- Open Access
Heart defibrillation: relationship between pacing threshold and defibrillation probability
BioMedical Engineering OnLine volume 18, Article number: 96 (2019)
Considering the clinical importance of the ventricular fibrillation and that the most used therapy to reverse it has a critical side effect on the cardiac tissue, it is desirable to optimize defibrillation parameters to increase its efficiency. In this study, we investigated the influence of stimuli duration on the relationship between pacing threshold and defibrillation probability.
We found out that 0.5-ms-long pulses had a lower ratio of defibrillation probability to the pacing threshold, although the higher the pulse duration the lower is the electric field intensity required to defibrillate the hearts.
The appropriate choice of defibrillatory shock parameters is able to increase the efficiency of the defibrillation improving the survival chances after the occurrence of a severe arrhythmia. The relationship between pulse duration and the probability of reversal of fibrillation shows that this parameter cannot be underestimated in defibrillator design since different pulse durations have different levels of safety.
Life-threatening arrhythmias (LTA) such as ventricular fibrillation (VF) are very serious conditions that may lead to death in few minutes. VF is characterized by chaotic and asynchronous cardiomyocyte electrical activity which leads to ineffective heart pumping . It has a prevalence of approximately ~ 25–50% of people with out-of-hospital cardiac arrest (OHCA) [2,3,4]. LTA are one of the major causes of death around the world. Annually, 35 per 100,000 people experience OHCA globally, including adults and children, and this number increases to 62 per 100,000 people when only adults are taken in account .
Once LTA are diagnosed, a high-intensity electric field (HEF) must be applied in the patient as soon as a defibrillator is available in a procedure called defibrillation . For effective defibrillation, a critical mass (75–90%) of ventricular cardiomyocytes has to be excited at the same time . However, the excitation of this large number of cardiomyocytes requires the application of HEF which reaches around 100 V/cm or higher in some regions of the myocardium . A HEF of this magnitude may lead to acute myocardial injury by electroporation , depression of contractile function  and blockage of electrical conduction by necrosis . Furthermore, our research team has already demonstrated that HEF of such intensity is able to kill cardiomyocytes [12,13,14]. Nevertheless, even a non-lethal HEF can make the cell unexcitable, generating a substrate to arrhythmia re-induction . These side effects might be related to the low survival rates reported after OHCA, which have been stable at 7–8% for the last 30 years despite the improvements in treatment and the increased availability of automated external defibrillators (AEDs) in public places [4, 16]. In this context, several studies have been carried out with the aim of improving defibrillation procedure to increase its success rate whilst reducing its side effects.
Our aim was to show the efficacy of a simple and feasible method able to improve the defibrillatory procedure, through the study of the strength–duration (SxD) curves. SxD curves have been exhaustively studied for the heart, but for the first time we present a paired study with heart pacing SxD curves and defibrillation SxD curves for the same hearts; from these data, we also propose a relationship between heart pacing electric field (E) and defibrillation HEF as a possible indicator of heart damage risk.
Also, a previous study of our research team has shown that the ratio of lethal HEF to excitation threshold for isolated rat cardiomyocytes changes with stimuli duration and is maximal for 0.5-ms stimuli , which indicates that this duration would probably be safer for defibrillation. In this study, we confirmed the relationship between defibrillation safety and pulse duration through SxD curves. We correlated the required HEF intensity for defibrillation with the shock intensity required for heart pacing to verify whether the existence of a previously observed optimum duration for cardiomyocyte stimulation would be translated to a higher efficiency in rat heart defibrillation.
Adult male Wistar rats were euthanized under deep anesthesia and the hearts were removed and cannulated in less than 30 s, avoiding physiological function loss due to prolonged ischemia [17, 18]. Hearts weighed on average 2.46 ± 0.07 g.
Pacing strength–duration (SxD) curve (Fig. 1) shows the correlation between the stimulatory pulse duration and the mean pacing threshold (ET). The SxD curve was well adjusted by the Weiss–Lapicque equation (Eq. 1, R2 = 0.96), where Y is the HEF intensity corresponding to a pulse duration d, Erh is the rheobase value (field modulus when d is infinity) and cr is the chronaxie (pulse duration corresponding to twice the rheobase). Chronaxie and rheobase values were 1.820 ± 0.20 ms and 0.16 ± 0.01 V/cm, respectively,
Defibrillation probability curves
Figure 2 shows the defibrillation probability curves as a function of the applied HEF in V/cm for all tested durations. These curves are significantly different (p < 0.0001). When compared in pairs, the 0.2-ms curve was different from all the others, whilst 0.5-ms and 1-ms curves were similar but different from all other curves. The average values of HEF corresponding to 50% of defibrillation probability (HEF50) values in V/cm were obtained from the nonlinear fit of the survival test results (Fig. 2) and were significantly different from each other (p < 0.0001). The HEF50 of curves with duration of 0.2-, 0.5-, 1- and 3- ms was different from all the others. On the other hand, 5-, 8- and 10-ms curves presented very close HEF50 values with no significant differences between them. Then the increase in the defibrillatory pulse duration beyond 5-ms did not promote intensity reduction of the shocks required for defibrillation. The adjustment of these HEF50 values by the Weiss–Lapicque equation generated a defibrillation SxD curve (Fig. 3) with rheobase and chronaxie of the 4.17 ± 0.561 V/cm and 1.41 ± 0.235 ms, respectively.
Ratio of defibrillation probability to pacing threshold
The HEF50 obtained from the probability curves (Fig. 4) for each pulse duration is shown in Fig. 5. This graphical representation puts emphasis on how greater than the pacing threshold a shock should be to succeed in defibrillation. In general, the HEF50 was over 20 times the threshold (×Threshold), the only exception was the 0.5-ms pulses, which HEF50 was 17.65 ×Threshold. Normalized HEF50 were different (p < 0.001) and when compared in pairs, the 0.2-ms pulse was not different from 0.5-ms and 3-ms pulses, but it was different from all the others. 1-ms pulse was different from the 3-ms, and 3-ms was different from the 8-ms pulse. The HEF50 values of the sigmoid adjustment were also significantly different (p < 0.001). When compared in pairs, the HEF50 of the 0.5-ms curve (the only one below 20 ×Threshold) is different from all others. There was no difference in the comparison between the HEF50 of the 0.2-ms and 3-ms curves, between the 1-ms and 5-ms curves, and the 8-ms and 10-ms curves.
To better understand this last result, we normalized the SxD curves (SxD for pacing—Fig. 1, and SxD for defibrillation—Fig. 3) to their respective rheobase (Fig. 6). For durations of 10-, 8-, 5-, 3- and 1-ms, the values of rheobase normalized ET and HEF50 are similar. For durations below 3-ms, we observed that not only both ET and HEF50 values increase, but also the difference between them. For 1- to 0.5-ms, the HEF50 increase was smaller (18%) than the ET increase (58%). For 0.5- to 0.2-ms, both HEF50 and ET values increased similarly, about 104% and 134%, respectively; therefore, the HEF50/ET ratio is lower for 0.5-ms since the ET variation is greater than the HEF50 for this duration, whereas it is not so pronounced for the others.
The present study shows that, within certain limits, the longer the pulse duration is, the lower is the threshold intensity for pacing and defibrillation, as expected for the stimulation of excitable tissues [19, 20]. Herein, we show for the first time, to the best of our knowledge, the stimulation and defibrillation SxD curves for the same hearts, considering an applied E homogeneously, which generated values more reliable and preparation independent.
We observed the same behavior for SxD curves for pacing and for defibrillation. Pacing requires a small pulse strength to be successful, because when a small number of cells are excited, the action potential propagation occurs throughout the heart . Thus, only a small number of cells need to be submitted to a supra-threshold E. In contrast, for defibrillation, a simultaneous excitation of a large portion of the myocardium (75–90%)  is required to make the cells non-excitable for a period and to terminate the fibrillatory mechanisms.
During the E application, non-uniform potential gradient formation happens because the cardiac tissue is anisotropic, composed by muscle fibers oriented in multiple directions with layers of connective tissue [22, 23]. Also, the heart region subjected to a higher potential gradient is closer to the electrodes; consequently, this region is easily stimulated, while other regions might not be stimulated depending on the applied E strength. However, when a critical mass of cardiomyocytes must be depolarized at the same time, as in the case of defibrillation, E has to be increased to stimulate cells which are not close to the electrodes; as a result, defibrillatory E is much larger than ET [21, 24]. In addition, pacing occurs during diastole, when most ventricular myocytes are relaxed in a vulnerable period. However, during defibrillation, the myocytes are not synchronized, each group of cells may be in a different action potential phase, requiring even higher amplitudes to excite cells during their relative refractory period and then terminate the fibrillation wave fronts [22, 24]. Hence, during defibrillation, the closest regions to the electrodes are exposed to a much higher E than the ET. The transmembrane potential variation (ΔVm) of each myocyte is proportional to the applied E module , then the maximal ΔVm is observed in the near-electrode myocytes; moreover, during threshold pacing, we may assume that the maximal ΔVm in the myocytes of this region is the stimulation threshold (ΔVmT); as a result, it is constant and does not change according to stimuli duration . During defibrillation, the ΔVm can be expressed by HEFdefibrillatory/ET multiplied by the ΔVmT, where HEFdefibrillatory is the high-intensity electric field necessary to successful defibrillation, which means that the HEFdefibrillatory/ET could be taken as an indirect index of the induced ΔVm in the cardiomyocytes during defibrillation. This information is very important because it allows to infer which duration induced a lower ΔVm, since a high ΔVm may lead to electroporation and consequent cell death [27, 28]. We observed a lower HEFdefibrillatory/ET ratio (17.65) when we defibrillated with 0.5-ms pulses. Thus, using this pulse duration, the induced ΔVm in the cardiomyocytes was probably lower and, consequently, it may be safer to be used in defibrillation procedures.
Defibrillatory pulses with duration of 0.5-ms are probably better for defibrillating rat hearts since not only the HEF50 in ×Threshold is smaller, but cells are also less susceptible to injury for this duration . Although the defibrillation success × pulse duration depends on the animal study [24, 29], the use of a short pulse duration might improve defibrillation procedures in human hearts, as Semenov et al.  also argued, since the commercial defibrillators use pulses with 5- or 10-ms duration, i.e., near the rheobase [21, 22]. Despite the difference in heart size between rodents and human, a factor that can influence the cardiac arrest mechanisms , models using rodent hearts have several advantages as presented by Patten et al. . These models, such as the one used in this study, produce results that cannot be directly related to the clinical context, but that generate important results, especially on the understanding, diagnosis and treatment of conditions such as VF because of the unavailability of studies on human subjects for ethical reasons. However, due to the limitations of the models, for results of basic science to be translated into clinical practice, studies in larger mammals, whose heart size is more similar to that of humans, are needed.
A possible limitation of this work was the time between heart removal and cannulation finalization (30 s). However, it was not sufficient to cause ischemia impairment in previous studies [17, 18]; additionally, contractile and chronotropic impairment may be present due to prolonged experiment time and cumulative effect of consecutive shocks.
Despite the fact that the rat hearts were placed in a Langendorff-adapted preparation for a maximum time of 3 h and that this type of preparation leads to contractile and chronotropic function deterioration of the heart ranging from 5 to 10% per hour , we believe that the randomized choice of the pulse duration sequence could minimize changes in the outcomes that were implied by this deterioration. However, we did not note any significant change in the heart function during the experiments involving all hearts included in this work.
We hope that this work can bring important clinical implications in the future, leading to an optimization of commercial defibrillators only by changing the pulse duration. A simple reduction of the shock duration, even on a small scale, may possibly lead to a significant increase in the effectiveness of defibrillatory procedures.
Considering our results, it is possible to conclude that defibrillated rat hearts by 0.5-ms pulses are less likely to suffer from injuries since the relationship between defibrillation probability and pacing threshold was lower for this duration, indicating that the impairment is smaller because the induced potential is lower in this case.
This outcome, along with a greater stimulatory safety factor for the duration of 0.5-ms , supports the hypothesis that a defibrillatory shock with this duration would be better for reversing VF in rats. Still, further studies should be performed to identify possible mechanisms underlying this finding.
Materials and methods
The protocols for animal care and use were approved by the Institutional Committee for Ethics in Animal Research (IB/UNICAMP, No. 4355-1). All the animals received care in accordance with relevant guidelines and regulations.
Isolated heart preparation
Ten male Wistar rats with age ranging from 5 to 6 months and average weight of 535.3 g ± 6.4 g were used. The animals received intraperitoneal injection of sodium heparin (3000 IU/kg), and they were anesthetized with an anesthetic button of lidocaine (5 mg/kg) and with an intraperitoneal injection of thiopental sodium (80 mg/kg). Following the chest opening, the heart was quickly and carefully excised. The aorta was cannulated in a Langendorff-adapted preparation (Fig. 7), where the heart was retrogradely perfused with Krebs–Henseleit solution composed of salts with the following concentrations (mM): NaCl 115, NaHCO3 25, KCl 4.6, MgSO4 1.2 KH2PO4 1.2, glucose 11.0 and CaCl2 1.4, constantly carbonated with carbogen mixture (5% CO2 and 95% O2) to maintain the pH between 7.35 and 7.45, and heated to 37 °C ± 0.5 °C.
After cannulation, the hearts were housed in the internal reservoir of the stimulation chamber (Fig. 7), heated (37 °C) by hot water circulation in the external reservoir provided by a pump (developed and manufactured by the Center for Biomedical Engineering, Campinas, Brazil). The heart was positioned in the middle of the chamber, 4 cm distant from the parallel plates located on the sides of the stimulation chamber, i.e., the stimulation electrodes, what allowed a uniform distribution of HEF within the chamber , which can be calculated according to the following equation :
where E is the electric field module, I is the current flowing through the chamber, σ is the physiological solution conductivity, h is the submerged height of the electrodes and w is the electrode width. In our setup, the conductive bath had 3.9 cm height (h) per 10.0 cm width (w), constituting a total volume of 152 ml and calculated resistance of 45.06 Ω. The Krebs–Henseleit solution σ was assumed to be 0.014 S/cm .
After 10 min for heart rate stabilization, the cardiac electrophysiological signal (ECG) was captured by Ag/AgCl electrodes (ECG electrodes), amplified (gain = 2000) and filtered (high-pass filter: fch = 3 Hz; low-pass filter: fcl = 100 Hz) by a electrophysiological signal amplifier (developed and manufactured by the Center of Biomedical Engineering, Campinas, Brazil). The ECG trace was visualized in an oscilloscope (manufactured by Tektronix Inc. Beaverton, OR, USA, model TDS 2014C, 100 MHz bandwidth) (Fig. 7). The spontaneous heart rate was determined by measuring the interval between five ECG R-waves.
Stimulation electrodes were connected to a low-intensity electrical stimulator (LIS, developed and manufactured by Center for Biomedical Engineering, Campinas, Brazil). The pacing threshold was determined for seven pulse durations (0.2-, 0.5-, 1-, 3-, 5-, 8- and 10-ms, total wave duration). The pulse duration sequence was randomly chosen for each heart and the stimulus frequency was set to 20% above the measured spontaneous heart rate. The stimulus amplitude was increased until the heart rate was equalized with the stimulation rate; the heart rate was inferred through the use of a video signal edge detector (VED, developed and manufactured by the Center for Biomedical Engineering, Campinas, Brazil). VED was coupled to a video camera (Ikegami Tsushinki Co., LTD, Japan—ICD-31 mod.) and to a video monitor (Kodo Electronics Co, LTD, Seoul, Korea—mod. KBM1200S, Fig. 7). The voltage output of the VED was proportional to the displacement of the heart border. When the electrical output signal of the VED synchronized with the stimulatory pulses, we considered that the heart was being paced. The minimum electric field (E) that kept the synchronism was considered the ET. This protocol was repeated for each stimuli duration.
The fibrillator (developed and manufactured by the Center for Biomedical Engineering Center, Campinas, Brazil) was coupled to the stimulation electrodes and the VF was induced by delivering a sine wave signal, with 60 Hz, amplitude from 1 to 3 V/cm and duration from 0.5 to 2 s [10, 35]. Duration and stimuli amplitude were adjusted to induce VF which was detected by monitoring the ECG record. When VF was maintained for at least 2 min, it was considered sustained and the fibrillator was disconnected; otherwise, a new amplitude and duration combination was set and VF was re-induced.
Once sustained VF was confirmed, the defibrillation protocol was started. A high-intensity electrical stimulator (HIS, developed and manufactured by the Center for Biomedical Engineering, Campinas, Brazil) was coupled with the stimulation electrodes and a monopolar electrical stimuli was applied with the truncated exponential waveform (decay less than 10%, with variable voltage from 1 to 1000 V and duration from 0.2- to 10-ms). The pulse duration was randomly chosen and the amplitude was initially set to five times the ET for the chosen duration of the same heart. This procedure was repeated for stimuli amplitudes between 10 and 35 times ET, or until defibrillation was confirmed, i.e., return of a clear QRS complex on the ECG record combined with heart contraction (Fig. 8, square A shows a case of defibrillation failure, and square B shows a case of success). For each heart, the procedures for fibrillation and defibrillation were performed once for each pulse duration, with intervals of 5 min to stabilize heart rate. The sequence of pulses with different durations was randomly chosen for each heart.
Defibrillation probability curves were based on the relationship between defibrillation probability and applied HEF in V/cm and ×Threshold (HEF values applied normalized to ET) by survival analysis . The curves obtained were compared by the Mantel–Cox test and adjusted by the following equation:
where L is defibrillation probability, HEF50 is the average value of HEF corresponding to 50% of defibrillation probability and h is the Hill coefficient . HEF50 values in V/cm and ×Threshold for each duration were compared by F test.
Two SxD curves were obtained: pacing SxD curve was made with ET values and the defibrillation probability SxD curve was plotted with the average values of HEF50, in V/cm obtained from the survival analysis. Both SxD curves were adjusted by Weiss–Lapicque equation (Eq. 1).
The ratio of defibrillation probability to pacing threshold was plotted with the average values of HEF50, in ×Threshold obtained from the survival analysis.
Statistical significance index α adopted for all tests was 0.05. All analyses and tests were made with the software Prism 5.03 (GraphPad Software, San Diego, US).
Availability of data and materials
The datasets used and/or analyzed during the current study are available from the corresponding author on reasonable request.
- ΔV m :
transmembrane potential variation
- ΔV mT :
transmembrane potential variation threshold
automated external defibrillators
- c r :
- E :
- E rh :
- E T :
- f ch :
- f cl :
- h :
high-intensity electric field
- HEF50 :
the average value of HEF corresponding to 50% of defibrillation probability
- HEFdefibrillatory :
high-intensity electric field necessary to successful defibrillation
high-intensity electrical stimulator
low-intensity electrical stimulator
out-of-hospital cardiac arrest
standard error of the mean
video signal edge detector
- w :
Jalife J. Ventricular fibrillation: mechanisms of initiation and maintenance. Annu Rev Physiol. 2000;62:25–50.
Hollenberg J, Herlitz J, Lindqvist J, Riva G, Bohm K, Rosenqvist M, et al. Improved survival after out-of-hospital cardiac arrest is associated with an increase in proportion of emergency crew-witnessed cases and bystander cardiopulmonary resuscitation. Circulation. 2008;118:389–96.
Cobb LA, Fahrenbruch CE, Olsufka M, Copass MK. Changing incidence of out-of-hospital. JAMA. 2002;288:3008–13.
Ringh M, Jonsson M, Nordberg P, Fredman D, Hasselqvist-Ax I, Håkansson F, et al. Survival after public access defibrillation in Stockholm, Sweden—a striking success. Resuscitation. 2015;91:1–7. https://doi.org/10.1016/j.resuscitation.2015.02.032.
Berdowski J, Berg RA, Tijssen JGP, Koster RW. Global incidences of out-of-hospital cardiac arrest and survival rates: systematic review of 67 prospective studies. Resuscitation. 2010;81:1479–87. https://doi.org/10.1016/j.resuscitation.2010.08.006.
Travers AH, Perkins GD, Berg RA, Castren M, Considine J, Escalante R, et al. Part 3: Adult basic life support and automated external defibrillation: 2015 international consensus on cardiopulmonary resuscitation and emergency cardiovascular care science with treatment recommendations. Circulation. 2015;132:S51–83.
Li Y, Chen B. Determinants for more efficient defibrillation waveforms. In: Anaesthesia, pharmacology, intensive care and emergency APICE, vol. 25. 2014. p. 203–18.
Yabe S, Smith WM, Daubert JP, Wolf PD, Rollins DL, Ideker RE. Conduction disturbances caused by high current density electric fields. Circ Res. 1990;66:1190–203.
Guensch DP, Yu J, Nadeshalingam G, Fischer K, Shearer J, Friedrich MG. Evidence for acute myocardial and skeletal muscle injury after serial transthoracic shocks in healthy swine. PLoS ONE. 2016;11:1–14.
Yamaguchi H, Weil MH, Tang W, Kamohara T, Jin X, Bisera J. Myocardial dysfunction after electrical defibrillation. Resuscitation. 2002;54:289–96.
Waldmann V, Narayanan K, Combes N, Jost D, Jouven X, Marijon E. Electrical cardiac injuries: current concepts and management. Eur Heart J. 2018;39:1459–65.
Prado LNS, Goulart JT, Zoccoler M, Oliveira PX. Ventricular myocyte injury by high-intensity electric field: effect of pulse duration. Gen Physiol Biophys. 2017;36:521–9.
Goulart JT, de Oliveira PX, Bassani JWM, Bassani RA. The influence of cell dimensions on the vulnerability of ventricular myocytes to lethal injury by high-intensity electrical fields. Rev Bras Eng Bioméd. 2012;28:337–45. https://doi.org/10.4322/rbeb.2012.040.
de Oliveira PX, Bassani RA, Bassani JWM. Lethal effect of electric fields on isolated ventricular myocytes. IEEE Trans Biomed Eng. 2008;55:2635–42.
Tovar O, Tung L. Electroporation of cardiac cell membranes with monophasic or biphasic rectangular pulses. Pacing Clin Electrophysiol. 1991;14:1887–92.
Sasson C, Rogers MAM, Dahl J, Kellermann AL. Predictors of survival from out-of-hospital cardiac arrest a systematic review and meta-analysis. Circ Cardiovasc Qual Outcomes. 2010;3:63–81.
Bell RM, Mocanu MM, Yellon DM. Retrograde heart perfusion: the Langendorff technique of isolated heart perfusion. J Mol Cell Cardiol. 2011;50:940–50. https://doi.org/10.1016/j.yjmcc.2011.02.018.
Skrzypiec-Spring M, Grotthus B, Szelag A, Schulz R. Isolated heart perfusion according to Langendorff—still viable in the new millennium. J Pharmacol Toxicol Methods. 2007;55:113–26.
Lapicque L. Definition experimentale de l’excitabilite. Soc Biol. 1909;77:280–3.
Weiss G. Sur la possibilite de rendre comparables entre eux les appareils servant a l’excitation electrique. Arch Ital Biol. 1901;35:413–45. https://doi.org/10.5935/1678-9741.20140015.
Tacker WA, Geddes LA. The laws of electrical stimulation of cardiac tissue. Proc IEEE. 1996;84:355–65.
Dosdall DJ, Fast VG, Ideker RE. Mechanisms of defibrillation. Annu Rev Biomed Eng. 2010;12:233–58.
Al-Kadra A, Nikolski V, Efimov IR. The role of electroporation in defibrillation. Circ Res. 2000;87:797–804.
Ideker RE, Zhou X, Knisley SB. Correlation among fibrillation, defibrillation, and cardiac pacing. Pacing Clin Electrophysiol. 1995;18:512–25.
Klee M, Plonsey R. Stimulation of spheroidal cells—the role of cell shape. IEEE Trans Biomed Eng. 1976;23:347–54.
Aidley DJ. The physiology of excitable cells. Cambridge: Cambridge University Press; 1998.
Saulis G. Electroporation of cell membranes: the fundamental effects of pulsed electric fields in food processing. Food Eng Rev. 2010;2:52–73.
Chen C, Smye SW, Robinson MP, Evans JA. Membrane electroporation theories: a review. Med Biol Eng Comput. 2006;44:5–14.
Gomes PAP, De Magalhães Galvão K, Mateus EF. Excitability of isolated hearts from rats during postnatal development. J Cardiovasc Electrophysiol. 2002;13:355–60.
Semenov I, Grigoryev S, Neuber JU, Zemlin CW, Pakhomova ON, Casciola M, et al. Excitation and injury of adult ventricular cardiomyocytes by nano- to millisecond electric shocks. Sci Rep. 2018;8:1–12. https://doi.org/10.1038/s41598-018-26521-2.
Hamlin RL. Animal models of ventricular arrhythmias. Pharmacol Ther. 2007;113:276–95.
Patten RD, Hall-Porter MR. Small animal models of heart failure development of novel therapies, past and present. Circ Heart Fail. 2009;2:138–44.
Jones JL, Lepeschkin E, Jones RE, Rush S. Response of cultured myocardial cells to countershock-type electric field stimulation. Am J Physiol. 1978;235:H214–22.
Sabo MZ, de Godoy CMG, Gomes PAP. Desenvolvimento e testes de um estimulador elétrico de alta potência para estimulação por campo elétrico em coração isolado de rato. Rev Bras Eng Bioméd. 2001;17:13–7.
von Planta I, Weil MH, von Planta M, Bisera J, Bruno S, Gazmuri RJ, et al. Cardiopulmonary resuscitation in the rat. J Appl Physiol. 1988;65:2641–7.
Kaplan EL, Meier P. Nonparametric estimation from incomplete observations. J Am Stat Assoc. 1958;53(282):457–81.
The authors are grateful to the R&D team at CEB/UNICAMP and NMCE-Núcleo de Medicina e Cirurgia Experimental at Faculty of Medical Science-UNICAMP for the valuable technical support. This study was supported by CAPES (Coordination of Improvement of Higher Education Personnel, in Portuguese, scholarship to Priscila C. Antoneli) and FAPESP (Foundation for Research of the State of São Paulo, in Portuguese, Proc. N 2011/51199-6).
This study was supported by CAPES (Coordination of Improvement of Higher Education Personnel, in Portuguese, scholarship to Priscila C. Antoneli) and FAPESP (Foundation for Research of the State of São Paulo, in Portuguese, Proc. N 2011/51199-6).
Ethics approval and consent to participate
The protocols for animal care and use were approved by the Institutional Committee for Ethics in Animal Research (IB/UNICAMP, No. 4355-1).
Consent for publication
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Antoneli, P.C., Goulart, J.T., Bonilha, I. et al. Heart defibrillation: relationship between pacing threshold and defibrillation probability. BioMed Eng OnLine 18, 96 (2019). https://doi.org/10.1186/s12938-019-0715-5
- Ventricular fibrillation
- Electric stimulation
- Isolated rat heart preparation