- Open Access
An ultra-low-power image compressor for capsule endoscope
© Lin et al; licensee BioMed Central Ltd. 2006
- Received: 02 November 2005
- Accepted: 25 February 2006
- Published: 25 February 2006
Gastrointestinal (GI) endoscopy has been popularly applied for the diagnosis of diseases of the alimentary canal including Crohn's Disease, Celiac disease and other malabsorption disorders, benign and malignant tumors of the small intestine, vascular disorders and medication related small bowel injury. The wireless capsule endoscope has been successfully utilized to diagnose diseases of the small intestine and alleviate the discomfort and pain of patients. However, the resolution of demosaicked image is still low, and some interesting spots may be unintentionally omitted. Especially, the images will be severely distorted when physicians zoom images in for detailed diagnosis. Increasing resolution may cause significant power consumption in RF transmitter; hence, image compression is necessary for saving the power dissipation of RF transmitter. To overcome this drawback, we have been developing a new capsule endoscope, called GICam.
We developed an ultra-low-power image compression processor for capsule endoscope or swallowable imaging capsules. In applications of capsule endoscopy, it is imperative to consider battery life/performance trade-offs. Applying state-of-the-art video compression techniques may significantly reduce the image bit rate by their high compression ratio, but they all require intensive computation and consume much battery power. There are many fast compression algorithms for reducing computation load; however, they may result in distortion of the original image, which is not good for use in the medical care. Thus, this paper will first simplify traditional video compression algorithms and propose a scalable compression architecture.
As the result, the developed video compressor only costs 31 K gates at 2 frames per second, consumes 14.92 mW, and reduces the video size by 75% at least.
- Celiac Disease
- Compression Ratio
- Discrete Cosine Transform
- Power Dissipation
- Image Compression
Gastrointestinal (GI) endoscopy has been popularly applied for the diagnosis of diseases of the alimentary canal including Crohn's Disease, Celiac disease and other malabsorption disorders, benign and malignant tumors of the small intestine, vascular disorders and medication related small bowel injury. There exist two classes of GI endoscopy; wired active endoscopy and wireless passive capsule endoscopy. The wired active endoscopy can enable efficient diagnosis based on real images and biopsy samples; however, it causes patients discomfort and pain to push flexible, relatively bulky cables into the digestive tube. To relief the suffering of patients, wireless passive capsule endoscopes are being developed worldwide [1–4]. The capsule moves passively through the internal GI tract with the aid of peristalsis and transmits images of the intestine wirelessly.
The scope of this paper is the design of an image compression processor for capsule endoscopes. Instead of applying existing compression standards, we developed simplified image compression specifically for capsule endoscopes. Unlike the general image compression techniques, the proposed image compression starts from raw images in the format of Bayer patterns and processes R, G1, G2, and B signals separately. Comparing with the traditional image compression, the proposed image compression is low-powered for three reasons. First, the proposed image compression does not need demosaicking, and hence saves the computing power of interpolation steps. Second, the proposed compression starts from the raw image, and does not need inner product operations for color-space transformation. Finally, the computation load of the 8-by-8 discrete cosine transform (DCT) can be reduced by the factor of 3.
The proposed image compression algorithm
Traditional image compression algorithms use the optimized quantization for YCbCr image to reduce compressed image size while the visual distortion is low. In order to quantize YCbCr image, the typical image compression requires two preprocessing steps that are demosaicking and the color space transformation. However, the demosaicking step requires weighted sums for color interpolation and the color space transformation requires calculation of inner products. From the view point of GICam, it is not worth it to dissipate power for both preprocessing steps as long as the compression quality and ratio are acceptable. The measure of compression quality is the peak signal-to-noise ratio (PSNR). The calculation of PSNR is formulated as Eq. (1):
Where MSE is the mean square error of decompressed image. The compression ratio (CR) is defined as the ratio of the raw image size to the compressed image size. The measure of the compression ratio is the compression rate. The formula of the compression rate is calculated by Eq. (2):
compression rate = (1-CR-1) × 100% (2)
In GICam, the Lempel-Ziv (LZ) coding  is employed for the entropy coding. The reason why we adopted the LZ coding as the entropy coding is that the LZ encoding does not need look-up tables and complex computation. Thus, the LZ encoding can consume less power and use smaller silicon size than the other candidates, such as the Huffman encoding and the arithmetic coding.
Architecture design and implementation of GIcam image compressor
This paper presents an ultra-low-power image compression processor for capsule endoscope or swallowable imaging capsules. In applications of capsule endoscopy, it is imperative to consider battery life/performance trade-offs. Instead of applying state-of-the-art video compression techniques, we propose an RGB-based compression algorithm in which the memory size and computational load can be significantly reduced. We first simplified traditional video compression algorithms by removing the color-space transformation. As shown in the result, the developed video compressor only costs 31 K gates at 2 frames per second, consumes 14.92 mW, and reduces the video size by 75% at least.
This work was supported in part by Chung-Shan Institute of Science and Technology, Taiwan, under the project BV94G10P and the National Science Council, R.O.C., under the grant number NSC 94-2220-E-009-023. The authors would like to thank National Chip Implementation Center(CIC) for technical support.
- Gong F, Swain P, Mills T: Wireless Endoscopy. Gastrointestinal Endoscopy 2000, 51: 725–729.View ArticleGoogle Scholar
- Park HJ, Nam HW, Song BS, Choi JL, Choi HC, Park JC, Kim MN, Lee JT, Cho JH: Design of Bi-directional And Multi-Channel Miniaturized Telemetry Module for Wireless Endoscopy. Proceedings of the 2nd Annual International IEEE-EMBS Special Topic Conference on Microtechnologies in Medicine and Biology 2002, 273–276.View ArticleGoogle Scholar
- Sendoh M, Ishiyama K, Arai K-I: Fabrication of Magnetic Actuator for Use In A Capsule Endoscope. IEEE Trans On Magnetics 2003, 39: 3232–3234.View ArticleGoogle Scholar
- Louis Phee , Dino Accoto , Arianna Menciassi* , Cesare Stefanini , Carrozza Maria, Paolo Dario : Analysis And Development of Locomotion Devices For The Gastrointestinal Tract. IEEE Trans On Biomedical Engineering 2002, 49: 613–616.View ArticleGoogle Scholar
- Peterson HA, Peng H, Morgan JH, Pennebaker WB: Quantization of Color Image Components In The DCT Domain. SPIE, Human Vision, Visual Processing, and Digital Display II 2002, 1453: 210–222.View ArticleGoogle Scholar
- Ziv J, Lempel A: A Universal Algorithm for Sequential Data Compression. IEEE Trans On Information Theory 1977, 23: 337–343.MathSciNetView ArticleGoogle Scholar
- Hwang SA, Wu CW: Unified VLSI systolic array design for LZ data compression. IEEE Trans On VLSI 2001, 9: 489–498.View ArticleGoogle Scholar
- Fu M, Jullien GA, Dimitrov* VS, Ahmadi M: A Low-Power DCT IP Code Based on 2D Algebraic Integer Encoding. IEEE International Symposium on Circuit and Systems 2004, 2: 765–768.Google Scholar
- Kim CY: Compression of color medical images in gastrointestinal endoscopy: a review. Medinfo 1998,9(Pt 2):1046–50.Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.