- Open Access
A feasibility study of altered spatial distribution of losses induced by eddy currents in body composition analysis
© Blomqvist and Sepponen; licensee BioMed Central Ltd. 2010
- Received: 18 June 2010
- Accepted: 4 November 2010
- Published: 4 November 2010
Tomographic imaging has revealed that the body mass index does not give a reliable state of overall fitness. However, high measurement costs make the tomographic imaging unsuitable for large scale studies or repeated individual use. This paper reports an experimental investigation of a new electromagnetic method and its feasibility for assessing body composition. The method is called body electrical loss analysis (BELA).
The BELA method uses a high-Q parallel resonant circuit to produce a time-varying magnetic field. The Q of the resonator changes when the sample is placed in its coil. This is caused by induced eddy currents in the sample. The new idea in the BELA method is the altered spatial distribution of the electrical losses generated by these currents. The distribution of losses is varied using different excitation frequencies. The feasibility of the method was tested using simplified phantoms. Two of these phantoms were rough estimations of human torso. One had fat in the middle of its volume and saline solution in the outer shell volume. The other had reversed conductivity distributions. The phantoms were placed in the resonator and the change in the losses was measured. Five different excitation frequencies from 100 kHz to 200 kHz were used.
The rate of loss as a function of frequency was observed to be approximately three times larger for a phantom with fat in the middle of its volume than for one with fat in its outer shell volume.
At higher frequencies the major signal contribution can be shifted toward outer shell volume. This enables probing the conductivity distribution of the subject by weighting outer structural components. The authors expect that the loss changing rate over frequency can be a potential index for body composition analysis.
- Voltage Divider
- Resonant Circuit
- Conductivity Distribution
- Body Composition Analysis
- Loss Resistance
Central (visceral) obesity has an important role in the development of type 2 diabetes . One recent study has also found that fat collection around the heart and aorta and within the liver is clearly associated with decreased heart functions . The same study also shows once again that body mass index (BMI) has limitations as a health metric.
Tomographic imaging is an accurate and reliable method to measure the visceral fat area (VFA), but it is typically slow and expensive. Thus it is not suitable for evaluating large groups of individuals. Moreover, X-ray computed tomography (CT) exposes patients to radiation. Therefore there is a need for a simple and cost-effective method to measure visceral fat accumulation [3, 4].
Waist circumference has been used as an alternative index because of its simplicity and reasonable correlation with CT. However, it is desired to replace it by a more direct measurement which would not be affected by the subcutaneous and the fat free volumes. Abdominal bioelectrical impedance analysis has been proposed as a suitable low-cost method for assessing VFA, but in spite of its potential, further exploration of the method is still needed .
This paper introduces a body electrical loss analysis (BELA) method for body composition analysis. BELA is an electromagnetic method, where the object is placed in a time-varying magnetic field. The object set in the magnetic field perturbs the field, and the magnitude of this perturbation is measured. The first work related to measurements of the perturbation of biological tissue in a magnetic field was carried out by Tarjan and McFee . Later on, magnetic fields have been used in the assessment of the composition of the human body, such as in the total body electrical conductivity (TOBEC) method . TOBEC operated on the principle that the impedance of a solenoidal coil is changed when a subject is placed into a coil. In the last decade much research has been done towards the realisation of an imaging method based on magnetic induction [8–10]. In magnetic induction tomography the excitation coils are used to induce eddy currents in the subject, and the magnetic field from these is then detected by sensing coils.
The device employed in the present study is a feasibility model designed specially for the studies of the abdominal region of an adult human subject. The sensitivity of the method to the eddy current losses arising from the sample was validated with simplified phantom tests and the results are shown below.
where ω loaded = 2πf loaded and R L is the equivalent sample resistance which loads the coil .
where is the effective conductivity (low frequency approximation) and E rms is the rms value of the electric field E .
The altered spatial distribution of the losses induced by eddy currents in the sample is a new approach to body composition analysis, and it is how the BELA method differs from the former non-imaging electromagnetic methods, such as TOBEC [17–19], electromagnetic resonance , and tissue resonant impedance monitor .
Sensor and instrumentation
The feasibility of the BELA method was tested with simplified phantoms. Four phantoms with different conductivity distributions were placed in the coil and the voltage change across the coil was measured. The measurement was performed as a trial test at five different resonance frequencies. The frequency was changed by switching a tuning capacitor and matching resistor. The number of consecutive sweeps was 16 (average factor) and the amplitude of the excitation frequency was set to 1 V. After the resonance was found and the span narrowed, the frequency increments of the sweep were 2 Hz.
Fluid conductivities used in phantoms
The four-point conductivity measurement method was used to measure the conductivities of the saline solution and sun flower oil used in the test phantoms. A known alternating current was driven through the volume filled with the fluid under test. Two electrodes with a known distance between them were placed in the fluid and the difference in the voltage between the electrodes was measured at six different frequencies from 100 kHz to 200 kHz in 20-kHz steps. A stand-alone preamplifier was used (Model 5113 PRE-AMP, Signal Recovery, United Kingdom). The electrode cell constant was 25.98 m-1.
Q of the sensor
The quality factor of the tuned coil was measured with an Agilent 4395A network analyser by using an Agilent 43961A RF Impedance Test Adapter. The measurements were made for a coil laid on a table and placed upright on the floor. The interesting bandwidth is from 100 kHz to 200 kHz, but some extra measurements were taken at higher frequencies to give an idea of the trend.
Driver amplifier stability
The stability of the output voltage of the driver amplifier was measured for every resonance frequency with different coil loadings. The coil was loaded by a loop (d = 285 mm) placed in the middle of it. The loop was closed with a resistor whose value varied from 10 Ω to 100 kΩ in one-decade steps.
To estimate the electrical loss distribution between the inner and outer parts of these measured phantoms, it is worthwhile to calculate the signal slopes for in-phase signals. The signal slopes for the fat-ns and ns-fat phantoms are approximately 0.14 mV/kHz and 0.41 mV/kHz, respectively. The signal slopes for the partially filled (air core) and entirely filled phantoms are approximately 0.47 mV/kHz and 0.73 mV/kHz, respectively. The difference between the samples and the estimated function values of linear fit (residuals) is below ± 2 mV.
Fluid conductivities used in phantoms
Q of the sensor
Driver amplifier stability
The stability of the amplifier output voltage V0 was approximately ± 1 mV for the various coil loadings. To ensure that electromagnetic interference did not play a part, the spectrum of the coil was measured from 50 kHz to 450 kHz. The highest peak of 38 dB μ V was observed at 66 kHz.
The results of this study show that the BELA method has the potential to be used in the assessment of body composition. It meets the requirement of a device suitable for medical check-ups and repeated individual use. It is non-invasive, easily transferable, fast, and inexpensive compared to tomography. The maximum current flowing in the coil is approximately 4-7 mA, depending on the resonance frequency. From this current the peak value of the magnetic field in the coil was estimated to be approximately 2.2 μ T. This is well below the ICNIRP guidelines (International Commission on Non-Ionizing Radiation Protection) for general public exposure, which is 4.6 μ T at a frequency of 200 kHz.
By analysing the results of the phantom test, the ratio of the signal slopes between ns-fat and fat-ns phantoms is approximately 2.9. This result is consistent with what was expected. The rate of loss as a function of frequency is higher for the phantom which has saline solution in its outer shell volume and fat in its middle part volume. The magnitude of the signal from the phantom entirely filled with saline solution is about 1.5 times higher than the magnitude of the signal observed from the partially filled (air core) phantom. This indicates a good penetration of magnetic field into the phantom. The magnitude of the signal slope for the phantom with an air core resembles that of the ns-fat phantom, but is slightly higher because air is an even better insulator than fat.
The change in channel Q indicates that there is no significant unwanted capacitive coupling between the coil and phantom below 150 kHz. The mean value of the signal is almost zero. However, at higher frequencies the mean value has slightly increased. A small dielectric loss may have been included as a result of capacitive coupling (recall inverted curves), although the detuning caused by the phantom was insignificant. An electrostatic shield was not used in this experiment, but it is generally essential: see, for example, .
The electrical loss resistances of 0.4-5.6 mΩ observed in this study can be compared to MRI studies. An experimental value of a 30 mΩ loss resistance was obtained for a 2-litre saline sample with an 100 mM concentration and ω 0 = 4 MHz, a = 0.13 m . By accounting for the factor of 20 in the difference of frequencies, it can be said that the loss resistances match those in this work.
Multiple frequency measurements as made in the BELA method have benefits in probing the conductivity distribution of the subject, by weighting outer structural components as the frequency increases. In body composition analysis this can be used in the assessment of fat distribution, where the single frequency measurement gives only the total conductivity of the subject, and thus indexes such as fat free mass and body water. An example use case for the BELA method can be the measurement of intra-abdominal (IA) fat, as the authors expect that the loss changing rate over frequency will be smaller in an individual who do not have significant amount of IA fat, than in an individual who has. The effect of the subcutaneous fat to the loss changing rate is expected to be minimal. Further studies are required to evaluate this in practice.
The feasibility of a new electromagnetic method for assessing body composition was introduced. The potential of the method was demonstrated with phantom tests. The results indicate that using different excitation frequencies, it will be possible to estimate the electrical loss distribution between the inner and outer parts of a sample consisting of two differently conductive volumes.
The authors wish to thank Tekes - the Finnish Funding Agency for Technology and Innovation - for funding this research. We also want to thank P. Eskelinen and H. Rimminen for comments and valuable discussion.
- Boyko EJ, Fujimoto WY, Loenetti DL, Newell-Morris L: Visceral adiposity and risk of type 2 diabetes: a prospective study among Japanese Americans. Diabetes Care 2000, 23: 465–471. 10.2337/diacare.23.4.465View ArticleGoogle Scholar
- Ruberg FL, Chen Z, Hua N, Bigornia S, Guo Z, Hal-lock K, Jara H, LaValley M, Phinikaridou A, Qiao Y, Viereck J, Apovian CM, Hamilton JA: The relationship of ectopic lipid accumulation to cardiac and vascular function in obesity and metabolic syndrome. Obesity advance online publication 2009.Google Scholar
- Shiga T, Oshima Y, Kanai H, Hirata M, Hosoda K, Nakao K: A simple measurement method of visceral fat accumulation by bioelectrical impedance analysis. IFMBE Proceedings 2007, 17: 687–690. full_textView ArticleGoogle Scholar
- Ryo M, Maeda K, Onda T, Katashima M, Okumiya A, Nishida M, Yamaguchi T, Funahashi T, Matsuzawa Y, Nakamura T, Shimomura I: A new simple method for the measurement of visceral fat accumulation by bioelectrical impedance. Diabetes Care 2005, 28: 451–453. 10.2337/diacare.28.2.451View ArticleGoogle Scholar
- Watson S, Blundell HL, Evans WD, Griffiths H, Newcombe RG, Rees DA: Can abdominal bioelectrical impedance refine the determination of visceral fat from waist circumference? Physiol Meas 2009, 30: N53-N58. 10.1088/0967-3334/30/7/N01View ArticleGoogle Scholar
- Tarjan PP, McFee R: Electrodeless measurements of the effective resistivity of the human torso and head by magnetic induction. IEEE Transactions on Bio-Medical Engineering 1968, 4: 266–278. 10.1109/TBME.1968.4502577View ArticleGoogle Scholar
- Harrison GG, Van Itallie TB: Estimation of body composition: a new approach based on electromagnetic principles. Am J Clin Nutr 1982, 35: 1176–1179.Google Scholar
- Griffiths H, Stewart WR, Gough W: Magnetic induction tomography: a measuring system for biological tissues. Ann NY Acad Sci 1999, 873: 335–345. 10.1111/j.1749-6632.1999.tb09481.xView ArticleGoogle Scholar
- Rosell-Ferrer J, Merwa R, Brunner P, Scharfetter H: A multifrequency magnetic induction tomography system using planar gradiometers: data collection and calibration. Physiol Meas 2006, 27: S271-S280. 10.1088/0967-3334/27/5/S23View ArticleGoogle Scholar
- Vauhkonen M, Hamsch M, Igney CH: A measurement system and imaging reconstruction in magnetic induction tomography. Physiol Meas 2008, 29: S445-S454. 10.1088/0967-3334/29/6/S37View ArticleGoogle Scholar
- Furse C, Christensen DA, Durney CH: Electromagnetics in medicine: Today and tomorrow. In Basic Introduction to Bioelectromagnetics. 2nd edition. Broken Sound Parkway, NW: CRC Press; 2009:235–237.Google Scholar
- Hambley AR: Chapter 11 Active filters and tuned circuits. In Electronics. Upper Saddle River, NJ: Prentice Hall; 1999:754–755.Google Scholar
- Vesselle H, Collin RE: The signal-to-noise ratio of nuclear magnetic resonance surface coils and application to a lossy dielectric cylinder model-Part II: the case of cylindrical window coils. IEEE Transactions on Biomedical Engineering 1995, 42: 507–520. 10.1109/10.376155View ArticleGoogle Scholar
- Furse C, Christensen DA, Durney CH: Appendix A: Electrical properties of the human body. In Basic Introduction to Bioelectromagnetics. 2nd edition. Broken Sound Parkway, NW: CRC Press; 2009:254.Google Scholar
- Furse C, Christensen DA, Durney CH: Electric and magnetic fields: Basic concepts. In Basic Introduction to Bioelectromagnetics. 2nd edition. Broken Sound Parkway, NW: CRC Press; 2009:35.Google Scholar
- Furse C, Christensen DA, Durney CH: EM behavior when the wavelength is large compared to the object size. In Basic Introduction to Bioelectromagnetics. 2nd edition. Broken Sound Parkway, NW: CRC Press; 2009:74.Google Scholar
- Baumgartner RN: Electrical impedance and total body electrical conductivity. In Human Body Composition. Edited by: Roche AF, Heymsfield SB, Lohman TG. Champaing, IL: Human Kinetics; 1996:94–107.Google Scholar
- Ellis KJ: Methods of measuring body composition. In Neonatal Nutrition and Metabolism. 2nd edition. Edited by: Thureen PJ, Hay WW. Cambridge: Cambridge University Press; 2006:603–608.Google Scholar
- Wojciech P, Pawel K, January W, Wojciech F: New way of body composition analysis using total body electrical conductivity method. Rev Sci Instrum 1995, 66: 3037–3041. 10.1063/1.1145525View ArticleGoogle Scholar
- Sutcliffe JF, Smith MA: Body composition assessed by electrical conductivity. In Body Composition Techniques in Health and Disease. Edited by: Davies PSW, J CT. Cambridge: Cambridge University Press; 1995:63–70.Google Scholar
- Chadwick PJ, Saunders NH: TRIM: An electromagnetic body composition analyser. In Advances in In Vivo Body Composition Studies. Edited by: Yasumura S, Harrison JE, Mcneill KG. New York: Plenum Press; 1990:387–390.View ArticleGoogle Scholar
- Redpath TW, Hutchison JMS: Estimating patient dielectric losses in NMR imagers. Magnetic Resonance Imaging 1984, 2: 295–300. 10.1016/0730-725X(84)90195-4View ArticleGoogle Scholar
- Hoult DI, Lauterbur PC: The sensitivity of the zeugmatographic experiment involving human samples. Journal of Magnetic Resonance 1979, 34: 425–433.Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.