- Open Access
Magnetic resonance flow velocity and temperature mapping of a shape memory polymer foam device
© Small et al; licensee BioMed Central Ltd. 2009
Received: 10 July 2009
Accepted: 31 December 2009
Published: 31 December 2009
Interventional medical devices based on thermally responsive shape memory polymer (SMP) are under development to treat stroke victims. The goals of these catheter-delivered devices include re-establishing blood flow in occluded arteries and preventing aneurysm rupture. Because these devices alter the hemodynamics and dissipate thermal energy during the therapeutic procedure, a first step in the device development process is to investigate fluid velocity and temperature changes following device deployment.
A laser-heated SMP foam device was deployed in a simplified in vitro vascular model. Magnetic resonance imaging (MRI) techniques were used to assess the fluid dynamics and thermal changes associated with device deployment.
Spatial maps of the steady-state fluid velocity and temperature change inside and outside the laser-heated SMP foam device were acquired.
Though non-physiological conditions were used in this initial study, the utility of MRI in the development of a thermally-activated SMP foam device has been demonstrated.
Shape memory polymers (SMPs) are a class of polymeric materials that can be fabricated into a primary shape, deformed into a stable secondary shape, and controllably actuated to recover the primary shape. The basis for the shape memory effect has been previously described in detail . Although there is wide chemical variation in these materials, they can be grouped into categories with high physical similarity based on the method of actuation, which can be achieved thermally, through photo-induced reaction, or by introduction of an external plasticizer . For SMPs that are actuated thermally, such as those in the present work, raising the temperature of the polymer above its characteristic glass transition temperature (Tg) results in a decrease in the elastic modulus from that of the glassy state (~109 Pa) to that of an elastomer (~106 to 107 Pa)  as the primary shape is recovered. Upon cooling, the original modulus is nearly completely recovered and the primary form is stabilized .
Encouraged by the shape memory behavior and biocompatibility [5, 6], many biomedical applications for SMP-based active devices have emerged . In particular, researchers are developing various interventional medical devices based on thermally responsive SMP. Such catheter-delivered devices include expandable stents [7, 8], microactuators for retrieving blood clots in ischemic stroke patients [9, 10], and embolic coils  and foams [12, 13] for filling aneurysms.
When the Tg of the SMP is above body temperature (37°C), an external heating mechanism such as laser (photothermal) [9, 14] or electroresistive [10, 14] heating is needed. Safe and effective device actuation requires limiting the thermal impact to the surrounding blood and tissue, posing a key challenge in SMP interventional device development. Another development consideration relevant for implantable devices (e.g., stent or embolic device) is the effect of the deployed device on the blood flow. Since changes in the hemodynamics and temperature induced by the intervention ultimately govern its safety and efficacy, there is a need to understand these changes and their physiological impact. A first step in the device development process is to investigate fluid velocity and temperature changes following device deployment in a simplified in vitro model. Though the results of such an investigation do not necessarily provide a direct assessment of the physiological impact in an actual clinical procedure, they may be used to modify device properties (e.g., foam density), adjust heating parameters (e.g., laser power), and validate computational models which can be extended to simulate physiological conditions.
The non-invasive methods of nuclear magnetic resonance (NMR) and magnetic resonance imaging (MRI) have been extensively used for chemical, material, biological, and medical applications. In MRI methods, a magnetic field gradient is used in conjunction with an NMR experiment to spatially encode spectral signatures based on numerous contrast parameters. These signatures may include structure (chemical shift), dynamics (relaxation times), or velocity (diffusion and flow). Spatial maps of fluid flow and temperature obtained by MRI can provide insight into the impact of the intervention.
Our team has previously reported the use of various tools to study the fluid and thermal dynamics associated with the deployment of SMP devices under development. These tools include computational fluid dynamics models , flow visualization and particle image velocimetry , and contact temperature measurement . The primary aim of this study is to demonstrate the utility of non-invasive MRI techniques to aid the early stage development of an SMP embolic foam device. We describe the use of established MRI methods to quantify the steady-state fluid flow and temperature during laser heating of a generic, non-clinical SMP embolic foam device in constant water flow in a straight tube. The SMP foam device, laser heating scheme, flow system, and MRI system are described, and the flow velocity and temperature change inside and outside the foam device are reported. The benefits of non-invasive MRI for preliminary device development are discussed.
SMP Foam Device
The chemically blown foam was based on SMP developed at Lawrence Livermore National Laboratory (LLNL) comprised of hexamethylene diisocyanate, N, N, N', N'-tetrakis(2-hydroxypropyl)ethylenediamine, and triethanolamine . Dye (Epolight™ 4121, Epolin, Inc.) was added during processing to aid laser light absorption. The predominantly open cell foam had a Tg of ~45°C, dye concentration of ~900 ppm (absorption coefficient ~10 cm-1), density of 0.020 g/cc, cell size of ~200-300 μm, and calculated volumetric void fraction of 98.4%, allowing for a theoretical volume expansibility from a fully collapsed state of 60×. Different foam properties, including Tg, may be achieved by adjusting the chemical composition.
A key property of the SMPs is their ability to maintain a secondary shape that is different than the original primary shape. The secondary shape is obtained by reforming the SMP while heated above Tg and then cooling it to stabilize the new shape. For the SMP foams, the secondary shape is obtained by compressing the foam to reduce its volume (a key feature enabling catheter delivery of the device). Heating near Tg will induce shape recovery (expansion). In the case of SMP foams, the transition from the secondary to the primary shape is not sharp like a crystalline melting point; the elastic modulus falls gradually over a span of ~30°C with the nominal Tg approximately centered in the decline.
To fabricate the diffusing fiber, LLNL SMP (Tg~81°C) was cast in a teflon tube (inner diameter = 300 μm) over a 100-μm-core cleaved optical fiber. The resulting SMP rod was then media blasted with 100-μm sodium bicarbonate particles to create a diffusing surface. A ST connector was added to the proximal end of the optical fiber for coupling to the laser light source, an 810-nm continuous-wave diode laser pigtailed into a 100-μm-core silica optical fiber (Model UM7800/100/20, Unique Mode). Nearly 90% of the light was emitted radially, with the remaining light emerging from the distal end of the diffuser. The SMP formulation was specifically designed to be optically transparent at the laser wavelength (absorption coefficient = 0.01 cm-1 at 810 nm).
MRI Experimental Parameters
Images were acquired at 9.4 T with a Bruker Avance spectrometer and a Bruker Micro2.5 microimaging probe equipped with a 5-mm coil. The fluid used in all experiments was ionized water doped with a small amount (0.3 ml/L) of MRI contrast agent (Magnevist, Berlex Laboratories) so that faster repetition time (TR) could be used, which shortens the experimental scan time. Images of the area occupied by the foam were acquired using a spin echo imaging sequence with a field of view (FOV) of 2 × 2 cm, imaging matrix of 128 × 128 (156 × 156 μm pixel resolution), TR of 400 ms, echo time (TE) of 7 ms, and an experiment acquisition time of 51 s. Fluid flow was turned off to acquire these images.
Fluid flow images, depicting the three-dimensional velocity patterns, were acquired using a diffusion weighted, spin echo, phase encoding sequence. Four slices were acquired for each scan with a thickness of 2 mm and FOV = 0.5 × 0.5 cm, imaging matrix = 128 × 128 (39 × 39 μm pixel resolution), TR = 400 ms, and TE = 15.2 ms. For flow encoding, the following parameter values were used: Δt = 7 ms, δ = 1 ms, and g = 0 and 5 mT/m, where Δt is the time between the two motion encoding gradients, δ is the duration of the motion encoding gradients, and g is the strength of the gradients. The resulting scan time was 2 min and 8 s. After subtracting the phase images from the two different g values, the phase difference, Δφ, was converted to velocity, V, using the formula V = Δφ/(δ Δtγg), where γ is the gyromagnetic ratio for protons (42.58 MHz/T) . In order to fully characterize the flow, three velocity encoded images were obtained, one for each spatial dimension, using the same parameters described above. Images were acquired using a fluid flow rate of 3.8 ml/min.
There exist several methods of measuring the temperature with MRI, including observing changes in spin-lattice relaxation (T1), determining the diffusion constant over a specific region, or through the proton resonance frequency (PRF). The PRF method for determining temperature with MRI is the method of choice because it is independent of tissue composition, unlike measuring temperature with MRI techniques, such as spin-lattice relaxation time and the diffusion times, which are dependent on tissue composition. PRF utilizes the average hydrogen bond strength and can be determined from the phase in gradient echo strength, which is based on the chemical shift properties. The chemical shift is the separation of resonance frequencies from an arbitrary chosen resonance frequency and is caused by the shielding effects of the electronic environment of the nucleus. This means that nuclei in different chemical environments give rise to MRI signals at different frequencies. Since the chemical shift depends on temperature, changes in the strength of hydrogen bonding causes changes in chemical shifts, which can be used to measure changes in temperature. For these reasons, we used a gradient echo sequence. The gradient echo sequence was optimized by maximizing TE, TR, and gradient properties.
A gradient echo, phase encoding pulse sequence was employed to obtain spatial temperature maps. Acquisition parameters of TR = 90 ms, TE = 10 ms, and flip angle = 45° were used. Two slices were acquired during each 46-s scan with a FOV = 1 × 1 cm, imaging matrix = 128 × 128 (78 × 78 μm pixel resolution), and a slice thickness of 2 mm. Temperature maps were obtained from the phase difference between an image at a reference temperature and an image with laser heating, with fluid flow turned on for both. The relationship between phase difference, Δφ, and temperature change, ΔT, is given by ΔT = Δφ/(αγB0TE), where α is the thermal coefficient (0.01 ppm/°C in water), B0 is the magnetic field strength, and TE is the echo time of the experiment [19–21]. Temperature images were acquired with laser powers of 0, 1, 3, 5 and 7 W and a fluid flow rate of 3.8 ml/min.
Steady-state x, y, and z components of the fluid velocity (with z along the tubing in the direction of the main flow) were acquired immediately proximal, inside, and distal to the expanded SMP foam. Steady-state fluid temperature images were acquired immediately proximal to and inside the expanded SMP foam. Only measurements of the fluid were made; the MRI technique is not sensitive to temperature changes of the silicone tubing, SMP, or silica optical fiber. Due to the relatively long scan time (~2 min), no images were acquired during the initial laser-induced expansion of the SMP foam. While it may be possible to slightly increase the temporal resolution, the MRI system is not capable of imaging the continuously changing flow and temperature profiles as the foam expands.
Given the sub-physiological fluid flow rate (blood flow in the basilar artery is ~70-170 ml/min during the cardiac cycle ), non-pulsatile flow, sub-37°C water temperature, use of water instead of blood, and lack of a mock aneurysm in the flow system, the in vitro model was not intended to accurately represent actual physiological conditions. Consequently, the measured values are not necessarily expected to represent those encountered in a clinical scenario. However, under certain conditions the data may be somewhat relevant. For example, an actual clinical device could include a baffle or other means of temporarily reducing the flow (e.g., balloon catheter) to facilitate thermal actuation of the SMP foam , in which case the low flow rate used in this study could be relevant. Regardless, testing under clinically relevant conditions is a necessary future step in the device development process, but is beyond the scope of this work.
The primary objective of this study was to illustrate the use of MRI as a tool in the preliminary stage of SMP foam device development using a simplified in vitro model. In this case, fluid velocity and temperature at several laser powers under a steady flow condition were acquired non-invasively. This type of experiment can be used to enhance the device design and deployment parameters without necessarily providing clinically relevant measurements. For example, experiments performed using devices with different foam densities would provide a means of elucidating the impact of foam density on the flow and temperature. Similarly, the thermal impact of different dye concentrations in the foam can also be quantified. By repeating the experiments at different flow rates, the effect of flow rate on the temperature can be gauged; this is important, for example, for evaluating methods of temporarily reducing the flow (e.g., baffle or balloon) to facilitate thermal actuation. The information provided can be used to refine the device and deployment parameters for subsequent testing under more clinically relevant conditions. In addition, since the experimental data is free from perturbation by invasive probes, it is particularly well-suited for development or validation of computational models of fluid and thermal dynamics.
Even with the limitations of steady-state measurements of flow velocity and temperature, the results were our first snapshot of the environment in the foam. The thermal result of reaching a maximum temperature change of 4°C at 7 W is not a surprise. We have previously described that, using simulations , flowing fluids (e.g. blood or water) are efficient at cooling the heated foams and distributing the thermal loads throughout the fluid volume. To date, however, we have had little intuition for the flow development in the foam. Fig. 3 shows the development of non-axial flows that are on the order of 50% of the axial flow rates. As we modify the structure of our open celled foams, these patterns will give us insight to local permeabilities or pockets of closed cells, for example. Also, as we move to anatomical geometries, the MRI flow data will permit characterization of flow in our treatment of aneurysms. Our ultimate goal is to use the MRI imaging to study the flow of blood, with and without anti-clotting agents, to help us in our design of the foam structure and deployment devices.
Spatial maps of the steady-state fluid velocity and temperature change inside and outside a laser-heated SMP foam device in a simplified in vitro model were acquired using MRI techniques. Despite the relatively low temporal resolution (i.e., long image acquisition time) which prevented image acquisition during laser-induced expansion of the foam, the post-expansion images contain useful information from a device development standpoint, potentially enabling quantitative comparison of different device designs. The non-invasive nature of MRI allowed measurement of the temperature rise and flow without introducing probes that perturb the system; this feature is particularly useful for comparison to computational models. Though non-physiological conditions were used in this initial study, the utility of MRI in the development of a thermally-activated SMP foam device has been demonstrated.
We would like to thank Robert S. Maxwell and Sarah C. Chinn for their support. This work was performed under the auspices of the U.S. Department of Energy by Lawrence Livermore National Laboratory under Contract DE-AC52-07NA27344 and supported by the National Institutes of Health, National Institute of Biomedical Imaging and Bioengineering, Grant R01EB000462, and Lawrence Livermore National Laboratory Directed Research and Development, Grant 04-ERD-093.
- Lendlein A, Kelch S: Shape-memory polymers. Agnew Chem Int Ed 2002, 41: 2034–2057. Publisher Full Text 10.1002/1521-3773(20020617)41:12%3C;2034::AID-ANIE2034%3E;3.0.CO;2-MView ArticleGoogle Scholar
- Behl M, Lendlein A: Shape-memory polymers. Mater Today 2007, 10: 20–28. 10.1016/S1369-7021(07)70047-0View ArticleGoogle Scholar
- Wilson TS, Bearinger JP, Herberg JL, Marion JE III, Wright WJ, Evans CL, Maitland DJ: Shape memory polymers based on uniform aliphatic urethane networks. J Appl Polym Sci 2007, 106: 540–551. 10.1002/app.26593View ArticleGoogle Scholar
- Baer G, Wilson TS, Matthews DL, Maitland DJ: Shape-memory behavior of thermally stimulated polyurethane for medical applications. J Appl Polym Sci 2007, 103: 3882–3892. 10.1002/app.25567View ArticleGoogle Scholar
- Fare S, Valtulina V, Petrini P, Alessandrini E, Pietrocola G, Tanzi MC, Speziale P, Visai L: In vitro interaction of human fibroblasts and platelets with a shape-memory polyurethane. J Biomed Mater Res 2005, 73A: 1–11. 10.1002/jbm.a.30193View ArticleGoogle Scholar
- Cabanlit M, Maitland D, Wilson T, Simon S, Wun T, Gershwin ME, Water J: Polyurethane shape-memory polymers demonstrate functional biocompatibility in vitro. Macromol Biosci 2007, 7: 48–55. 10.1002/mabi.200600177View ArticleGoogle Scholar
- Wache HM, Tartakowska DJ, Hentrich A, Wagner MH: Development of a polymer stent with shape memory effect as a drug delivery system. J Mater Sci Mater Med 2003, 14: 109–112. 10.1023/A:1022007510352View ArticleGoogle Scholar
- Baer GM, Small W IV, Wilson TS, Benett WJ, Matthews DL, Hartman J, Maitland DJ: Fabrication and in vitro deployment of a laser-activated shape memory polymer vascular stent. BioMedical Engineering OnLine 2007, 6: 43. 10.1186/1475-925X-6-43View ArticleGoogle Scholar
- Small W IV, Wilson TS, Benett WJ, Loge JM, Maitland DJ: Laser-activated shape memory polymer intravascular thrombectomy device. Opt Express 2005, 13: 8204–8213. 10.1364/OPEX.13.008204View ArticleGoogle Scholar
- Hartman J, Small W IV, Wilson TS, Brock J, Buckley PR, Benett WJ, Loge JM, Maitland DJ: Embolectomy in a rabbit acute arterial occlusion model using a novel electromechanical extraction device. AJNR Am J Neuroradiol 2007, 28: 872–874.Google Scholar
- Hampikian JM, Heaton BC, Tong FC, Zhang Z, Wong CP: Mechanical and radiographic properties of a shape memory polymer composite for intracranial aneurysm coils. Mater Sci Eng C 2006, 26: 1373–1379. 10.1016/j.msec.2005.08.026View ArticleGoogle Scholar
- Metcalfe A, Desfaits AC, Salazkin I, Yahia L, Sokolowski WM, Raymond J: Cold hibernated elastic memory foams for endovascular interventions. Biomaterials 2003, 24: 491–497. 10.1016/S0142-9612(02)00362-9View ArticleGoogle Scholar
- Maitland DJ, Small W IV, Ortega JM, Buckley PR, Rodriguez J, Hartman J, Wilson TS: Prototype laser-activated shape memory polymer foam device for embolic treatment of aneurysms. J Biomed Opt 2007, 12: 030504. 10.1117/1.2743983View ArticleGoogle Scholar
- Koerner H, Price G, Pearce NA, Alexander M, Vaia RA: Remotely actuated polymer nanocomposites--stress-recovery of carbon-nanotube-filled thermoplastic elastomers. Nature Mater 2004, 3: 115–120. 10.1038/nmat1059View ArticleGoogle Scholar
- Ortega J, Maitland D, Wilson T, Tsai W, Savas O, Saloner D: Vascular dynamics of a shape memory polymer foam aneurysm treatment technique. Ann Biomed Eng 2007, 35: 1870–1884. 10.1007/s10439-007-9358-yView ArticleGoogle Scholar
- Tsai WW, Savas O, Maitland D, Ortega J, Small W, Wilson TS, Saloner D: Experimental study of the vascular dynamics of a saccular basilar aneurysm. Proceedings of the IMECE2006 ASME International Mechanical Engineering Congress and Exposition: 5–10 November 2006; Chicago 2006, 14662.Google Scholar
- Small W IV, Buckley PR, Wilson TS, Loge JM, Maitland KD, Maitland DJ: Fabrication and characterization of cylindrical light diffusers comprised of shape memory polymer. J Biomed Opt 2008, 13: 024018. 10.1117/1.2904952View ArticleGoogle Scholar
- Han S, Pierce KL, Pines A: NMR velocity mapping of gas flow around solid objects. Phys Rev E 2006, 74: 016302. 10.1103/PhysRevE.74.016302View ArticleGoogle Scholar
- Elkins CJ, Markl M, Iyengar A, Wicker R, Eaton JK: Full-field velocity and temperature measurements using magnetic resonance imaging in turbulent complex internal flows. Int J Heat Fluid Flow 2004, 25: 702–710. 10.1016/j.ijheatfluidflow.2004.05.017View ArticleGoogle Scholar
- Ishihara Y, Calderon A, Watanabe H, Okamoto K, Suzuki Y, Kuroda K, Suzuki Y: A precise and fast temperature mapping using water proton chemical shift. Magn Reson Med 1995, 34: 814–823. 10.1002/mrm.1910340606View ArticleGoogle Scholar
- Maswadi SM, Dodd SJ, Gao J, Glickman RD: Temperature mapping of laser-induced hyperthermia in an ocular phantom using magnetic resonance thermography. J Biomed Opt 2004, 9: 711–718. 10.1117/1.1751399View ArticleGoogle Scholar
- Jou LD, Quick CM, Young WL, Lawton MT, Higashida R, Martin A, Saloner D: Computational approach to quantifying hemodynamic forces in giant cerebral aneurysms. AJNR Am J Neuroradiol 2003, 24: 1804–1810.Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.