- Open Access
Movies of cellular and sub-cellular motion by digital holographic microscopy
© Mann et al; licensee BioMed Central Ltd. 2006
- Received: 11 September 2005
- Accepted: 23 March 2006
- Published: 23 March 2006
Many biological specimens, such as living cells and their intracellular components, often exhibit very little amplitude contrast, making it difficult for conventional bright field microscopes to distinguish them from their surroundings. To overcome this problem phase contrast techniques such as Zernike, Normarsky and dark-field microscopies have been developed to improve specimen visibility without chemically or physically altering them by the process of staining. These techniques have proven to be invaluable tools for studying living cells and furthering scientific understanding of fundamental cellular processes such as mitosis. However a drawback of these techniques is that direct quantitative phase imaging is not possible. Quantitative phase imaging is important because it enables determination of either the refractive index or optical thickness variations from the measured optical path length with sub-wavelength accuracy.
Digital holography is an emergent phase contrast technique that offers an excellent approach in obtaining both qualitative and quantitative phase information from the hologram. A CCD camera is used to record a hologram onto a computer and numerical methods are subsequently applied to reconstruct the hologram to enable direct access to both phase and amplitude information. Another attractive feature of digital holography is the ability to focus on multiple focal planes from a single hologram, emulating the focusing control of a conventional microscope.
A modified Mach-Zender off-axis setup in transmission is used to record and reconstruct a number of holographic amplitude and phase images of cellular and sub-cellular features.
Both cellular and sub-cellular features are imaged with sub-micron, diffraction-limited resolution. Movies of holographic amplitude and phase images of living microbes and cells are created from a series of holograms and reconstructed with numerically adjustable focus, so that the moving object can be accurately tracked with a reconstruction rate of 300ms for each hologram. The holographic movies show paramecium swimming among other microbes as well as displaying some of their intracellular processes. A time lapse movie is also shown for fibroblast cells in the process of migration.
Digital holography and movies of digital holography are seen to be useful new tools for visualization of dynamic processes in biological microscopy. Phase imaging digital holography is a promising technique in terms of the lack of coherent noise and the precision with which the optical thickness of a sample can be profiled, which can lead to images with an axial resolution of a few nanometres.
- Phase Image
- Optical Thickness
- Angular Spectrum
- Digital Holography
- Coherent Noise
Dennis Gabor invented holography in 1948 while attempting to improve upon the resolution of electron microscopy. Although at the time this new invention was a significant advance, the requirement of a coherent light source precluded its practical use. The invention of the laser and the introduction of off-axis holography provided the critical elements to make holography a practical and powerful tool for large areas of applications from metrology, data storage, optical processing, device fabrication, and even fine arts. While the use of lasers and the off-axis technique made holographic recording and reconstruction a valuable and practical technique, the conventional process of performing holography using photographic plates is time-consuming and cumbersome. Real time processing of the hologram is not feasible unless one uses photorefractives and other nonlinear optical materials.
Recently, the field of holography has been undergoing another paradigm shift by electronic image capture using CCD array cameras and digital processing of the holographic images [1, 2]. With the development of higher performance CCD and computational techniques, digital holography is fast becoming an increasingly attractive alternative to conventional film-based holography. It offers a number of significant advantages, such as simple, fast image acquisition and the availability of many powerful digital processing algorithms. By calculating the complex optical field of an image volume, the amplitude and phase of the optical field are available for direct manipulation [3–7]. There are numerous digital processing techniques for manipulating the optical field information in ways that are difficult or impossible in real space processing. For example, optical system aberration can be numerically corrected[8, 9] and holographic interferometry can be performed between remotely situated objects through telecommunication links. We have recently introduced digital interference holography, where short coherence lengths can be synthesized for optical tomography by superposition of many holograms using different wavelengths.
The applications of digital holography to microscopy are particularly advantageous because in conventional microscopy the very narrow depth of focus requires mechanical scanning of the focal plane, whereas a single hologram records the information of a three-dimensional object space. These advantages have been noted since early in the development of digital holography and applied to imaging analysis of microstructures[12, 13] and biological microscopy [14–18]. In contrast with other interferometric configurations and interference microscopy which require the phase to be measured through the process of multiple image acquisition and phase modulation, digital holography requires a single image (hologram) and no phase modifying devices in order to obtain phase information. This is a significant advantage for real-time living cell analysis.
In this paper we present the results of our recent experiments on digital holography for biological microscopy. Cellular and sub-cellular features are imaged with diffraction limited resolution of 0.5 μm and the optical thickness profile is determined from the analysis of the measured phase information. As examples to determine the resolution of the optical system as well as illustrate the unique properties of digital holography we examined a standard optics test object, the USAF 1951 resolution target and a diverse range of biological objects such as red blood cells, mouse embryo fibroblast cells, and paramecium. To show the adaptability of digital holography in dynamic imaging of living specimens and cells, movies of holographic amplitude and phase images were created from a series of holograms and reconstructed with numerically adjustable focus.
The hologram which is created by the interference between the object and reference fields, |H|2 is recorded by the camera and consists of;
|H|2 = |O|2 + |R|2 + O* R + OR* (1)
where O and R are the object and reference fields, respectively. Separate exposures and subtraction of the zero-order background |O|2 and |R|2 helps to reduce the zero order term significantly. Once |O|2 and |R|2 are removed the holographic twin image O* R + OR* terms remain. A slight angle is introduced between the object and the reference beams by tilting the beam splitter BS2 in order to separate the conjugate twin images, O* R + OR* for an off-axis hologram. The object is moved back a distance z from the focal plane of the CCD to create the out of focus image of the object.
Once the hologram has been captured the optically diffracted field is numerically propagated by use of reconstruction algorithms to calculate the optically diffracted field [19–22]. Here we briefly describe some of the commonly applied methods of reconstruction.
Huygens convolution method
In the convolution process  the reconstructed complex wave-field E(x, y) is found by
and the convolution is
= E 0 (x, y) ⊕ S H (x, y, z) (3)
= F-1[F (E 0)·F (S H )]
where S H is the Huygens PSF
The whole process requires three Fourier transforms, which are carried out using the FFT algorithm. The pixel sizes of the images reconstructed by the convolution approach are equal to that of the hologram. In order to achieve as high a lateral resolution as possible, one keeps the object-hologram distance as short as possible, but the discrete Fourier transform necessitates a minimum distance such that:
where a x is the frame size and n x is the number of pixels in a x . At too close a distance, the spatial frequency of the hologram is too low and aliasing occurs. Normally the object is placed just outside this minimum distance.
Fresnel transform method
The Fresnel transformation is the most commonly used method in holographic reconstruction. The approximation of spherical Huygens wavelet by a parabolic surface allows the calculation of the diffraction integral using a single Fourier transform. The PSF can be simplified by the Fresnel approximation as:
and the reconstructed wave-field is:
The pixel resolution Δx of the reconstructed images determined directly from the Fresnel diffraction formula will vary as a function of the reconstruction distance z as:
where N is the number of pixels and Δx 0 is the pixel width of the CCD camera. As with the Huygens convolution method there is a minimum z distance requirement set by Equation 5.
Angular spectrum method
The angular spectrum method of reconstruction has the significant advantage that it has no minimum reconstruction distance requirement as is this case for the Fresnel and convolution methods. If E 0(x 0, y 0;0) is the wave-field at plane z = 0, the angular spectrum A(k x , k y ;0) at this plane is obtained by taking the Fourier transform:
A(k x , k y ;0) = ∬E 0(x 0, y 0;0) exp[-i(k x x 0 + k y y 0)]dx 0 dy 0 (9)
where k x and k y are corresponding spatial frequencies of x and y. Fourier-domain filtering can be applied to the spectrum to block unwanted spectral terms in the hologram and select a region of interest corresponding only to the object spectrum. Subsequently the wave-field E 0(x 0, y 0;0) can be rewritten as the inverse Fourier transform of its angular spectrum,
E 0(x 0, y 0;0) = ∬A(k x , k y ;0) exp[i(k x x 0 + k y y 0)]dk x dk y (10)
The new angular spectrum at plane z, A(k x , k y ;z) is calculated from A(k x , k y ;0) as
A(k x , k y ;z) = A(k x , k y ;0) exp[ik z z] (11)
The reconstructed complex wave-field of any plane perpendicular to the propagating z axis is found by
E(x, y;z) = ∬A(k x , k y ;z) exp[i(k x x + k y y)]dk x dk y (12)
Two Fourier transforms are needed for the calculation in comparison to the one needed by the Fresnel transform. However once the field is known at any one plane, only one additional Fourier transform is needed to calculate the field at different values of z. This method allows frequency-domain spectrum filtering to be applied, which for example can be used to block or remove the disturbance of the zero order and twin image components. A signficant advantage of the angular spectrum is that there is no minimum z reconstruction distance requirement.
Δξ = 0.61λ/N.A ~ 0.81 μm. (13)
The resolution target used is a positive mask with opaque chrome film on a transparent glass plate. The physical thickness of film is given by
d = λ(Δφ/2π)/(n - n 0) (13)
where λ is the wavelength, Δφ is the phase step, and n-n 0 is the index difference between the film and air. From the measured optical path length change it is possible to determine either the optical thickness or refractive index provided knowledge of one or the other is known.
Quantitative phase imaging is particularly effective in digital holography – one only needs to plot the phase of the calculated complex optical field, as shown in Fig. 2c). In Fig. 2d) we demonstrate the use of a flood-fill phase unwrapping algorithm to remove the 2π ambiguities in the wrapped phase image and apply a linear phase offset correction to the image in Fig. 2c). The optical path length resolution was determined quantitatively by analysis of the noise level in the flat area, which corresponds to glass thickness variation of around 5 nanometers as shown in the z-profile cross section Fig. 2e) (On the film-coated bar areas, the lack of light causes larger uncertainty in phase.). Pseudocolor 3D rendering of 2d) is shown in 2f). The noise in the phase image is not dependant on the quality of the beam profile (interferometric arrangement) and is significantly less than that of the noise observed in the amplitude image Fig. 2b).
As well as obtaining both amplitude and phase information, another advantage of digital holography is the ability to calculate the optical field at any number of image planes from a single hologram, emulating the mechanical focusing control of a conventional microscope. To illustrate this focusing ability we show a sequence of twenty one images of a paramecium with image area 90 × 90 μm2 and calculated in the range of z = 50 ~ 250 μm in steps of 10 μm. The sequence of images is then composed into an AVI movie (see Additional file 1). As the focus is scanned, one observes the food vacuoles as well as the cell wall passing through the various focuses.
Digital holographic movies
A digital hologram contains information of the whole optical field in a three-dimensional image volume, which allows for calculation of the image field on any focal plane. A time series of digital holograms then provides complete four-dimensional information of the object's three-dimensional spatial images as well as the time evolution of those images. Once a movie of digital holograms are recorded, one can reconstruct the images of the object and be able to focus on any focal plane as a specimen under observation moves up and down in the image space. This is the most unique capability of digital holographic movie in contrast to conventional video microscopy, where only the images of the specific focal plane that was used in recording are preserved and the information of all the other planes is lost. This is a critical advantage especially in the microscopic recording of animated microbes that are constantly swimming in and out of a large range of focal distances. In conventional video microscopy, it is not feasible to track the focal distances of rapidly moving microbes, whereas in digital holography a series of holograms can be recorded at a fixed distance and the images are later numerically focused at leisure and one is able to track the rapidly moving microbes as needed. We have recorded a series of holograms by synchronizing the camera with the laser pulses under the control of the digital delay generator. The amplitude and phase images are calculated by the Angular Spectrum method while adjusting image distances for best focus on the object under observation. The reconstructed images are then composed into AVI files.
(Additional file 2) shows the hologram movie of a paramecium swimming around with a number of euglenas. The amplitude and phase images are reconstructed from the holograms while adjusting the image distances over a 20 μm range in order to maintain the paramecium in focus (see Addition files 3 and 4). The reconstruction rate from each recorded hologram is achieved in around 300 ms. Higher magnification hologram, amplitude, phase and phase unwrapped movies are shown in (Additional files 5, 6, 7, 8) each with an image area of 80 × 80 μm2. In order to maintain the paramecium long enough within the field of view, the paramecium is slowed down using a drop of thickening agent methylcellulose. In the amplitude movie, one observes the flickering due to celial motion in the oral groove that forces food (bacteria) into the food vacuoles. A notable phenomenon in the phase movie is the shrinking of the contractile vacuole as it pumps out water. (Additional files 9, 10, 11, 12, 13) are the hologram, amplitude, phase, phase unwrapped and 3D optical profile movies of image area 170 × 170 μm2 showing selected frames of the migration process by living mouse-embryo fibroblast cells (3T3).
We have presented experimental results that demonstrate the capabilities of digital holography for biological microscopy. A series of holograms are recorded and the images are reconstructed with numerically adjustable focus so that the moving objects can be accurately tracked. The holographic reconstruction is carried out by the Angular Spectrum method in an off-axis configuration so that the twin images and zero-order background can be subtracted out using frequency domain analysis. The lateral and longitudinal resolutions obtained are consistent with diffraction limited imaging. Digital holography and movies of digital holography are seen to be a useful new tool for biological microscopy, with noteworthy advantages over traditional microscopic techniques for biological imaging.
The speckle noise of a coherent imaging system is a significant issue for biological microscopy where there is often a large range of structural scales. For example, in the presented holographic amplitude images of paramecium the intracellular regions contain various assortments of unresolved particles, which may be food particles or other organelles. These degrade the quality of images to various degrees. On the other hand, one also notices that the phase images tend to suffer from the coherent noise to a significantly lesser degree compared to the amplitude images. We note that in this study we have presented the images with minimal post-processing, the goal being the demonstration of digital holography processes. Other than the overall brightness and contrast adjustments, we have not applied any of the numerous image enhancing techniques that are available, which can significantly improve the perceived image quality for biological applications. This will be the subject of a future study.
Phase imaging digital holography is particularly promising in terms of the lack of coherent noise and the precision with which the optical thickness can be profiled, which can lead to images of high axial resolution of a few nanometres. However the phase images are often required to be unwrapped in order to simplify their interpretation. Of particular significance is the 2π ambiguity problem in phase-imaging holography. A conventional approach is to apply one of many phase-unwrapping algorithms as we have done in this paper, but often these require both substantial user intervention and strict requirements on the level of phase noise and the phase discontinuity. They also have difficulty in correctly interpreting the phase images of rough and irregular objects. We are currently investigating multi-wavelength phase-imaging digital holography for generating movies of phase-unwrapped images by using two or more laser wavelengths for illumination of the holographic system without the addition of any noise in the image.
This work is supported in part by the National Science Foundation.
- Schnars U, Jueptner WP: Direct recording of holograms by a CCD target and numerical reconstruction. Appl Opt 1994, 33: 179–181.View ArticleGoogle Scholar
- Schnars U, Jueptner WP: Digital recording and numerical reconstruction of holograms. Meas Sci Technol 2002, 13: R85-R101. 10.1088/0957-0233/13/9/201View ArticleGoogle Scholar
- Cuche E, Bevilacqua F, Depeursinge C: Digital holography for quantitative phase-contrast imaging. Opt Lett 1999, 24: 291–293.View ArticleGoogle Scholar
- Ferraro P, De Nicola S, Coppola G, Finizio A, Alfieri D, Pierattini G: Controlling image size as a function of distance and wavelength in Fresnel-transform reconstruction of digital holograms. Opt Lett 2004, 29: 854–856. 10.1364/OL.29.000854View ArticleGoogle Scholar
- Ferraro P, Grilli S, Alfieri D, De Nicola S, Finizio A, Pierattini G, Javidi B, Coppola G, Striano V: Extended focused image in microscopy by digital holography. Opt Exp 2005, 13: 6738–6749. 10.1364/OPEX.13.006738View ArticleGoogle Scholar
- Palacios F, Ricardo J, Palacios D, Gonalves E, Valin J, De Souza R: 3D image reconstruction of transparent microscopic objects using digital holography. Opt Commun 2005, 248: 41–50. 10.1016/j.optcom.2004.11.095View ArticleGoogle Scholar
- Cuche E, Marquet P, Depeursinge C: Simultaneous amplitude-contrast and quantitative phase-contrast microscopy by numerical reconstruction of Fresnel off-axis holograms. Appl Opt 1999, 38: 6994–7001.View ArticleGoogle Scholar
- Ferraro P, De Nicola S, Finizio A, Coppola G, Grilli S, Magro C, Pierattini G: Compensation of the inherent wave front curvature in digital holographic coherent microscopy for quantitative phase-contrast imaging. Appl Opt 2003, 42: 1938–1946.View ArticleGoogle Scholar
- Stadelmaier A, Massig J: Compensation of lens aberrations in digital holography. Opt Lett 2000, 25: 1630–1632.View ArticleGoogle Scholar
- Osten W, Baumbach T, Jueptner WP: Comparative digital holography. Opt Lett 2002, 27: 1764–1766.View ArticleGoogle Scholar
- Kim MK: Tomographic three-dimensional imaging of a biological specimen using wavelength-scanning digital interference holography. Opt Exp 2000, 7: 305–310.View ArticleGoogle Scholar
- Jacquot M, Sandoz P, Tribillon G: High resolution digital holography. Opt Commun 2001, 190: 87–94. 10.1016/S0030-4018(01)01046-XView ArticleGoogle Scholar
- Xu L, Peng X, Miao J, Asundi AK: Studies of digital microscopic holography with applications to microstructure testing. Appl Opt 2001, 40: 5046–5051.View ArticleGoogle Scholar
- Haddad WS, Cullen D, Solem JC, Longworth JW, McPherson A, Boyer K, Rhodes CK: Fourier-transform holographic microscop. Appl Opt 1992, 31: 4973–4978.View ArticleGoogle Scholar
- Yamaguchi I, Kato J, Ohta S, Mizuno J: Image formation in phase-shifting digital holography and applications to microscopy. Appl Opt 2001, 40: 6177–6186.View ArticleGoogle Scholar
- Marquet P, Rappaz B, Magistretti PJ, Cuche E, Emery Y, Colomb T, Depeursinge C: Digital holographic microscopy: a noninvasive contrast imaging technique allowing quantitative visualization of living cells with subwevelength axial accuracy. Opt Lett 2005, 30: 468–470. 10.1364/OL.30.000468View ArticleGoogle Scholar
- Carl D, Kemper B, Wernicke G, Von Bally G: Parameter optimized digital holographic microscope for high resolution living cell analysis. Appl Opt 2004, 43: 6536–6544. 10.1364/AO.43.006536View ArticleGoogle Scholar
- Sun H, Player M, Watson J, Hendry D, Perkins R, Gust G, Paterson D: The use of digital/electronic holography for biological applications. J Opt A-Pure Appl Opt 2005, 7: S399-S407. 10.1088/1464-4258/7/6/022View ArticleGoogle Scholar
- Mann C, Yu L, Lo C, Kim MK: High-resolution quantitative phase-contrast microscopy by digital holography. Opt Exp 2005, 13: 8693–8698. 10.1364/OPEX.13.008693View ArticleGoogle Scholar
- Ma L, Wang H, Li Y, Jin H: Numerical reconstruction of digital holograms for three-dimensional shape measurement. J Opt A-Pure Appl Opt 2004, 6: 396–400. 10.1088/1464-4258/6/4/016View ArticleGoogle Scholar
- Grilli S, Ferraro P, De Nicola S, Finizio A, Pierattini G, Meucci R: Whole optical wavefields reconstruction by digital holography. Opt Exp 2001, 9: 294–302.View ArticleGoogle Scholar
- Aaron Dakoff, James Gass, Myung Kim K: Microscopic three- dimensional imaging by digital interference. Journal of Electronic Imaging 2003, 12: 643–647. 10.1117/1.1604784View ArticleGoogle Scholar
- Demetrakopoulos TH, Mittra R: Digital and optical reconstruction of images from suboptical diffraction patterns. Appl Opt 1974, 13: 665–670.View ArticleGoogle Scholar
- Yu L, Kim MK: Wavelength-scanning digital interference holography for tomographic 3D imaging using the angular spectrum method. Opt Lett 2005, 30: 2092. 10.1364/OL.30.002092View ArticleGoogle Scholar
- Farinas J, Verkman AS: Cell volume and plasma membrane osmotic water permeability in epithelial cell layers measured by interferemetry. Biophys J 1996, 71: 3511–3522.View ArticleGoogle Scholar
- Xu W, Jericho MH, Meinertzhagen IA, Kreuzer HJ: Digital in-line holography for biological applications. Proc Natl Acad Sci USA 2001, 98: 11301–11305. 10.1073/pnas.191361398View ArticleGoogle Scholar
- Gass J, Dakoff A, Kim MK: Phase imaging without 2π ambiguity by multiple-wavelength digital holography. Opt Lett 2003, 28: 1141–1143.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.