- Open Access
Mobile GPU-based implementation of automatic analysis method for long-term ECG
© The Author(s) 2018
- Received: 6 December 2017
- Accepted: 23 April 2018
- Published: 3 May 2018
Long-term electrocardiogram (ECG) is one of the important diagnostic assistant approaches in capturing intermittent cardiac arrhythmias. Combination of miniaturized wearable holters and healthcare platforms enable people to have their cardiac condition monitored at home. The high computational burden created by concurrent processing of numerous holter data poses a serious challenge to the healthcare platform. An alternative solution is to shift the analysis tasks from healthcare platforms to the mobile computing devices. However, long-term ECG data processing is quite time consuming due to the limited computation power of the mobile central unit processor (CPU).
This paper aimed to propose a novel parallel automatic ECG analysis algorithm which exploited the mobile graphics processing unit (GPU) to reduce the response time for processing long-term ECG data. By studying the architecture of the sequential automatic ECG analysis algorithm, we parallelized the time-consuming parts and reorganized the entire pipeline in the parallel algorithm to fully utilize the heterogeneous computing resources of CPU and GPU.
The experimental results showed that the average executing time of the proposed algorithm on a clinical long-term ECG dataset (duration 23.0 ± 1.0 h per signal) is 1.215 ± 0.140 s, which achieved an average speedup of 5.81 ± 0.39× without compromising analysis accuracy, comparing with the sequential algorithm. Meanwhile, the battery energy consumption of the automatic ECG analysis algorithm was reduced by 64.16%. Excluding energy consumption from data loading, 79.44% of the energy consumption could be saved, which alleviated the problem of limited battery working hours for mobile devices.
The reduction of response time and battery energy consumption in ECG analysis not only bring better quality of experience to holter users, but also make it possible to use mobile devices as ECG terminals for healthcare professions such as physicians and health advisers, enabling them to inspect patient ECG recordings onsite efficiently without the need of a high-quality wide-area network environment.
- Automatic ECG analysis
- Parallel computing
- Mobile GPU
- Energy consumption
With the development of information and healthcare technologies, large number of people tend to monitor their health condition at home using wearable holters. Due to the unique nature of healthcare service, users typically follow a similar schedule and upload the electrocardiogram (ECG) data collectively, creating a high computational burden on the healthcare platform at a certain time of day. In particular, concurrent processing of long-term ECG data, which is an important assistant approach to capture intermittent cardiac arrhythmias, tortures the healthcare platform heavily. How to efficiently process long-term ECG data collectively is an important issue to be solved.
ECG, as one of the vital physiological data, shows the time evolution of the heart’s electrical activity, which is caused by distinct electrical depolarization–repolarization patterns of the heart. Disorders of heart rate or rhythm, or changes in the morphological patterns, are indicators of underlying diseases . For example, myocardial infarction, cardiomyopathy, and myocarditis can lead to obvious ECG changes. Various ECG heartbeat classification methods were proposed by researchers in previous studies [2, 3]. An automatic ECG heartbeat classification pipeline mainly consists of basic steps including noise reduction [4–6], QRS complex detection [7, 8], feature extraction [9–11], and heartbeat classification [3, 12–18]. Heartbeat classification is a critical step in automatic ECG analysis. In the past few years, heartbeat classification methods are still under development with novel algorithms being proposed continuously. Liang et al. used hidden markov models (HMMs) to classify patient ECG signals in the free-living environment . Lannoy et al. proposed weighted conditional random fields classifier for the automatic classification of heartbeats . Furthermore, with the development of machine learning, many artificial intelligent methods have been applied in ECG heartbeat classification. Ye et al. applied wavelet transform and independent component analysis separately to each heartbeat to extract morphological features, then support vector machine (SVM) was used for heartbeat classification . Yu et al. used wavelet transformation and probabilistic neural network (PNN) to classify the ECG heartbeats . Oliveira et al. employed a dynamic bayesian network to predict premature ventricular heartbeats . Lagerholm applied unsupervised clustering methods to partition the QRS complex into clusters, then self-organized neural networks were used to identify heartbeat types . However, most ECG heartbeat classification methods are time-consuming, especially for processing long-term ECGs such as 24-h long ECGs. Although healthcare cloud platforms have been widely built to collect and manage ECG data from large populations, it is technically infeasible to provide prompt feedback for a large number of concurrent ECG analysis requests for even a median platform managing ten thousands of users.
To reduce the burden of remote healthcare platforms, many researchers shifted analysis tasks from remote healthcare platforms to local mobile computing devices [19–21]. However, processing long-term ECG data on mobile computing devices is time-consuming and leads to poor quality of experience as it occupies too many central processor unit (CPU) computation resources which are limited on mobile devices. At the same time, the heavy computation demanded by long-term ECG analysis leads to a long period in which the CPU runs at full frequency, which greatly shortens the working hours of battery in case the ECG analyzing tasks are frequently launched, i.e., in a mobile device of a physician or a health advisor. With the aforementioned limitations of mobile devices, a fast and energy-efficient analysis algorithm for long-term ECG is urgently needed.
This paper aimed to propose a novel parallel automatic ECG analysis algorithm based on mobile graphics processing unit (GPU). The 24-h long ECG recordings collected from volunteers were utilized to evaluate the computing performance and energy efficiency of the parallel algorithm. The test was conducted on a smartphone named OnePlus 3 equipped with an Qualcomm Snapdragon 820 processor along with 6 GB memory and an integrated Adreno 530 GPU. The parallel automatic ECG analysis algorithm was optimized through workgroup size tuning, data vectorization, and zero memory copy technology to take full advantage of the potential of heterogeneous computation. Additionally, the optimized parallel automatic ECG analysis algorithm consumed less energy per second in average compared with the sequential automatic ECG analysis algorithm. This led to a significant energy saving and longer working hours of battery for mobile devices when they were served as ECG processing terminals. Note a preliminary version of this paper has been reported .
The subsequent sections of the paper are organized as follows. Section of “Methods” introduces the OpenCL parallel computing model, parallel optimization technologies, the sequential automatic ECG classification algorithm, and the parallel implementation of the sequential automatic ECG classification algorithm on mobile GPUs. The results including experimental environment, data source, classification performance of automatic ECG analysis algorithm, and parallel efficiency and energy efficiency are described in the section of “Results and discussion”. The final section summarizes the paper.
The OpenCL parallel computing model
Automatic ECG analysis algorithm
Detailed judgment rules of 5 types of ECG rhythms.
Type of rhythms
QRS complex width < 120 ms
0.75× average RR interval < current RR interval
Current RR interval < 1.2× average RR interval
QRS complex width < 120 ms
Current RR interval < 0.75× current average RR interval
Current RR interval + next RR interval < 2× current average RR interval
QRS complex width > 120 ms
Current RR interval < 0.75× current average RR interval
Current RR interval + next RR interval ≥ 2× current average RR interval
Alternating appearance of VPB and NB
Alternating appearance of 2 NBs and 1 VPB
Parallel implementation of automatic ECG classification method on OpenCL framework
Parallel optimization technologies
The parallel algorithm could obtain a higher speedup through workgroup size tuning, data vectorization, and zero memory copy technologies. The details as follows: (1) workgroup size tuning. Workgroup size is an important factor affecting program efficiency. No general formula is available to calculate the optimal workgroup size, besides some constraints. Larger workgroups are better at hiding memory latency, while they would lead to cache thrashing. Proper workgroup size must be determined considering the register usage and the presence of barrier instructions inside kernels. (2) Data vectorization technology. As vector operations are common in OpenCL framework, GPUs are designed to be particularly powerful in dealing with vectors. Native vector operations, including addition and multiplication, are supported by most of GPU architectures with the development of mobile GPU technologies. Meanwhile, GPU memory architecture has also been developed to be efficient in loading/storing batches of data concurrently. (3) Zero memory copy technology. Data transfer between CPU and GPU memory is quite slow. It is a common case that the time cost of pure computation in parallelized programs takes up only a small portion of the entire time cost. The intrinsic latency for data transfer between CPU and GPU memory is a big obstacle in parallelization practice on the traditional CPU/GPU heterogeneous architecture. The CPU and GPU in mobile devices are usually integrated in one chip owing to the compact hardware structure of smartphones, making it possible for them to share the same memory. The CPU can recognize the memory allocated by the GPU as a part of its own memory with the zero memory copy technology and access it directly. Instead of repeatedly copying data between CPU and GPU whenever needed, the algorithm spends only its efforts on the initial mapping that takes comparatively much lesser time.
In order to maximally utilize the computing resources, these technologies were applied on the proposed parallel program. For workgroup size, we started from 64, and scaled up and down until the peak performance was found. The result was that most tasks were most efficiently completed when workgroup size was 128, while a workgroup size of 256 offered the highest performance for the task morphological transformation, as they relied on less registers. For data vectorization, the data were loaded/processed/stored in batches of 128 bits for all the transactions. As 32-bit float-point numbers were employed, four numbers were packed as a vector and operations on them were re-written using vector instructions. For zero memory copy, it was utilized throughout the design of the entire parallel algorithm. When CPU requires processed data from GPU, data were mapped directly to host memory from GPU memory directly, as it was shown in Fig. 4.
The compiling and linking from Android application through Java and C/C++ to OpenCL could be complemented owing to the Java Native Interface designed to enable native libraries’ calls from Java code. The experimental environment is benchmarked on a smartphone named OnePlus 3 equipped with a Qualcomm Snapdragon 820 processor along with memory of 6 GB. In detail, the processor consists of two CPU cores running at 2.15 GHz and another two cores running at 1.6 GHz, together with an integrated Adreno 530 GPU in which 256 arithmetic and logic units run at 624 MHz.
In this study, a multi-parameter patient simulator (FLUKE ProSim 8), the heart rate of which was 60 beats/min, was used to generate 5 types of ECG signals. Then, our developed wearable Holter (Mini-Holter, 58 \(\times\) 50 \(\times\) 10 mm, 20 g, over 30 h of battery working time, single channel three leads, 150/250/500 Hz sampling frequency) was connected with the multi-parameter simulator via ECG lead wire cables to capture 40 5-min long ECG recordings at a sampling frequency of 150 Hz. A total of 3120 NB, 240 APB, 240 VPB, 600 BG and 400 TG were generated. These simulated ECG records were utilized to evaluate the classification performance of the automatic ECG analysis. On the other hand, a public ECG dataset named MIT-BIH arrhythmia database  (total 48 records including 74,695 NB, 2745 APB, and 5976 VPB) was also utilized for the classification performance. These ECG records were sampled at 360 Hz with 11-bit resolution over a 10 mV range. Heartbeat types of NB, APB and VPB were selected for identification in this paper.
To further verify the performance of the proposed method on clinical ECG records, seven subjects (2 males and 5 females) from the General Hospital of People’s Liberation Army (301 hospital) were recruited to generate independently representative dataset. The age of the subjects ranged from 49 to 75 years, reaching an average of 63 ± 9 years. Each subject wore the Mini-Holter (150 Hz sampling frequency) to collect a long-term ECG signal, the duration of which is 23 ± 1.04 h (ranging from 21.1 to 24.0 h). The clinical long-term ECG dataset consists of 673147 NB, 481 APB, 268 VPB, 2 BG, and 1 TG. All the subjects were requested to sign the informed consent form before the experiment. At the same time, abnormal arrhythmia heartbeats of the dataset were marked by an expert physician.
As Fig. 5 shown, Mini-Holter was placed on the skin of volunteer’s chest with glue-like gel to record a long-term ECG signal. The attached position was between the middle line of the body and the heart. Volunteers would wear Mini-Holters for 12–24 h as they go about daily routine. After ECG data collection was completed, the Mini-Holter would be taken down from volunteer’s body. The long-term ECG data was transferred to smartphone via Bluetooth. The first and last 2 min of data were cut off to ensure that the signal is stable as volunteers need time to attach/detach the Mini-Holter. The data was analyzed by aforementioned mobile GPU-based automatic ECG analysis method. Then, subjects could browse the analysis results and original ECG signals. At the same time, the executing time of the program was recorded backstage as well as the energy consumption using Trepn Profiler from Qualcomm. Experiments of power consumption were conducted at a starting battery life of 100% or close to 100%. Both parallel and sequential program were executed five times on each tested ECG record and results were saved to be averaged. As for throttling, we launched a third-party CPU adjuster to keep CPU running at full frequency while the tested experiments were conducted.
Classification performance of the sequential and parallel automatic ECG algorithm on synthetic ECG data
Type of rhythms
Number of records
Classification performance of the sequential and parallel automatic ECG algorithm on MIT-BIH arrhythmia database
Type of rhythms
Number of records
Classification performance of the sequential and parallel automatic ECG algorithm on clinical long-term ECG data
Type of rhythms
Number of records
Parallel program efficiency
In order to give users real-time response after their physiological data were transmitted to smartphones, it was considered only viable to run automatic ECG analysis on powerful servers. In this study, the focus was shifted from powerful computing servers to long-underestimated potential computing capability of smartphone-based GPU. Parallel structures in the sequential automatic ECG analysis algorithm were fully extracted, and then these specific parts were redesigned using a framework designed for computing heterogeneous platforms known as OpenCL.
The same 24-h long ECG signal was also selected to evaluate the computing efficiency of the parallel program in each step. As shown in Table 5 and Fig. 6, the time cost of the automatic ECG analysis algorithm was reduced from 7.57 to 1.45 s. The parallel automatic ECG analysis algorithm achieved a speedup of 5.22× compared with the sequential algorithm. Then the parallel algorithm was further optimized using workgroup size tuning, data vectorization and zero memory copy technologies. With the optimization mentioned above, the time cost of parallel algorithm was further reduced to 1.26 s, which achieved a speedup of 6.02×. In detail, the optimized parallel automatic ECG algorithm achieved a speedup of 23.05×, 6.99×, and 4.36× in the artifact removal, QRS complex detection, and waveform classification respectively. Meanwhile, the remained long-term ECG signals were also utilized to test the computing efficiency. As shown in Table 6, the average executing time of the sequential program and the parallel program on long-term ECG dataset (duration: 23.0 ± 1.0 h) were 7.018 ± 0.592 and 1.215 ± 0.140 s correspondingly. The parallel program achieved an average speedups of 5.81 ± 0.39×, which showed its great superiority over the sequential algorithm.
Speedup of the parallel automatic ECG analysis algorithm in each step on a 24-h long ECG signal
Sequential program (s)
Parallel program (s)
Parallel-optimized program (s)
Speedup (parallel vs. sequential)
Speedup (parallel-optimized vs. sequential)
QRS complex detection
ECG waveform classification
Speedup of the parallel program on the long-term ECG dataset collected from the 301 hospital
Sequential program (s)
Parallel program (s)
23.0 ± 1.0
7.018 ± 0.592
1.215 ± 0.140
5.81 ± 0.39×
Program energy efficiency
Power consumption for sequential and parallel algorithm on ECG data from 310 hospital
1860.95 ± 42.17
33.93 ± 1.95
1826.91 ± 21.44
12.16 ± 0.65
This paper proposed a parallel automatic ECG analysis algorithm based on smartphone GPUs. The processing steps of artifact removal, QRS complex detection, and ECG waveform classification were implemented in parallel according to time cost and parallel feasibility analysis of the sequential automatic ECG analysis algorithm. Compared with the sequential algorithm, the average time cost of the parallel automatic ECG analysis algorithm on the long-term ECG dataset from the 301 hospital was reduced from 7.018 ± 0.592 to 1.215 ± 01.40 s, and a speedup of 5.81 ± 0.39× was achieved. Additionally, the average power and execution time of the parallel automatic ECG analysis algorithm was less than that of the sequential automatic ECG analysis algorithm, and about 64.16% of the battery energy was saved. If the energy consumption from data loading was excluded, the parallel program could save 79.44% of the energy consumption.
The reduction of response time and battery energy consumption in ECG analysis not only bring better quality of experience to holter users, but also make it possible to use mobile devices as ECG terminals for healthcare professions such as physicians and health advisers, enabling them to inspect patient ECG recordings onsite efficiently without the need of a high-quality wide-area network (WAN) environment. This is very helpful to extend the use of onsite ECG monitoring in rural, outdoor or ambulatory situations. Moreover, the migration of ECG processing pipeline to mobile device without heavy CPU occupation paves the way for real-time identification of cardiac arrhythmias which is of great value in prevention of acute cardiovascular event (e.g., stroke and myocardial infractions). This would be the further direction of our research.
XF and QY designed this study and performed literature review; YL was responsible for concept study; RC performed data preparation and analysis; YC guided the project and revised the writing of the final manuscript. All authors read and approved the final manuscript.
The authors would like to thank Liyan Yin, Shangguan Weihua, Huhe Dai, and other related members who developed the sequential automatic ECG classification algorithm. At the same time, the authors would appreciate Jun Wang who helped to generate the arrhythmia ECG records to validate the performance of the automatic ECG analysis algorithm.
The authors declare that they have no competing interests.
Availability of data and materials
Consent for publication
All authors approved the publication of this manuscript.
Ethics approval and consent to participate
The research involving patient data described in this paper were approved by the Shenzhen Institute of Advanced Technology (SIAT) Institutional Review Board (IRB Number: SIAT-IRB-151115-H0084).
This work was supported in part of the National High Technology Research and Development Plan (863 Plan: SS2015AA020109), the Science and Technology Planning Project of Guangdong Province (2015B010129012), the National Nature Science Foundation (61379136), the Shenzhen Science and Technology Projects (JCYJ20170413161515911), Major Special Project of Guangdong Province (2017B030308007), and the Science and Technology Service Network Plan (STS Plan) Project of Chinese Academy of Sciences (KFJ-SW-STS-161).
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Menown I, Mackenzie G, Adgey A. Optimizing the initial 12-lead electrocardiographic diagnosis of acute myocardial infarction. Eur Heart J. 2000;21(4):275–83.View ArticleGoogle Scholar
- Pan J, Tompkins WJ. A real-time ORS detection algorithm. IEEE Trans Biomed Eng. 1985;3:230–6.View ArticleGoogle Scholar
- De Chazal P, O’Dwyer M, Reilly RB. Automatic classification of heartbeats using ecg morphology and heartbeat interval features. IEEE Trans Biomed Eng. 2004;51(7):1196–206.View ArticleGoogle Scholar
- Li C, Zheng C, Tai C. Detection of ECG characteristic points using wavelet transforms. IEEE Trans Biomed Eng. 1995;42(1):21–8.View ArticleGoogle Scholar
- Leski JM, Henzel N. Ecg baseline wander and powerline interference reduction using nonlinear filter bank. Signal Process. 2005;85(4):781–93.View ArticleMATHGoogle Scholar
- Sun Y, Chan KL, Krishnan SM. ECG signal conditioning by morphological filtering. Comput Biol Med. 2002;32(6):465–79.View ArticleGoogle Scholar
- Slimane Z-EH, Naït-Ali A. QRS complex detection using empirical mode decomposition. Digit Signal Process. 2010;20(4):1221–8.View ArticleGoogle Scholar
- Chouakri S, Bereksi-Reguig F, Taleb-Ahmed A. QRS complex detection based on multi wavelet packet decomposition. Appl Math Comput. 2011;217(23):9508–25.MathSciNetMATHGoogle Scholar
- Mazomenos EB, Biswas D, Acharyya A, Chen T, Maharatna K, Rosengarten J, Morgan J, Curzen N. A low-complexity ECGecg feature extraction algorithm for mobile healthcare applications. IEEE J Biomed Health Inform. 2013;17(2):459–69.View ArticleGoogle Scholar
- Saxena S, Kumar V, Hamde S. Feature extraction from ECG signals using wavelet transforms for disease diagnostics. Int J Syst Sci. 2002;33(13):1073–85.View ArticleMATHGoogle Scholar
- Cvetkovic D, Übeyli ED, Cosic I. Wavelet transform feature extraction from human PPG, ECG, and EEG signal responses to elf pemf exposures: a pilot study. Digit Signal Process. 2008;18(5):861–74.View ArticleGoogle Scholar
- Li P, Wang Y, He J, Wang L, Tian Y, Zhou T-S, Li T, Li J-S. High-performance personalized heartbeat classification model for long-term ECG signal. IEEE Trans Biomed Eng. 2017;64(1):78–86.View ArticleGoogle Scholar
- Liang W, Zhang Y, Tan J, Li Y. A novel approach to ECG classification based upon two-layered HMMS in body sensor networks. Sensors. 2014;14(4):5994–6011.View ArticleGoogle Scholar
- Ye C, Kumar BV, Coimbra MT. Heartbeat classification using morphological and dynamic features of ECG signals. IEEE Trans Biomed Eng. 2012;59(10):2930–41.View ArticleGoogle Scholar
- de Oliveira LS, Andreão RV, Sarcinelli-Filho M. Premature ventricular beat classification using a dynamic Bayesian network. In: IEEE EMBC annual international conference of the engineering in medicine and biology Society; 2011. p. 4984–7.Google Scholar
- Lagerholm M, Peterson C, Braccini G, Edenbrandt L, Sornmo L. Clustering ECG complexes using hermite functions and self-organizing maps. IEEE Trans Biomed Eng. 2000;47(7):838–48.View ArticleGoogle Scholar
- De Lannoy G, François D, Delbeke J, Verleysen M. Weighted conditional random fields for supervised interpatient heartbeat classification. IEEE Trans Biomed Eng. 2012;59(1):241–7.View ArticleGoogle Scholar
- Yu S-N, Chen Y-H. Electrocardiogram beat classification based on wavelet transformation and probabilistic neural network. Pattern Recognit Lett. 2007;28(10):1142–50.View ArticleGoogle Scholar
- Li P, Wang Y, Tian Y, Zhou T-S, Li J-S. An automatic user-adapted physical activity classification method using smartphones. IEEE Trans Biomed Eng. 2017;64(3):706–14.Google Scholar
- Ryoo D-W, Bae C. Design of the wearable gadgets for life-log services based on UTC. IEEE Trans Consumer Electron. 2007;53(4):1477–82.View ArticleGoogle Scholar
- Wang L-H, Hsiao Y-M, Xie X-Q, Lee S-Y. An outdoor intelligent healthcare monitoring device for the elderly. IEEE Trans Consumer Electron. 2016;62(2):128–35.View ArticleGoogle Scholar
- Fan X, Yao Q, Cai Y, Li Y. An efficient automatic electrocardiogram analysis method using smartphones. In: IEEE international conference on Consumer Electronics (ICCE); 2018.Google Scholar
- Hong J-H, Ahn Y-H, Kim B-J, Chung K-S. Design of opencl framework for embedded multi-core processors. IEEE Trans Consumer Electron. 2014;60(2):233–41.View ArticleGoogle Scholar
- Berjon D, Cuevas C, Moran F, Garcia N. GPU-based implementation of an optimized nonparametric background modeling for real-time moving object detection. IEEE Trans Consumer Electron. 2013;59(2):361–9.View ArticleGoogle Scholar
- Chavan MS, Agarwala R, Uplane M. Suppression of baseline wander and power line interference in ECG using digital IIR filter. Int J Circ Syst Signal Process. 2008;2(2):356–65.MATHGoogle Scholar
- Chiarugi F, Sakkalis V, Emmanouilidou D, Krontiris T, Varanini M, Tollis I. Adaptive threshold GRS detector with best channel selection based on a noise rating system. In: IEEE computers in cardiology; 2007. p. 157–60.Google Scholar
- Zhang F, Lian Y. QRS detection based on multiscale mathematical morphology for wearable ECG devices in body area networks. IEEE Trans Biomed Circ Syst. 2009;3(4):220–8.View ArticleGoogle Scholar
- Drew BJ, Califf RM, Funk M, Kaufman ES, Krucoff MW, Laks MM, Macfarlane PW, Sommargren C, Swiryn S, Van Hare GF. Practice standards for electrocardiographic monitoring in hospital settings. Circulation. 2004;110(17):2721–46.View ArticleGoogle Scholar
- Tan Y, Du L. Study on automatic analysis technology for dynamic ECG (electrocardiogram) signal. Microcomput Inf. 2009;18:235–7.Google Scholar
- Prineas RJ, Crow RS, Zhang Z-M. The minnesota code manual of electrocardiographic findings. Berlin: Springer; 2009.Google Scholar
- Niederhauser T, Wyss-Balmer T, Haeberlin A, Marisa T, Wildhaber RA, Goette J, Jacomet M, Vogel R. Graphics-processor-unit-based parallelization of optimized baseline wander filtering algorithms for long-term electrocardiography. IEEE Trans Biomed Eng. 2015;62(6):1576–84.View ArticleGoogle Scholar
- Mark R, Schluter P, Moody G, Devlin P, Chernoff D. An annotated ECG database for evaluating arrhythmia detectors. IEEE Trans Biomed Eng. 1982;29:600.Google Scholar
- Eager DL, Zahorjan J, Lazowska ED. Speedup versus efficiency in parallel systems. IEEE Trans Comput. 1989;38(3):408–23.View ArticleGoogle Scholar
- Hoque MA, Siekkinen M, Khan KN, Xiao Y, Tarkoma S. Modeling, profiling, and debugging the energy consumption of mobile devices. ACM Comput Surv. 2016;48(3):39.Google Scholar