 Research
 Open Access
 Published:
Referencefree determination of tissue absorption coefficient by modulation transfer function characterization in spatial frequency domain
BioMedical Engineering OnLine volume 16, Article number: 100 (2017)
Abstract
Background
Spatial frequency domain (SFD) measurement allows rapid and noncontact widefield imaging of the tissue optical properties, thus has become a potential tool for assessing physiological parameters and therapeutic responses during photodynamic therapy of skin diseases. The conventional SFD measurement requires a reference measurement within the same experimental scenario as that for a test one to calibrate mismatch between the real measurements and the model predictions. Due to the individual physical and geometrical differences among different tissues, organs and patients, an ideal reference measurement might be unavailable in clinical trials. To address this problem, we present a referencefree SFD determination of absorption coefficient that is based on the modulation transfer function (MTF) characterization.
Methods
Instead of the absolute amplitude that is used in the conventional SFD approaches, we herein employ the MTF to characterize the propagation of the modulated lights in tissues. With such a dimensionless relative quantity, the measurements can be naturally corresponded to the model predictions without calibrating the illumination intensity. By constructing a threedimensional database that portrays the MTF as a function of the optical properties (both the absorption coefficient μ _{ a } and the reduced scattering coefficient \(\mu^{\prime}_{s}\)) and the spatial frequency, a lookup table approach or a leastsquare curvefitting method is readily applied to recover the absorption coefficient from a single frequency or multiple frequencies, respectively.
Results
Simulation studies have verified the feasibility of the proposed referencefree method and evaluated its accuracy in the absorption recovery. Experimental validations have been performed on homogeneous tissuemimicking phantoms with μ _{ a } ranging from 0.01 to 0.07 mm^{−1} and \(\mu^{\prime}_{s}\) = 1.0 or 2.0 mm^{−1}. The results have shown maximum errors of 4.86 and 7% for \(\mu^{\prime}_{s}\) = 1.0 mm^{−1} and \(\mu^{\prime}_{s}\) = 2.0 mm^{−1}, respectively. We have also presented quantitative ex vivo imaging of human lung cancer in a subcutaneous xenograft mouse model for further validation, and observed high absorption contrast in the tumor region.
Conclusions
The proposed method can be applied to the rapid and accurate determination of the absorption coefficient, and better yet, in a referencefree way. We believe this referencefree strategy will facilitate the clinical translation of the SFD measurement to achieve enhanced intraoperative hemodynamic monitoring and personalized treatment planning in photodynamic therapy.
Background
Recently, the spatial frequency domain (SFD) measurement has attracted increasing interests since it allows rapid and noncontact widefield imaging of tissue optical properties [1,2,3,4,5]. Compared to the traditional nearinfrared imaging modalities in epiillumination mode, the emerging modality features scanfree widefield illumination together with mesoscopicscale detection, all benefiting from the implementation of the spatial modulated excitation. The advantages of the SFD mode lend itself well suited for imaging skin tissues during photodynamic therapy [6,7,8,9,10]. Specifically, by measuring the tissue absorption coefficients at two or more wavelengths, quantitative mapping of the concentrations of the tissue chromophores (dominant by oxy and deoxyhemoglobins in the nearinfrared window ~600–900 nm) is achieved [11,12,13], and thus hemodynamic and oxygenation status as well as vascular distribution for disease diagnosing, staging and therapeutic response assessment are obtained.
In 1998, Dognitz and Wagnières reported the first use of the SFD method for measuring the tissue optical properties [14]. A widefield light source modulated with radiallyvarying square wave was employed and both the diffuse reflectivity and the modulation depth of the backscattering light were used to recover the optical properties at a single point in space. In 2005, Cuccia et al. proposed a singlefrequency sinusoidal modulation based imaging, where images of the direct current (DC) and alternating current (AC) components of the modulated reflectance were simultaneously extracted using a phaseshifting demodulation technique, and pixelbypixel recovery of the optical properties was achieved from joint use of the AC and DC amplitudes [2, 15]. This approach has been a mainstream technique for SFD imaging. Since then, further studies on improved techniques such as the fast demodulation and depthresolved recovery, etc., as well as clinical applications, have been comprehensively reported [16,17,18,19,20,21,22,23].
Nevertheless, to our knowledge, all the reported SFD methods require a reference measurement to calibrate the mismatch between the absolute intensity and the model predictions of the diffuse reflectance. According to the explanations in Ref. 15 (Eqs. (22) and (23)), a reference phantom with the known optical properties must be measured within the same experimental setup and physical condition as those for the test one. In practice, this consistency can be individually affected by the net incident fluence, which is relevant to both the optical reflectivity and geometrical morphology of sample surface, as well as by the system modulation transfer function (MTF), which varies with the objectimage distance at each spatial location [24,25,26]. Therefore, it is usually difficult to prepare a universal reference measurement in clinical trials.
To address the adversity, we attempt in this study a referencefree SFD method for the determination of the optical properties. Instead of using the absolute reflectance amplitude, we herein employ the MTF to characterize the propagation of the modulated lights in tissue. This dimensionless relative quantity naturally enables correspondence of the SFD measurements to the model predictions without the intensity calibration. Through the establishment of a database that links the MTF to the tissue optical properties (both the absorption coefficient μ _{ a } and the reduced scattering coefficient \(\mu^{\prime}_{s}\)) and the modulation frequency (ranges from 0.05 to 0.30 mm^{−1}), the optical properties can be theoretically retrieved using the SFD measurement at one or more frequencies.
It is widely believed that separating the scattering from the absorption in SFD requires at least one highfrequency modulation (typically at the spatial frequency of f > 0.5 mm^{−1}), with the highfrequency response dominant by scattering and the lowfrequency response dominant by both absorption and scattering [1, 4, 27]. However, due to the lowpass feature of highscattering media such as tissues, intensity of the modulated diffuse reflectance attenuates severely as the modulation frequency increases, leading to great difficulties and complexities in detecting highfrequency modulated reflectance. In addition, the MTF, as a relative measure, also lacks the constraints on the absolute amplitude of the frequency response, and thus can further aggravate the inverse coupling between the absorption and scattering as the modulation frequency is not high enough. In view of the fact, we assume in the remainder of this manuscript a priori knowledge of the reduced scattering coefficient, and focus on the absorptiononly determination. The practicability of the assumption have been justified in some of the clinical applications such as intraoperative monitoring of tissue oxygenation, where tracking the dynamic variations in oxygen metabolism is a primary concern and requires fast and convenient determination of absorption coefficient, while the scattering background is reasonably regarded as a constant and easy to predeterminate using the space or timeresolved spectroscopy [28, 29].
Methods
Modulation transfer function
We begin with assuming a linear optical medium illuminated by a spatially modulated light in the xdirection of the Cartesian coordinate system. The SFD expression of the modulated source fluence is given as \(S(f) = A_{S}^{(0)} \delta (f) + A_{S}^{{(f_{x} )}} \delta (f  f_{x} )\), with \(A_{S}^{\left( 0 \right)}\) and \(A_{S}^{{(f_{x} )}}\) being the amplitude of the DC component and the amplitude of the AC component at the modulation frequency f _{ x }, respectively. The modulated source light gives rise to a reflected diffuse photon fluence modulated at the same frequency, with its SFD expression correspondingly given as \(R(f) = A_{R}^{(0)} \delta (f) + A_{R}^{{(f_{x} )}} \delta (f  f_{x} )\), with \(A_{R}^{\left( 0 \right)}\) and \(A_{R}^{{(f_{x} )}}\) being the DC and AC amplitudes, respectively.
By definition, the medium MTF at the modulation frequency is experimentally calculated as the ratio of the modulation depth in reflected fluence to that in the source, i.e.,
where M _{ S } and M _{ R } denote the modulation depths of the source and reflected fluence, respectively, i.e., \(M_{S} (f_{x} ) = A_{S}^{{(f_{x} )}} /A_{S}^{(0)}\) and \(M_{R} (f_{x} ) = A_{R}^{{(f_{x} )}} /A_{R}^{(0)}\).
In the real spatial domain, the spatial response of medium to excitation of a highcontrast line beam, \(R_{d}^{LSF} \left( x \right)\), is defined as the line spread function (LSF), and specified in the SFD by its onedimension (1D) Fourier transformation, referred to as the system transfer function (STF), i.e., \(H\left( {f_{\text{x}} } \right) = \int_{  \propto }^{ + \propto } {R_{d}^{LSF} \left( x \right) { \exp } \left( {  j 2\pi f_{\text{x}} x} \right)dx}\). The SFDSTF can be simply calculated as the ratio of the output AC amplitude to the input one, i.e., \(H(f_{x} ) = A_{R}^{{(f_{x} )}} /A_{S}^{{(f_{x} )}}\). Accordingly, Eq. (1) can be also given in the form of the SFDSTF
Note that although it is physically meaningless to define the MTF for the DC case, it converges to 1 as f _{ x } approaches to 0 in terms of Eq. (2). This implies that the MTF calculation can be extended to include the scenario of zero modulation frequency when necessary. Equation (2) is the basis for the following threedimension (3D) database construction.
Construction of 3D MTF database
In terms of Eq. (2), the MTF is essentially calculated as the normalized Fourier series expansion of the real spatial domain LSF. Therefore, the conventional photon propagation models, such as the Monte Carlo (MC) simulation, radiative transfer equation and its approximations, can be applied to predicting the MTF through the LSF calculation. Based on the reported optical properties of human skin and subcutaneous tissue [30, 31], and meanwhile taking into account the sensitivity of the prototype SFD measuring system, we calculate in this study the MTF in the spatial frequency range of 0.05 to 0.30 mm^{−1}, for μ _{ a } ranging from 0.005 to 0.100 mm^{−1} at a step of 0.005 mm^{−1} and \(\mu^{\prime}_{s}\) ranging from 0.4 to 2.2 mm^{−1} at a step of 0.2 mm^{−1}, while keeping a constant anisotropy factor of g = 0.9. The MTF values for all combinations of the 20 absorption coefficients and the 10 reduced scattering coefficients are obtained by the following steps: Firstly, the spatially resolved diffuse reflectance of tissue for a singlepoint incidence \(R_{d}^{PSF} \left( {x, y} \right)\), referred to as the point spread function (PSF), is predicted at a spatial resolution of 0.1 mm, using the steadystate MC simulation, with 10^{8} photons injected [32]. Secondly, the LSF \(\left( {R_{d}^{LSF} \left( x \right)} \right)\) is deduced from convoluting the PSF with the distribution function of a line source that extends infinitely in ydirection, simply achieved by binning (summing up) the 2D response along the ydirection. Thirdly, the MTF is calculated according to Eq. (2) in the frequency range of 0.05–0.3 mm^{−1} at a step of 0.01 mm^{−1}, by Fourier transforming \(R_{d}^{PSF} \left( x \right)\). Since our MC simulations have demonstrated that the intensity of the diffuse reflectance at a site 50 mm far from the source is below the shot noise level for all the above combinations of μ _{ a } and \(\mu^{\prime}_{s}\), the LSF calculation is truncated to a data length of 999 for the Fourier transform. In this way, a database of the MTF curves that correspond to the all 200 pairs of the optical properties can be established. Finally the linear interpolation procedure is applied to the MTF curves on a finer μ _{ a }grid, to enhance the resolution of the μ _{ a }determination to 0.001 mm^{−1}.
As examples, the MTF curves for varying μ _{ a } are illustrated in Fig. 1, at four fixed \(\mu^{\prime}_{s}\) values of 0.4, 1.0, 1.6 and 2.0 mm^{−1}, respectively. The calculations manifest the lowpass filtering effect of tissue on spatially modulated lights, where all the MTFs drop off with the increase in the spatial frequency but at a slowingdown slop as the scattering increases, and intuitively demonstrates the feasibility of using the MTF for the μ _{ a }determination. In addition, it is shown in Fig. 1 that the MTF curves shift up on whole with the increase in μ _{ a }, indicating an increase in the modulation depth of the diffuse reflectance with the increase in μ _{ a }, and thus inferring an more significant effect of μ _{ a } on the DC attenuation than the AC one. A further investigation, as shown in Fig. 2, analogously calculates the sensitivity of the MTF to the absorption coefficient (the μ _{ a }sensitivity), simply defined as the ratio of the MTF change to the absorption change, at the aforementioned set of \(\mu^{\prime}_{s}\). It is shown that the μ _{ a }sensitivity trends to achieve a higher value at lower absorption, and decreases on whole with the maximum moving toward the direction of frequency increase as \(\mu^{\prime}_{s}\) increases. This observations are implicitly in accordance with the past findings that the low frequency component is more sensitive to variations in μ _{ a } while the high frequency component is more sensitive to variations in \(\mu^{\prime}_{s}\), and could presage an optimized selection of the working frequency according to the \(\mu^{\prime}_{s}\)range [1, 18, 27].
Inversion methods
As a priori \(\mu^{\prime}_{s}\) is available, only one MTF measurement at a single frequency is sufficient to determine μ _{ a } from the established database simply by a lookup table method. To enhance the noise robustness, the inversion scheme can be generalized to the following least square optimization for multifrequency measurements.
where \(MTF_{m} \left[ {f_{x}^{(n)} } \right]\) and \(MTF\left[ {\mu_{a} , f_{x}^{(n)} } \right]\) are the measured and modelpredicted MTFs, respectively, with the latter rapidly calculated by interpolating the database; \(f_{x}^{(n)}\) \(\left( {n = 1,{ 2}, \ldots ,N} \right)\) is the nth spatial frequency. Owing to the monotonic dependence of the MTF on the absorption coefficient, the above minimization can be uniquely achieved when μ _{ a } approaching to the true values. Obviously, by employing the multifrequency optimization of the MTFs, the noise influence in the single measurement on the μ _{ a }determination can be greatly alleviated thanks to the averaging effect of the leastsquare fitting. While the multifrequency optimization provides more robust determination of μ _{ a } than the lookup table method does, it is essential in practice to make a compromise between the determination accuracy and the measurement cost.
For homogeneous optical media, the diffuse reflectance will ideally maintain the modulation frequency without spectrum broadening. In this case, a composited illumination of multifrequency modulation patterns is employed and the MTFs at multiple frequencies are extracted from the single snapshot reflectance image using the Fourier frequency spectrum analysis, and determine the bulk μ _{ a } using the leastsquare curvefitting method. To reduce the truncation errors, the raw data is weighted by a Blackman window function prior to the Fourier transform. The Blackman window is widely believed to be excellent in estimating the amplitudefrequency characteristics but suffers from degraded spectral resolution. As a result, a minimal frequency spacing should be assured for reliably extracting the multiple frequency components from the raw data, according to the field of view (FOV) and the spatial resolution of the system.
For inhomogeneous optical medium, due to adverse effect of the frequency spectrum broadening of the diffuse reflectance on selection of the multiple modulation frequencies, a successive illumination of multifrequency patterns is used to avoid the spectrum aliasing. To map the μ _{ a }distribution, the raw data successively measured at each frequency is then demodulated in a pixelbypixel fashion by employing the threephase amplitude demodulation technique [2]
where x _{ i } is the position of the ith pixel; I _{1}, I _{2} and I _{3} are the measured reflectance images at modulation frequency of f _{ x } with the phase offsets of 0, 2π/3 and 4π/3, respectively.
Prototype SFD system
The setup of the used prototype SFD measuring system is shown in Fig. 3. Light emitted from a lightemitting diode (LED) source at the wavelength of 660 nm (M660F1, Thorlabs, USA) is expanded and then coupled to a digital micromirror device (DMD) (LightCrafter 4500, Texas Instruments, USA) by a lens system. The spatial modulated patterns on the DMD are formed by assigning an 8bit value (0–255) to each micromirror with customized control software. To reduce the specular reflection artifact, the spatially modulated light is projected onto a sample with a slight angle (≈3°) relative to the normal of the sample. Diffuse reflectance images of 51.2 mm × 51.2 mm are captured by a 16bit, 512 × 512 pixel charge coupled device (CCD) camera (RoleraMGi Plus, QImaging, Canada) placed right above the sample. This configuration leads to an approximate sampling rate of 0.1 mm and a raw data length of 512, and accordingly, a minimum frequency spacing of 0.06 mm^{−1} should be assured as the composited illumination of multi frequency patterns is to be applied.
System calibration
In a realistic scenario, the overall MTF of the measurement is the combined contribution from both the MTF of the sample, MTF _{ sample }(f _{ x }), and that of the measuring system (both the DMD and CCD), MTF _{ system }(f _{ x }). Thus, the modulation depth of the measured reflectance, M _{ R }(f _{ x }), is given by
where M _{ I }(f _{ x }) is the known modulation depths of the input to the DMD. To obtain MTF _{ system }(f _{ x }), a BaSO_{4} plate which has been widely used as a 100% reflectance standard is illuminated with the modulated source, and the modulation depth of its reflectance, M _{ P }(f _{ x }), is measured. We then get
Now, with M _{ I }(f _{ x }) and MTF _{ system }(f _{ x }) available, the genuine sample MTF required for the μ _{ a }determination, MTF _{ sample }(f _{ x }), can be readily calculated from Eq. (6).
Results and discussion
To validate the proposed method, simulation, phantom and experimental investigations were performed.
Simulation validations
The simulation validation was conducted on a total of 36 homogeneous samples, with their optical properties coming from 36 combinations of μ _{ a } = 0.01, 0.02, …, 0.09 mm^{−1} and \(\mu^{\prime}_{s}\) = 0.4, 1.0, 1.4, 2.0 mm^{−1}, are tested. A composited illumination of multifrequency modulation patterns at frequencies of 0.07, 0.13, 0.19, and 0.25 mm^{−1} was used. The spatial domain diffuse reflectance for each sample was simulated with the following steps: Firstly, the LSF of each sample was generated following steps 1 and 2 in “Construction of 3D MTF database”, except that the number of the photons being run in the MC simulation was 10^{6} for the purpose of increasing the shot noise level. Secondly, the LSF was Fouriertransformed to extract the SFD responses at the DC and the four modulation frequencies, from which the real spatial domain response to the fourfrequency modulated source was recovered using the inverse Fourier transform. Thirdly, the recovered data was corrupted by a 20 dB Gaussian noise to emulate the realistic measurement.
For the inversion, the Fourier frequency spectrum analysis was firstly employed, and then μ _{ a } of each sample was extracted from the four measured MTFs using the leastsquare curvefitting method, with a priori knowledge on \(\mu^{\prime}_{s}\). The results are shown in Fig. 4, where samples are divided into 9 groups according to the true values of μ _{ a }. It is observed that, with increase of μ _{ a }, the discrepancy between the true and measured μ _{ a }values gradually increases, as a result of decreased μ _{ a }sensitivity, with a maximum of 0.003 mm^{−1} occurred at a sample case of μ _{ a } = 0.09 mm^{−1} and \(\mu^{\prime}_{s}\) = 1.0 mm^{−1}.
The accuracy of the proposed MTFcharacterizationbased method for the μ _{ a }determination is contingent on the measurement errors of MTFs as well as the errortolerance of the inversion algorithm. We define the measurement error of the MTF as
where \(MTF_{M} (\mu_{a} ,\mu^{\prime}_{s} ,f)\) and \(MTF_{T} (\mu_{a} ,\mu^{\prime}_{s} ,f)\) are the measured and the true sample MTFs at the spatial frequency f _{ x }, respectively. To assess the influence of the measurement errors of the sample MTF on the lookup table scheme of the μ _{ a }determination, we define the up and down error tolerance
where \(\mu_{a}^{k}\) is the kth absorption coefficient in the μ _{ a }grid with \(\mu_{a}^{k} < \mu_{a}^{k + 1}\). It is inferred from Fig. 1 that d _{ U } ≥ 0 and d _{ D } ≤ 0. Assuming that the MTFs for a fixed \(\mu^{\prime}_{s}\) vary as a linear function of μ _{ a }, the lookup table method can accurately determine μ _{ a } provided that the condition \(d_{D} (\mu_{a} ,\mu^{\prime}_{s} ,f_{x} )/2 \le e(\mu_{a} ,\mu^{\prime}_{s} ,f_{x} ) \le d_{U} (\mu_{a} ,\mu^{\prime}_{s} ,f_{x} )/2\) is satisfied. In this view, we defined the interval \([d_{D} (\mu_{a} ,\mu^{\prime}_{s} ,f_{x} )/2,\,d_{U} (\mu_{a} ,\mu^{\prime}_{s} ,f_{x} )/2]\) as the errortolerance range (ETR) of a measured MTF for the μ _{ a }determination. Figure 5 contrasts the MTFETRs with the measurement errors of the simulated MTFs for \(\mu^{\prime}_{s}\) = 0.4, 1.0, 1.4, 2.0 mm^{−1} at the spatial frequency of f _{ x } = 0.07 mm^{−1}, and Fig. 6 compares the MTFETRs and the measurement errors of the simulated MTFs for \(\mu^{\prime}_{s}\) = 1.4 mm^{−1} at the spatial frequencies of f _{ x } = 0.07, 0.13, 0.19, 0.25 mm^{−1}.
In Fig. 5, the MTFETRs decrease with the increase in \(\mu^{\prime}_{s}\), due to the decrease in the μ _{ a }sensitivity and the increase in the MTF value, and the measurement errors of the MTFs decrease conformably, indicating no clear correlation between \(\mu^{\prime}_{s}\) and the accuracy of the μ _{ a }determination. It is noted that in our simulations, the SNR is set to be the same for all the measured reflectance without regard to \(\mu^{\prime}_{s}\). But in practice, the SNR of the measured diffuse reflectance might increase with the increase in \(\mu^{\prime}_{s}\), owing to the increased intensity of the diffuse reflectance. Therefore it is expected that accuracy of the μ _{ a }determination could be improved with increasing \(\mu^{\prime}_{s}\).
In Fig. 6, with the increase in the spatial frequency, it is observed that both the ETRs and the measurement errors of MTFs trend to increase. As the spatial frequency reaches 0.25 mm^{−1}, the measurement errors of the MTFs are all beyond the corresponding ETRs. This implies that a lookuptablemethodbased retrieval of μ _{ a } at this frequency would be inaccurate. A reason for the degraded performance with the increase in the spatial frequency is that the high frequency components account for little of the whole spatial response, and thus suffer from deteriorated SNRs. It is thus concluded that low frequency modulation is more preferable as the lookup table method is employed.
Phantom experiments
Experiments were conducted on liquid tissuesimulating phantoms consisting of deionized water as the diluent, India ink as the absorber and Intralipid10% as the scatterer. According to the spectral absorbance of a preprepared diluted India ink measured by a spectrophotometer (UV2550, Shimadzu, Japan), and the reported scattering coefficient and the anisotropy of Intralipid10% [33], a set of the optical properties were obtained from 14 combinations of μ _{ a } = 0.01, 0.02, 0.03, 0.04, 0.05, 0.06, 0.07 mm^{−1} and \(\mu^{\prime}_{s}\) = 1.0, 2.0 mm^{−1} for the phantoms. A semiinfinite plane geometry was realized by a tank with a length of 120 mm, a width of 120 mm, and a height of 50 mm. The tank was painted black to reduce the wall reflection.
Each sample was illuminated by composited threefrequency modulation patterns with three frequency sets of f _{ x,1} = {0.06, 0.14, 0.22 mm^{−1}}, f _{ x,2} = {0.08, 0.16, 0.24 mm^{−1}} and f _{ x,3} = {0.10, 0.18, 0.26 mm^{−1}}. The diffuse reflectance images corresponding to the three frequency sets were captured successively, and then summed up along the ydirection for 1D Fourier transform along the x direction. Finally, the MTFs of each sample at the 9 modulation frequencies were calculated by Eq. (9).
For the μ _{ a }determination, we recombined the 9 measured MTFs of each sample into three sets, i.e., the lowfrequency set: LF = {MTF(f)f = 0.06, 0.08, 0.10}, the middlefrequency set: MF = {MTF(f)f = 0.14, 0.16, 0.18}, and the highfrequency set: HF = {MTF(f)f = 0.22, 0.24, 0.26}, respectively. We thereafter retrieved 27 absorption coefficients using the leastsquare curvefitting scheme, from 27 combinations of the threefrequency MTFs, with the MTFs in each combination coming from the LF, MF and HF, respectively. Boxplot graphs of the retrieved absorption coefficients are shown in Fig. 7a, b, for \(\mu^{\prime}_{s}\) = 1.0 mm^{−1} and \(\mu^{\prime}_{s}\) = 2.0 mm^{−1}, respectively, with the relevant statistical and error analyses listed in Table 1.
For the accuracy assessment of the μ _{ a }determination, the relative error between the expected absorption coefficient, \(\mu_{a}^{{expect}}\), and the mean of the measured absorption coefficient, \(\mu_{a}^{mean}\), is used in the above table, i.e., \(e = \left {\left( {\mu_{a}^{mean}  \mu_{a}^{{expect}} } \right)/\mu_{a}^{{expect}}} \right\).
Results in Table 1 demonstrate a maximum μ _{ a }determination error of 4.86% for \(\mu^{\prime}_{s}\) = 1.0 mm^{−1} and 7.00% for \(\mu^{\prime}_{s}\) = 2.0 mm^{−1}. A gradually increasing trend of the variance is observed as μ _{ a } increases. This is probably due to the fact that as μ _{ a } increase, the μ _{ a }sensitivity decreased while the measurement errors of the MTFs increased. In contrast, as \(\mu^{\prime}_{s}\) increases, decreased variance is observed in Table 1, which might be a result of the decreased measurement errors of the MTFs.
Exvivo imaging of a subcutaneous tumor
A subcutaneous xenograft model of human lung cancer (ATCC number: CCL185) in a 4weekold female nude mouse was investigated, as shown in Fig. 8. To obtain a reasonably flat sample, a specimen consisting of tumor, vessels, skin and subcutaneous tissues was cut out from the mouse and put on the upper surface of a solid tissuemimicking phantom made from polyformaldehyde. The phantom was 80 mm in length, 80 mm in width and 40 mm in height, with μ _{ a } = 0.0038 mm^{−1} and \(\mu^{\prime}_{s}\) = 1.0 mm^{−1} according to the product manual. The size of the tumor was ~7 mm in largest dimension. The 3D surface profile of the specimen was measured using the prototype system with the upper surface of the phantom being taken as the reference. The surface height map is reconstructed using a phaseshifting profilometry, and shown in Fig. 8c, with the maximum height of the tumor being approximated 2.4 mm [34].
To achieve spatially resolved imaging of the specimen, the sample was sequentially illuminated three times at the same modulation frequency of f _{ x } = 0.06 mm^{−1} with the different phase offsets of 0, 2π/3, 4π/3, and the diffuse reflectance images were captured with the CCD camera in full 512 × 512 resolution. Profile corrections of the reflectance intensity using a multiheight calibration approach along with a Lambertian model were firstly conducted [35]. Then the corrected images were put into Eqs. (4) and (5) to extract both the DC and the AC modulated reflectance images, from which the MTF was calculated pixelbypixel. Finally, pixelbypixel μ _{ a }determination is achieved using the lookup table method, with the assumption of \(\mu^{\prime}_{s}\) = 1.0 mm^{−1} over the whole FOV. The reconstructed image is shown in Fig. 9.
In Fig. 9, we recognize an extremely high absorption region which is basically coincident with the tumor bulge observed in the surface height map, demonstrating the high absorption contrast of the tumor to the surrounding tissue as a result of tumor angiogenesis [36]. The circumambient high absorption region near the tumor suggests potential tumor invasion that is invisible to naked eyes, implying that the method may serve as a powerful tool for identifying tumor margins. It should be noted that up to now SFD imaging still suffers from degraded quantitation and spatial resolution, known as the partial volume effect, because the data acquired by each detector (i.e., each pixel on the CCD array) is analyzed independently of all the other detectors. As a result, in Fig. 9, the absorption coefficient of the specimen might be underestimated due to the averaging of the background, and image of the tumor target is slightly blurred due to the averaging of the neighboring volumes. Cuccia et al. have illustrated well about the lateral and depthdependent partial volume effects in SFD imaging [15]. To solve the problem, we will work on the development of an accurate SFD forward model for inhomogeneous optical medium as well as a reconstruction method that employs rigorous sensitivity function of detector in the future.
Conclusions
In summary, we have developed a novel approach based on the MTF characterization to achieve the referencefree determination of absorption coefficients. A 3D database that depicts the MTF as a function of the absorption coefficient, the reduced scattering coefficient and the spatial frequency have been established, enables the inversion methods of both the leastsquare curvefitting and the lookup table being adopted for μ _{ a }determination. Simulation results have verified the feasibility of the method as well as evaluated its performance in μ _{ a }determination. Experimental performance have been evaluated for μ _{ a } ranging from 0.01 to 0.07 mm^{−1}, and the maximum errors in μ _{ a }determination are 4.86% for \(\mu^{\prime}_{s}\) = 1.0 mm^{−1} and 7.00% for \(\mu^{\prime}_{s}\) = 2.0 mm^{−1}. The results suggest that the proposed method can be applied to the accurate determination of the bulk absorption coefficient of tissue, and better yet, in a referencefree way. We have also presented quantitative ex vivo imaging of human lung cancer in a subcutaneous xenograft mouse model, and the result indicates that this referencefree method will facilitate the clinical translation of SFD measurement toward the diagnosis, staging and prognosis of those diseases that are accompanied by significant changes in optical absorption, such as nonmelanoma skin cancer and portwine stain.
Abbreviations
 SFD:

spatial frequency domain
 MTF:

modulation transfer function
 DC:

direct current
 AC:

alternating current
 LSF:

line spread function
 1D:

onedimension
 STF:

system transfer function
 MC:

Monte Carlo
 PSF:

point spread function
 FOV:

field of view
 LED:

lightemitting diode
 DMD:

digital micromirror device
 CCD:

charge coupled device
 ETR:

errortolerance range
References
 1.
O’Sullivan TD, Cerussi AE, Cuccia DJ, Tromberg BJ. Diffuse optical imaging using spatially and temporally modulated light. J Biomed Opt. 2012;17:071311.
 2.
Cuccia DJ, Bevilacqua F, Durkin AJ, Tromberg BJ. Modulated imaging: quantitative analysis and tomography of turbid media in the spatialfrequency domain. Opt Lett. 2005;30:1354–6.
 3.
Erickson TA, Mazhar A, Cuccia DA, Durkin AJ, Tunnell JW. Lookuptable method for imaging optical properties with structured illumination beyond the diffusion theory. J Biomed Opt. 2010;15:036013.
 4.
Kanick SC, McCiatchy DM 3rd, Krishnaswarny V, Elliott JT, Paulsen KD, Pogue BW. Subdiffusive scattering parameter maps recovered using widefield highfrequency structured light imaging. Biomed Opt Express. 2014;5:3376–90.
 5.
McClatchy ШDM, Rizzo EJ, Wells WA, Cheney PP, Hwang JC, Paulsen KD, Pogue BW, Kanick SC. Widefield quantitative imaging of tissue microstructure using subdiffuse spatial frequency domain imaging. Optica. 2016;3:613–21.
 6.
Rohrbach DJ, Zeitouni NC, Muffoletto D, Saager R, Tromberg BJ, Sunar U. Characterization of nonmelanoma skin cancer for light therapy using spatial frequency domain imaging. Biomed Opt Express. 2015;6:1761–6.
 7.
Saager RB, Cuccia DJ, Saggese S, Kelly KM, Durkin AJ. A Light emitting diode (LED) based spatial frequency domain imaging system for optimization of photodynamic therapy of nonmelanoma skin cancer: quantitative reflectance imaging. Laser Surg Med. 2013;45:207–15.
 8.
Mazhar A, Sharif SA, Cuccia DJ, Nelson SJ, Kelly KM, Durkin AJ. Spatial frequency domain imaging of port wine stain biochemical composition in response to laser therapy: a pilot study. Laser Surg Med. 2012;44:611–21.
 9.
Konecky SD, Rice T, Tromberg BJ, Owen CM, Valdés PA, Roberts DW, Kolste K, Leblond F, Paulsen KD, Wilson BC. Spatial frequency domain tomography of protoporphyrin IX fluorescence in preclinical glioma models. J Biomed Opt. 2012;17:056008.
 10.
Sunar U, Rohrbach DJ, Morgan J, Zeitouni N, Henderson BW. Quantifiation of PpIX concentration in basal cell carcinoma and squamous cell carcinoma models using spatial frequency domain imaging. Biomed Opt Express. 2013;4:531–7.
 11.
Ponticorvo A, Taydas E, Mazhar A, Scholz T, Kim HS, Rimler J, Evans GR, Cuccia DJ, Durkin AJ. Quantitative assessment of partial vascular occlusion in a swine pedicle flap model using spatial frequency domain imaging. Biomed Opt Express. 2013;4:298–306.
 12.
Mazhar A, Dell S, Cuccia DA, Gioux S, Durkin AJ, Frangioni JV, Tromberg BJ. Wavelength optimization for rapid chromophore mapping using spatial frequency domain imaging. J Biomed Opt. 2010;15:061716.
 13.
Doornbos RM, Lang R, Aalders MC, Cross FW, Sterenborg HJ. The determination of in vivo human tissue optical properties and absolute chromophore concentrations using spatially resolve diffuse reflectance spectroscopy. Phys Med Biol. 1999;44:967–81.
 14.
Dögnitz N, Wagnières G. Determination of tissue optical properties by steadystate spatial frequencydomain reflectometry. Lasers Med Sci. 1998;13:55–65.
 15.
Cuccia DJ, Bevilacqua F, Durkin AJ, Ayers FR, Tromberg BJ. Quantitation and mapping of tissue optical properties using modulated imaging. J Biomed Opt. 2009;14:024012.
 16.
Vervandier J, Gioux S. Single snapshot imaging of optical properties. Biomed Opt Express. 2013;4:2938–44.
 17.
Nadeau KP, Durkin AJ, Tromberg BJ. Advanced demodulation technique for the extraction of tissue optical properties and structural orientation contrast in the spatial frequency domain. J Biomed Opt. 2014;19:056013.
 18.
Bodenschatz N, Krauter P, Liemert A, Wiest J, Kienle A. Modelbased analysis on the influence of spatial frequency selection in spatial frequency domain imaging. Appl Opt. 2015;54:6725–31.
 19.
Weber JR, Cuccia DJ, Durkin AJ, Tromberg BJ. Noncontact imaging of absorption and scattering in layered tissue using spatially modulated structured light. J Appl Phys. 2009;105:102028.
 20.
Nadeau KP, Rice TB, Durkin AJ, Tromberg BJ. Multifrequency synthesis and extraction using square wave projection patterns for quantitative tissue imaging. J Biomed Opt. 2015;20:116005.
 21.
Nandy S, Mostafa A, Kumavor PD, Sanders M, Brewer M, Zhu Q. Characterizing optical properties and spatial heterogeneity of human ovarian tissue using spatial frequency domain imaging. J Biomed Opt. 2016;21:101402.
 22.
Tabassum S, Zhao Y, Istfan R, Wu J, Waxman DJ, Roblyer D. Feasibility of spatial frequency domain imaging (SFDI) for optically characterizing a preclinical oncology model. Biomed Opt Express. 2016;7:4154–70.
 23.
Baruch D, Abookasis D. Multimodal optical setup based on spectrometer and cameras combination for biological tissue characterization with spatially modulated illumination. J Biomed Opt. 2017;22:046007.
 24.
Sandbymoller J, Poulsen T, Wulf HC. Epidermal thickness at different body sites: relationship to age, gender, pigmentation, blood content, skin type and smoking habit. Acta Derm Venereol. 2003;83:410–3.
 25.
Bhandari A, Hamre B, Frette O, Stamnes K, Stamnes JJ. Modeling optical properties of human skin using Mie theory for particles with different size distributions and refractive indices. Opt Express. 2011;15:14549–67.
 26.
Bashkatov AN, Genina EA, Tuchin VV. Optical properties of skin, subcutaneous, and muscle tissues: a review. J Innov Opt Health Sci. 2011;4:9–38.
 27.
Krishnaswamy V, Elliott JT, McClatchy DM, Barth RJ Jr, Wells WA, Pogue BW, Paulsen KD. Structured light scatteroscopy. J Biomed Opt. 2014;19:070504.
 28.
Yücel MA, Aasted CM, Petkov MP, Borsook D, Boas DA, Becerra L. Specificity of hemodynamic brain responses to painful stimuli: a cunctional nearinfrared spectroscopy study. Sci Rep. 2015;5:9469.
 29.
Foschum F, Kienle A. Broadband absorption spectroscopy of turbid media using a dual step steadystate method. J Biomed Opt. 2012;17:037009.
 30.
Bashkatov AN, Genina EA, Kochubey VI, Tuchin VV. Optical properties of human skin, subcutaneous and mucous tissues in the wavelength range from 400 to 2000 nm. J Phys D Appl Phys. 2005;38:2543–55.
 31.
Jacques SL. Optical properties of biological tissues: a review. Phys Med Biol. 2013;58:R37–61.
 32.
Wang L, Jacques SL, Zheng L. MCML—Monte Carlo modeling of light transport in multilayered tissues. Comput Methods Progr Biomed. 1995;47:131–46.
 33.
van Staveren HJ, Moes CJ, van Marle J, Prahl SA, van Gemert MJ. Light scattering in Intralipid10% in the wavelength range of 400–1100nm. Appl Opt. 1991;30:4507–14.
 34.
Gioux S, Mazhar A, Cuccia DJ, Tromberg BJ, Frangioni JV. Threedimensional surface profile intensity correction for spatially modulated imaging. J Biomed Opt. 2009;14:034045.
 35.
van de Giessen M, Angelo JP, Gioux S. Realtime, profilecorrected single snapshot imaging of optical properties. Biomed Opt Express. 2015;6:4051–62.
 36.
Heijblom M, Klaase JM, van den Engh FM, van Leeuwen TG, Steenbergen W, Manohar S. Imaging breast tumor vascularization for detection and diagnosis of breast cancer. Technol Cancer Res Treat. 2011;10:607–23.
Authors’ contributions
WC carried out the design of the algorithms, performed the simulations and drafted the manuscript, and also participated in the experimental design and operation. TL and CQ built the prototype system and performed the surface profile extraction and correction. PY and KZ contributed to the implementation of the experiments. HZ and FG directed the design of the algorithm and the experiments. All authors read and approved the final manuscript.
Acknowledgements
Not applicable.
Competing interests
The authors declare that they have no competing interests.
Availability of data and materials
The datasets used and/or analyzed during the current study are available from the corresponding author on reasonable request.
Consent for publication
Not applicable.
Ethics approval
Animal experiments were conducted under a protocol approved by the institutional ethical committee, and the procedures were in accordance with institutional guidelines.
Funding
The authors acknowledge the funding supports from the National Natural Science Foundation of China (61475116, 81371602, 61575140, 81671728, 81401453, 81571723), the Tianjin Municipal Government of China (17JCZDJC32700, 17JCQNJC12700, 15JCZDJC31800) and the “111” Project of the Ministry of Education of China (B07014).
Publisher’s Note
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Author information
Rights and permissions
Open Access This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
About this article
Cite this article
Chen, W., Zhao, H., Li, T. et al. Referencefree determination of tissue absorption coefficient by modulation transfer function characterization in spatial frequency domain. BioMed Eng OnLine 16, 100 (2017) doi:10.1186/s129380170394z
Received:
Accepted:
Published:
Keywords
 Determination of absorption coefficient
 Spatial frequency domain measurement
 Modulation transfer function
 Referencefree measurement