
Estimation of current density distribution under electrodes for external defibrillation
© Krasteva and Papazov; licensee BioMed Central Ltd. 2002
- Received: 4 October 2002
- Accepted: 16 December 2002
- Published: 16 December 2002
Transthoracic defibrillation is the most common life-saving technique for the restoration of the heart rhythm of cardiac arrest victims. The procedure requires adequate application of large electrodes on the patient chest, to ensure low-resistance electrical contact. The current density distribution under the electrodes is non-uniform, leading to muscle contraction and pain, or risks of burning. The recent introduction of automatic external defibrillators and even wearable defibrillators, presents new demanding requirements for the structure of electrodes.
Method and Results
Using the pseudo-elliptic differential equation of Laplace type with appropriate boundary conditions and applying finite element method modeling, electrodes of various shapes and structure were studied. The non-uniformity of the current density distribution was shown to be moderately improved by adding a low resistivity layer between the metal and tissue and by a ring around the electrode perimeter. The inclusion of openings in long-term wearable electrodes additionally disturbs the current density profile. However, a number of small-size perforations may result in acceptable current density distribution.
The current density distribution non-uniformity of circular electrodes is about 30% less than that of square-shaped electrodes. The use of an interface layer of intermediate resistivity, comparable to that of the underlying tissues, and a high-resistivity perimeter ring, can further improve the distribution. The inclusion of skin aeration openings disturbs the current paths, but an appropriate selection of number and size provides a reasonable compromise.
- Specific Resistivity
- Current Density Distribution
- Wearable Electrode
- Automatic External Defibrillator
- Circular Electrode
Defibrillation of the heart is widespread and well-established procedure for resuscitation of cardiac arrest victims . The most accessible approach for electrical cardiac therapy is via external electrodes, placed on selected locations on the surface of the thorax. The electrodes have large contact area (70–120 cm2)  and provide high and supposedly uniform current density distribution in the heart, needed for excitation of most myocardial cells, thus forcing them to return to normal rhythm. Many authors have investigated optimal electrode positions and sizes via two-dimensional (2D) [3, 4] and three-dimensional (3D) [5–7] finite-element method (FEM) models, with the aim of obtaining uniform current distribution in the heart. The uniformity is evaluated by the ratio of the maximum current (which could result in myocardial damage) and the threshold current needed for defibrillation. For example Camacho et al.  found values of 2 to 4.7 for the anterior electrode position. Panescu et al.  reported that about 25% of the myocardium volume could be subjected to current densities more than 4 times higher than the threshold density.
Another aspect of the problem is the predominance of high current density along the perimeter of large size electrodes applied on human skin. In defibrillation and electrosurgery it can lead to unwanted damage and even severe skin burns [8–12] or electroporation , under the electrode perimeter area. In transthoracic pacing, it results in strong excitation of sensory nerve endings and provokes skeletal muscle contractions and pain [14, 15].
To reduce these adverse effects, Wiley and Webster  suggested concentric segmented electrodes with higher resistance in the periphery, by the use of external resistors adjusted to equalise the currents in the separate segments. A similar approach has been considered in more detail by Kim and co-workers . They have proposed covering the electrode metal with resistive gel of increasing resistivity toward the periphery, according to a specific relation with respect to the electrode radius. An implementation following such a design was tested on patients undergoing atrial cardioversion . The skin damage after defibrillation was assessed by skin biopsy in selected points under the electrodes. The results showed less damage with the use of the modified electrodes compared to the standard ones. However, the authors noted that after separation of the cases where high energies and currents were applied, no difference was found in the skin damage data. This result was not explained. It might be due to breakdown of the higher resistivity electrode layer, or to the overall increase of the current density. Also, the skin resistivity and reaction to increased current density and temperature may very probably be nonlinear.
Another problem is related to skin irritation, resulting from long-term application of the electrodes, when a protective wearable defibrillator-monitor is used . In such cases certain specific electrode designs were developed, providing openings for improvement of skin aeration or "breathing" .
The aim of the present work is to assess the current density distribution under electrodes of different structure, including shape, size, interfacing layer thickness and specific conductivity.
The problem can be reduced to the solution of the pseudo-elliptic differential equation of Laplace type, provided the effects of quasi-stationarity are neglected.
Dirichlet boundary conditions imposed on the surface SEi in contact with the ith electrode at potential Vi:
Neumann boundary conditions state that the normal component of the derivative of the potential is zero in the remaining boundary plane (SB), not in contact with the electrodes:
The current density distribution is defined by the potential gradient and the specific conductivity σ of the different regions:
J(x,y,z) = σ (-gradV(x,y,z)), (4)
In this case, the propagation of the low frequency electromagnetic field is virtually free of the feedback action from eddy currents, as the human tissue is weakly conductive. Therefore, the skin effect can be neglected.
Using software for FEM modelling and 3D-computer graphics (parts of ANSYS 5.7 and MATLAB 5.2), a simplified 3D finite element model with over 50,000 eight-node tetrahedron elements was developed (Fig. 1b). Since the measurements were taken 0.5 mm under the electrode-skin interface, the geometry of the thorax was simplified and simulated as a cylindrical domain (10 cm radius; 10 cm height), with specific resistivity 20 Ωm. This value was chosen as an approximate average of very high and very low conductivities (soft tissue, blood and bone, lung air). The estimated inter-electrode resistance R ≈ ρ l/S ≈ 65Ω (l – distance between the electrodes, S – electrode surface), corresponds to the real conditions in defibrillation. The electrodes, of about 80 cm2 area and 1 mm thickness, were located on the surface of the upper and lower cylinder bases. The defibrillation voltage was applied on the group of nodes forming the exterior electrode surface. The inter-electrode potential difference used in all comparative studies was set at 1000 V. This value was selected for convenience, since the relative distribution is independent of the voltage applied. The interfacing gel was simulated by a thick (0.5 – 1.5 mm) low-resistivity (20 – 60 Ωm) layer under the electrodes. Thus a varying layer resistance was represented, in a range of 1.5% to 15% of the total thoracic resistance, i.e. value from 1 to 10 Ω.
Standard electrode configurations
Circular electrode analysis
Comparative results for two electrodes with different radii.
Electrode Radius [cm]
Interfacing layer (5 mm thickness)
R [Ω ]
Various electrode configurations
Commonly used rectangular electrodes (area ~80 cm2) exhibit high non-uniformity of the current density distribution at the electrode-skin interface. The profile across the corners shows a non-uniformity coefficient of 4.6 (Fig. 2a). The standard circular electrode with the same area yields 32% lower non-uniformity (Fig. 2b).
Smaller electrodes produce higher current density, with slightly lower non-uniformity (Fig. 3). However, the electrode radius cannot be less than 4.7 cm for the minimum area of 70 cm2, recommended for efficient defibrillation.
The resistance of the electrode-skin interfacing layer is a major determinant of the maximum current density in the distribution profile. Two electrodes with different radii show equal non-uniformity, assessed by the coefficient K= Jmax/Jmin, if the under-electrode layer specific resistivity is chosen to provide the same electrode resistance (Table 1). This result was confirmed also for interfacing layers of different thicknesses (Fig. 5). For example, a thick layer with lower specific resistance (ρ = 20 Ωm and 1.5 mm thickness) has the same performance as a thin layer with higher ρ (ρ = 60 Ωm and 0.5 mm thickness). This result did not confirm our expectations that the thickness of the interface has a certain straightening effect on the current lines. A higher layer resistance is associated with lower non-uniformity, but it should not add more than 3–5% to the total resistance of the defibrillation current path. Many authors investigated the advantage of covering the electrode metal with resistive layers of increasing resistivity toward the periphery [14, 21]. Such a technique seems technologically difficult and expensive for disposable electrodes. The use of a ring of higher resistivity along the electrode perimeter seems acceptable, as the resistance in the current pathway was not increased, the maximum periphery current was reduced by 12% and the non-uniformity coefficient dropped to K = 2.71 (1570/580 A/m2). The same result could be achieved with 3 times higher resistance of a uniform layer. However, the problem of technological difficulties remains open.
Various electrode structures with openings for skin "breathing" (Figs. 7,8,9,10,11), increase the effect of non-uniformity, as the current density under the vents drops strongly. However, the distribution non-uniformity becomes negligible with increased distance from the electrode surface, as evident in Fig. 6. The bell-shaped distribution in Fig. 12 is estimated for the central region of the cylindrical thorax model, 50 mm under the interfacing layer surface. Consequently, the current density distribution under various electrode structures relates to effects on the skin, rather than to the defibrillation efficiency.
The current density distribution non-uniformity of circular electrodes is about 30% smaller than that of square-shaped electrodes. The use of an interface layer of intermediate resistivity, comparable to that of the underlying tissues can further improve the distribution. A high-resistivity perimeter ring adds a further 13% improvement without increasing the total interface resistance, hence the resistance to the defibrillation current, which is an important advantage in defibrillation.
The inclusion of skin aeration openings for wearable electrodes disturbs the current paths, but an appropriate selection of number and size provides a reasonable compromise.
The authors would like to express their deep gratitude to Prof. I. Daskalov for the scientific guidance and help in the biomedical engineering aspect of the study. The authors thank the Technical University of Sofia for granting the use of the FEM software.
- International Liaison Committee on Resuscitation: Guidelines 2000 for Cardiopulmonary Resuscitation and Emergency Cardiovascular Care. Circulation 2000, 102: I86–90.Google Scholar
- Geddes LA: Electrodes for transchest and ICD defibrillation and multifunctional electrodes. In: Defibrillation of the heart. ICDs, AEDs and Manual (Edited by: Tacker WA). Mosby, St Louis 1994, 82–118.Google Scholar
- Fahy BJ, et al.: Optimal electrode configurations for external cardiac pacing and defibrillation: an inhomogeneous study. IEEE Transactions on Biomedical Engineering 1987, 34: 743–748.View ArticleGoogle Scholar
- Lehr JL, Ramirez IF, Karlon WJ, Eisenberg SR: Test of four defibrillation dosing strategies using a two-dimensional finite-element model. Medical & Biological Engineering & Computing 1992, 30: 621–628.View ArticleGoogle Scholar
- Camacho MA, Lehr JL, Eisenberg WJ: A three-dimensional finite element model of human transthoracic defibrillation: paddle placement and size. IEEE Transactions on Biomedical Engineering 1995, 42: 572–578. 10.1109/10.387196View ArticleGoogle Scholar
- Panescu D, Webster JG, Tompkins WJ, Stratbrucker RA: Optimization of cardiac defibrillation by three dimensional finite element modelling of the human thorax. IEEE Transactions on Biomedical Engineering 1995, 42: 185–191. 10.1109/10.341831View ArticleGoogle Scholar
- Jorgenson DB, Haynor DR, Bardy GH, Kim Y: Computational studies of transthoracic and transvenous defibrillation in a detailed 3D human thorax model. IEEE Transactions on Biomedical Engineering 1995, 42: 172–183. 10.1109/10.341830View ArticleGoogle Scholar
- Wiley JD, Webster JG: Analysis and control of the current distribution under circular dispersive electrodes. IEEE Transactions on Biomedical Engineering 1982, 29: 381–385.View ArticleGoogle Scholar
- Wiley JD, Webster JG: Distributed equivalent-circuit model for circular dispersive electrodes. IEEE Transactions on Biomedical Engineering 1982, 29: 385–389.View ArticleGoogle Scholar
- McNaughton GW, Wyatt JP, Byrne JC: Defibrillation – a burning issue in coronary care units. Scottish Medical Journal 1996, 41: 47–48.Google Scholar
- Battig CG: Electrosurgical burns and their prevention. JAMA 1968, 204: 1025–1029. 10.1001/jama.204.12.1025View ArticleGoogle Scholar
- Pagan-Carlo LA, Stone MS, Kerber RE: Nature and determinants of skin 'burns' after transthoracic cardioversion. Am J Cardiology 1997, 79: 689–691. 10.1016/S0002-9149(96)00845-4View ArticleGoogle Scholar
- Vogel U, Wanner T, Bultmann B: Extensive pectoral muscle necrosis after defibrillation: nonthermal skeletal muscle damage caused by electroporation. Intensive Care Medicine 1998, 24: 743–745. 10.1007/s001340050656View ArticleGoogle Scholar
- Kim Y, Schimpf PH: Electrical behavior of defibrillation and pacing electrodes. Proceedings of the IEEE 1996, 84: 446–456. 10.1109/5.486746View ArticleGoogle Scholar
- Birkui PJ, Trigano LA, Zoll PM, eds: Noninvasive transcutaneous cardiac pacing. Futura Publ Inc Mount Kisco, New York 1992, 179–191.Google Scholar
- Garcia LA, Pagan-Carlo LA, Stone MS, Kerber RE: High perimeter impedance defibrillation electrodes reduce skin burns in transthoracic cardioversion. American Journal of Cardiology 1998, 82: 1125–1127. 10.1016/S0002-9149(98)00571-2View ArticleGoogle Scholar
- Schott R: Wearable defibrillator. Cardiovasc J Nurs 2002, 16: 44–52.View ArticleGoogle Scholar
- Totman M, O'Leary J, Owen JM, Fincke RW: Defibrillation system. US Patent No 6374138 2002.Google Scholar
- Tacker WA, Geddes LA: The laws of electrical stimulation of cardiac tissue. Proceedings of the IEEE 1996, 84: 355–65. 10.1109/5.486739View ArticleGoogle Scholar
- Al Hatib F, Trendafilova E, Daskalov I: Thransthoracic electrical impedance during external defibrillation: comparison of measured and modeled waveforms. Physiological Measurement 2000, 21: 145–153. 10.1088/0967-3334/21/1/318View ArticleGoogle Scholar
- Papazov S, Kostov Z, Daskalov I: Electrical current distribution under transthoracic defibrillation and pacing electrodes. J Med Eng & Technol 2002, 26: 22–27. 10.1080/03091900110102454View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article: verbatim copying and redistribution of this article are permitted in all media for any purpose, provided this notice is preserved along with the article's original URL.