- Open Access
Review and standardization of cell phone exposure calculations using the SAM phantom and anatomically correct head models
© Beard and Kainz; licensee BioMed Central Ltd. 2004
- Received: 21 May 2004
- Accepted: 13 October 2004
- Published: 13 October 2004
We reviewed articles using computational RF dosimetry to compare the Specific Anthropomorphic Mannequin (SAM) to anatomically correct models of the human head. Published conclusions based on such comparisons have varied widely. We looked for reasons that might cause apparently similar comparisons to produce dissimilar results. We also looked at the information needed to adequately compare the results of computational RF dosimetry studies. We concluded studies were not comparable because of differences in definitions, models, and methodology. Therefore we propose a protocol, developed by an IEEE standards group, as an initial step in alleviating this problem. The protocol calls for a benchmark validation study comparing the SAM phantom to two anatomically correct models of the human head. It also establishes common definitions and reporting requirements that will increase the comparability of all computational RF dosimetry studies of the human head.
- Cell Phone
- Human Head
- Correct Model
- Finite Difference Time Domain
- Head Model
Cell phone safety remains a topic of broad public concern that attracts frequent media attention. This attention is focused on two areas of scientific controversy concerning cell phone safety. The first area is that of non-thermal biological effects. The existence of these effects is an important open question, but it is not the topic of this paper. However, if these effects exist, their manifestation will certainly be related to the amount of RF energy deposited in the tissue – RF dosimetry . The second area of controversy, and the topic of this paper, is that of RF dosimetry, specifically computational RF dosimetry. Simply put, this is a computer simulation that estimates the deposition of RF energy, the specific absorption rate (SAR), in the head of a user. Because live human heads can not be safely instrumented for these measurements, computational RF dosimetry provides the best estimate of SAR in actual human heads. For this same reason, compliance testing is done with phantom heads.
The phantom head that is now the world-wide standard for compliance testing is the Specific Anthropomorphic Mannequin (SAM). SAM was developed by members of IEEE Standards Coordinating Committee 34, SubCommittee 2, Working Group 1 (SCC34/SC2/WG1). This working group was created to develop recommended practices for determining SAR in the head via measurement techniques . SAM has also been adopted by the European Committee for Electrical Standardization (CENELEC) , the International Electrotechnical Commission , Association of Radio Industries and Businesses , and Federal Communications Commission .
SAM is a lossless plastic shell and ear spacer. Because current technology does not allow reliable measurement of SAR in small complex structures, like a simulated pinna, SCC34/SC2 chose to use a lossless ear spacer on SAM to maximize the energy reaching the head and minimize measurement uncertainty. SAM's dimensions were taken from the 90th-percentile anthropometric data corresponding to the adult male head as tabulated by the US Army . The SAM shell is filled with a homogeneous fluid having the average electrical properties of head tissue at the test frequency.
A primary design goal for SAM was that, "SAM shall produce a conservative SAR for a significant majority of persons during normal use of wireless handsets" . To test whether this goal has been met, investigators have used computational RF dosimetry to compare the SAR in SAM to that in anatomically correct models of the human head. These anatomically correct head models are commonly derived from MRI scans. Each two-dimensional scan must be analyzed to identify individual tissue types. The two-dimensional scans must then be merged into a three-dimensional model that maintains smooth boundaries between tissue types [8, 9]. Some investigators have found that SAM underestimates SAR in adults and children by a factor of two or more . Other investigators have found that SAM overestimates SAR in both adults and children [11, 12]. These contradictory findings produce confusion on the part of the public and regulatory agencies, and call the validity of computational RF dosimetry into question. While the published results of computational RF dosimetry comparing the SAM to anatomically correct models appear contradictory, a close examination of the work reveals that there are several procedural and reporting problems that may well account for the discrepancies in results.
The groups headed by Gandhi and Kuster are not the only ones pursuing computational RF dosimetry using anatomically correct models of the human head [13–25]. Not all of these studies included SAM but, to various extents, all evidenced the same procedural and reporting problems that make comparison of results difficult.
2.1 Treatment of the pinna
The first, and the most significant of these problems, is the treatment of the external ear (pinna). Specifically, the problem is determining whether the pinna may, or may not, be considered as part of the 1- or 10-gram SAR averaging volumes. When considering SAR averaging volumes the head and the pinna should be viewed as mutually exclusive, in-other-words the pinna is not part of the head but it is attached to the head. Some investigators have chosen to treat the pinna in accordance with IEEE Std C95.1-1999  and the ICNIRP Guidelines . These standards do not consider the pinna to be an extremity. This means the pinna is subject to the same exposure limit, for peak spatial SAR, as the head. Investigators that refer to these standards include pinna tissue in the 1- or 10-gram averaging volumes used to compute SAR in anatomically correct models. Because the pinna is usually the tissue closest to the feed-point of the cell phone antenna the highest point SAR values are usually found in the pinna; consequently, averaging volumes that include pinna tissue will produce higher SAR.
SAR limits SAR limits from three different standards for extremities and other tissues. These limits are for exposure of the general public in an uncontrolled environment.
4 W/kg over 10 g
4 W/kg over 10 g
4 W/kg over 10 g
2 W/kg over 10 g
1.6 W/kg over 1 g
2 W/kg over 10 g
When comparing published results it is often difficult, or impossible, to determine whether head tissue SAR values are based on averaging volumes that include or exclude the pinna. In fact, some papers make no mention of how the pinna was treated. Although head tissue SAR is the major focus of attention, papers that consider the pinna as an extremity can not simply ignore its existence, the pinna must still meet the higher spatial peak SAR for extremities.
Another part of the problem dealing with the treatment of the pinna is simply determining what tissue constitutes the pinna. The IEEE defines the pinna as, the largely cartilaginous projecting portion of the outer ear consisting of the helix, lobule, and anti-helix . Unfortunately these anatomical structures vary with each individual and their boundaries are subjective. Consequently, when excluding the pinna some investigators have excluded considerably more or less tissue than others. Because the pinna contains high SAR values, excluding or including tissue near the pinna from the averaging volume, markedly changes the peak spatial 1- or 10-gram average.
The second problem area is the lack of common models. The only computer models that are common to all the computational RF dosimetry studies are the SAM and the Visible Human Male. The anatomic data for the Visible Human Male originated at the National Institutes of Health but many groups and individuals lent a hand in converting it into a computational model. While a few investigators have different models the only ones that can be compared across all the published results are SAM and the Visible Human. This also means that the only repeatable comparison that can be made is between the SAM and the Visible Human. It seems obvious that one can neither prove nor disprove that SAM produces a SAR greater than the maximum local SAR induced in humans for a significant majority of persons during normal use of wireless handsets, when there is only one anatomically correct model available for comparison.
Although not a major problem, it is still true that dielectric properties and names of tissue types in anatomically correct models have varied between investigators.
Of course the head model is only half of any computational RF dosimetry study, the model of the RF source is the other half. The only common source that has been used in several published studies is a dipole [15, 28–30]. Simulated cell phones have varied in size, shape, antenna type, antenna length, and sophistication. Like the anatomical head models there are some very realistic models of cell phones in use but they are either proprietary or too expensive for widespread use.
2.4 Finite Difference Time Domain (FDTD) considerations
2.4.1 Rotation artifacts
Usual practice is to align a monopole cell phone antenna with the FDTD grid to avoid the stairstep effect. The head model is then rotated to the correct position relative to the cell phone. After rotation, the voxelized model must be remeshed to align the voxels with the FDTD grid. This is not a trivial task and algorithms to perform remeshing are constantly being improved.
Grooving has not been observed with all FDTD software and even when it has been seen it has not occurred with all models. Researchers should routinely examine their models after rotation to insure grooving is not a problem.
2.4.2 SAR Calculations
Because the FDTD method calculates the electric fields at the voxel edges, the X, Y and Z-directed power components associated with a voxel are defined in different spatial locations. These components must be combined to calculate SAR in the voxel. There are three possible approaches to calculate the SAR: the 3-, 6-, and 12-field components approaches. The 12-field components approach is the most complicated but it is also the most accurate and the most appropriate from the mathematical point of view . The 12-field components approach correctly places all E field components in the center of the voxel using linear interpolation. Therefore, the power distribution is now defined at the same location as the tissue mass. For these reasons the 12-field components approach is preferred by IEEE 1529 . However, the actual approach used to calculate SAR in the FDTD voxels is usually not reported.
After the SAR in every voxel is determined multiple voxels must be combined to compute the 1- or 10-gram SAR spatial averaging volumes. These normally cubic volumes become difficult to construct at the surface of a model or when the volume is constrained to a particular tissue type. The particular algorithm used to construct these volumes can influence the resultant 1- or 10-gram SAR values. However, the actual algorithm used to construct the spatial averaging volumes is usually not reported.
2.5 Reporting results
When the cell phone model is placed next to the SAM or anatomically correct model, it changes the cell phone's antenna feed-point impedance. The antenna feed-point impedance (Z), feed-point current (I) and net input power (Pnet) are related by
Because net power and feed-point current are usually not initial conditions in FDTD simulations, different feed-point impedances will produce different results for net power, feed-point current, and SAR. If different head models produce the same feed-point impedance this would not be a concern; however, several studies [16, 17] have shown that the feed-point impedance depends on the head model, the size of the head next to the mobile phone and the mobile phone model itself.
To compare SAM with various anatomic models it is necessary to assume the same cell phone model at the same emission level for all simulations. Typically, for a given simulation, the SAR is normalized by feed-point current or net power. The normalized value is then multiplied by the feed-point current or net power level chosen for comparison. Commonly SAR is compared for net input power levels of 125 mW, 600 mW, or 1 W or for the corresponding feed-point current assuming a 50 ohm feed-point impedance.
Some investigators have chosen to scale their results to net power while others have used feed-point current. Unfortunately the choice of scaling is frequently omitted and the feed-point impedance is almost never reported making it impossible to compare differently scaled results.
To address the controversy and its underlying problems the Protocol for the Computational Comparison of the SAM Phantom to Anatomically Correct Models of the Human Head was developed by IEEE Standards Coordinating Committee 34, SubCommittee 2, Working Group 2 (SCC34/SC2/WG2). This working group was created to develop recommended practices for determining SAR in the head via computational techniques . This standard is still in draft.
The protocol has two parts; a benchmark validation study; and a set of common definitions, models, and reporting requirements. The benchmark validation study is underway with fifteen participants. All participants should finish the study by mid-2004 and the results should be published by early 2005. Hopefully the common definitions, models, and reporting requirements will be used in future investigations making comparison of results easier.
3.1 Treatment of the pinna
The protocol asks all participants to report peak spatial SAR for averaging volumes that both include and exclude the pinna. The voxels comprising the pinna in the provided anatomic models are flagged so all participants will conform to one definition of the pinna. The pinna voxels are flagged by prefixing the standard tissue type with pinna-; such as pinna-skin, pinna-cartilage, and pinna-fat. The electrical properties of the flagged pinna voxels are unchanged. The IEEE Std 1528 definition for the pinna was followed and the choice of each flagged voxel was confirmed by an Ear-Nose-Throat surgeon.
The Benchmark Validation Study calls for each participant to run twelve simulations: three head models, at two frequencies (835 and 1900 MHz), and in two cell phone positions (touch and tilted). The models are SAM, the Visible Human, and a seven year old Japanese male . Each model is provided as a voxel file with an ASCII header file.
For the two anatomically correct models, the tissue names and properties in the header file were made consistent with the definitions found on the Italian National Research Council, Institute for Applied Physics web site .
Although they are not part of the benchmark validation study, SCC34/SC2/WG2 plans on releasing several new anatomically correct models in the next few months to expand the population of models available for study.
The reference points, necessary for positioning the cell phone relative to the anatomically correct model, are also contained in the header file for each model.
3.4 FDTD considerations
The FDTD technique is called for by the P1529 draft . FDTD was chosen because it is stable and accurate, doesn't require enormous computational resources and can handle complex geometries.
The participants are asked to report the following FDTD data for all simulations: The boundary conditions used and the minimum distance between the model and the boundary of the computational space. The time step size and the number of time steps used. The grid (voxel) size and whether the grid was homogeneous or graded. To calculate the SAR in each voxel the protocol recommends the 12-field components approach. All 1- or 10-gram spatial averaging volumes are to be constructed in accordance with IEEE C95.3, Annex E.
The participants are asked to report the following SAR data for all simulations: The peak spatial SAR, both 1 g and 10 g averages, for all tissue (head plus pinna), head only, and pinna only averaging volumes and the location of the averaging cubes. The peak point value SAR and its location. A color coded SAR distribution for both 1 g and 10 g averages, in the ZY plane.
3.5.1 Scaling reported results
For a realistic study it would be ideal to simulate the real world situation. The question that remains is: "Does a real-world cell phone keep the power or the feed-point current constant when placed next to different human beings with different head shapes and head sizes"? Unfortunately there is not a definitive answer to this question. The behavior of a mobile phone depends on the system design and the power amplifier circuits. A detailed discussion for real-world mobile telephones has to be addressed by future projects with more realistic mobile phones models for numerical simulations.
For now it is important to scale the calculated SAR values to net input power and feed-point current and to present both results. The behavior of a real-world mobile phone is within the SAR range of scaling to the net input power and scaling to the feed-point current. For human health and safety considerations a worst case approach is desirable. Until further knowledge on the behavior of a real-world cell phone is available, the scaling producing the worst case result (largest SAR value) must be taken into account.
The current version of IEEE Std C95.1  does not classify the pinna as an extremity making it subject to the basic SAR exposure limitation of 1.6 W/kg over 1 g. However the much anticipated 200X revision of C95.1 will reclassify the pinna as an extremity raising its SAR exposure limit to 4 W/kg over 10 g. Confusion over the inclusion or exclusion of the pinna in the SAR averaging volume will continue until the IEEE officially releases C95.1-200X. The IEEE should release C95.1-200X as soon as practical, and if this can not be done in a reasonably short time, a supplement should be published clarifying the new status of the pinna as an extremity.
Investigators should inspect all models after rotation to be sure they are free of artifacts caused by meshing along the new coordinate axes. If necessary, artifacts should be manually corrected before running the simulation. Blindly accepting the output of meshing algorithms can lead to errors. All relevant data and assumptions for the computational RF dosimetry study, as discussed in section 2 "Problem areas", must be reported in such detail that the reader is able to compare the results to other studies. The names and electrical properties for all anatomically correct models should comply with those shown on the Italian National Research Council, Institute for Applied Physics web site.
To facilitate broad based comparisons, new anatomically correct models should be placed in the public domain or made available for a modest fee. The number of anatomically correct models suitable for electromagnetic modelling and widely available for comparison to the SAM is still low. Because the SAM is intended to represent a significant majority of persons during normal use of wireless handsets, comparison to a large variety of anatomically correct models is desirable. It is the hope of IEEE SCC34/SC2/WG2 that consistent results in the benchmark validation will show that, by adhering to some common definitions and procedures, FDTD studies from different investigators using different anatomically correct models are comparable.
Disclaimer: The opinions and conclusions stated in this article are those of the authors and do not represent the official position of the United States Food and Drug Administration.
- Lin JC: Cellular Mobile Telephones and Children. IEEE Antennas and Propagation Magazine. 2002,44(5):142–145.Google Scholar
- IEEE Std 1528–2003: Recommended Practice for Determining the Peak Spatial-Average Specific Absorption Rate (SAR) in the Human Head from Wireless Communications Devices – Measurement Techniques. Institute of Electrical and Electronics Engineers, New York 19 December 2003Google Scholar
- EN 50361: Basic Standard for the Measurement of Specific Absorption Rate Related to Exposure to Electromagnetic Fields from Mobile Phones (300 MHz – 3 GHz). European Committee for Electrical Standardization (CENELEC), Brussels 2001.Google Scholar
- IEC 62209: Procedure to measure the Specific Absorption Rate (SAR) in the frequency range of 300 MHz to 3 GHz – Part 1: hand-held mobile wireless communication devices. International Electrotechnical Commission, committee draft for vote Google Scholar
- ARIB STD-T56: Specific Absorption Rate (SAR) Estimation for Cellular Phone. Association of Radio Industries and Businesses 2002.Google Scholar
- Supplement C to OET Bulletin 65 (Edition 9701): Evaluating Compliance with FCC Guidelines for Human Exposure to Radio Frequency Electromagnetic Fields. Federal Communications Commission (FCC), Washington, DC 1997.Google Scholar
- Gordon CC, Churchill T, Clauser CE, Bradtmiller B, McConville JT, Tebbetts I, Walker RA: 1988 Anthropometric Survey of U.S. Army Personnel: Methods and Summary Statistics, Technical Report NATICK/TR-89/044. U.S. Army Natick Research, Development and Engineering Center, Natick, Massachusetts 1989.Google Scholar
- Mazzurana M, Sandrini L, Vaccari A, Malacarne C, Cristoforetti L, Pontalti RA: Semi-automatic method for developing an anthropomorphic numerical model of dielectric anatomy by MRI. Physics in Medicine and Biology 2003, 48: 3157–3170. 10.1088/0031-9155/48/19/005View ArticleGoogle Scholar
- Gjonaj E, Bartsch M, Clemens M, Schupp S, Weiland T: High-resolution human anatomy models for advanced electromagnetic field computations. IEEE Transactions on Magnetics 2002, 38: 357–360. 10.1109/20.996096View ArticleGoogle Scholar
- Gandhi OP, Kang G: Some present problems and a proposed experimental phantom for SAR compliance testing of cellular telephones at 835 and 1900 MHz. Physics in Medicine and Biology 2002, 47: 1501–1518. 10.1088/0031-9155/47/9/306View ArticleGoogle Scholar
- Kuster N, Christ A, Chavannes N, Nikoloski N, Frolich J: Human Head Phantoms for Compliance and Communication Performance Testing of Mobile Telecommunication Equipment at 900 MHz. 2002 Interim International Symposium on Antennas and Propagation, Yokosuka Reserach Park, Japan November 26–28, 2002Google Scholar
- Christ A, Chavannes N, Pokovic K, Gerber H, Kuster N: Numerical and Experimental Comparison of Human Head Models for SAR Assessment. Proceedings of the Millennium Workshop on Biological Effects of Electromagnetic Fields, Heraklion, Kreta, Greece 2000, 234–240.Google Scholar
- Lee A, Choi H, Lee H, Pack J: Human head size and SAR characteristics for handset exposure. ETRI Journal 2002, 24: 176–179.View ArticleGoogle Scholar
- Dimbylow PJ, Mann SM: SAR calculations in an anatomically realistic model of the head for mobile communications transceivers at 900 MHz and 1.8 GHz. Physics in Medicine and Biology 1994, 39: 1537–1553. 10.1088/0031-9155/39/10/003View ArticleGoogle Scholar
- Van de Kamer JB, Lagendijk JJW: Computation of high-resolution SAR distributions in a head due to a radiating dipole antenna representing a hand-held mobile phone. Physics in Medicine and Biology 2002, 47: 1827–1835. 10.1088/0031-9155/47/10/316View ArticleGoogle Scholar
- Wang J, Fujiwara O: Comparison and evaluation of electromagnetic absorption characteristics in realistic human head models of adult and children for 900-MHz mobile telephones. IEEE Transactions on Microwave Theory and Techniques 2003, 51: 966–971. 10.1109/TMTT.2003.808681View ArticleGoogle Scholar
- Okoiewski M, Stuckly MA: A study of the handset antenna and human interaction. IEEE Trans. Microwave Theory Tech 1996, 44: 1855–1864. 10.1109/22.539944View ArticleGoogle Scholar
- Hombach V, Meier K, Burkhardt M, Kühn E, Kuster N: The dependence of EM energy absorption on human head modeling at 900 MHz. IEEE Transactions on Microwave Theory and Techniques 1996, 44: 1865–1873. 10.1109/22.539945View ArticleGoogle Scholar
- Bernardi P, Cavagnaro M, Pisa S: Evaluation of the SAR distribution in the human head for cellular phones used in a partially closed environment. IEEE Transactions of Electromagnetic Compatibility 1996, 38: 357–366. 10.1109/15.536066View ArticleGoogle Scholar
- Wiart J, Dale C, Bosisio AV, Le Cornec A: Analysis of the influence of the power control and discontinuous transmission on RF exposure with GSM mobile phones. IEEE Transactions on Electromagnetic Compatibility 2000, 42: 376–385. 10.1109/15.902307View ArticleGoogle Scholar
- Luebbers R, Baurle R: FDTD Predictions of Electromagnetic Field in and near Human Bodies Using Visible Human Project Anatomical Scans. IEEE AP-S International Symposium and URSI Radio Science Meeting, Baltimore, MD July 21–26, 1996Google Scholar
- Martinez-Burdalo M, Martin A, Anguiano M, Villar R: Comparison of FDTD-calculated specific absorption rate in adults and children when using a mobile phone at 900 and 1800 MHz. Physics in Medicine and Biology 2004, 49: 345–354. 10.1088/0031-9155/49/2/011View ArticleGoogle Scholar
- Gandhi OP, Lazzi G, Furse CM: Electromagnetic Absorption in the Human Head and Neck for Mobile Telephones at 835 and 1900 MHz. IEEE Trans Microwave Theory and Techniques 1996, 44: 1884–1897. 10.1109/22.539947View ArticleGoogle Scholar
- Hadjem A, Lautru D, Dale C, Wong M, Fouad-Hanna V, Wiart J: Comparison of Specific Absorption Rate (SAR) Induced in Child-Sized and Adult Heads Using a Dual Band Mobile Phone. Proceedings on IEEE MTT-S International Microwave Symposium IMS 2004.Google Scholar
- Mochizuki S, Watanabe S, Taki M, Yamanaka Y, Shirai H: Size of Head Phantoms for Standard Measurements of SAR Due to Wireless Communication Devices. Electronics and Communications in Japan, Part 1 2004, 87.Google Scholar
- IEEE Std C95.1–1999: IEEE Standard for Safety Levels with Respect to Human Exposure to Radio Frequency Electromagnetic Fields, 3 kHz to 300 GHz. Copyright 1992 by the Institute of Electrical and Electronics Engineers (IEEE), Inc., New York, NY Google Scholar
- ICNIRP (International Commission on Non-Ionizing Radiation Protection), Guidelines for Limiting Exposure to Time-Varying Electric, Magnetic and Electromagnetic Fields (Up to 300 GHz) Health Physics 1998, 74: 494–522.Google Scholar
- Anderson V: Comparisons of peak SAR levels in concentric sphere head models of children and adults for irradiation by a dipole at 900 MHz. Physics in Medicine and Biology 2003, 48: 3263–3275. 10.1088/0031-9155/48/20/001View ArticleGoogle Scholar
- Kanda M, Balzano Q, Russo P, Faraone A, Bit-Babik G: Effects of ear-connection modeling on the electromagnetic-energy absorption in a human head phantom exposed to a dipole antenna field at 835 MHz. IEEE Transactions on Electromagnetic Compatibility 2002, 44: 4–10. 10.1109/15.990704View ArticleGoogle Scholar
- Schönborn F, Burkhardt M, Kuster N: Differences in energy absorption between heads of adults and children in the near field of sources. Health Physics 1998, 74: 160–168.View ArticleGoogle Scholar
- Kuster N, Balzano Q: Energy absorption mechanism by biological bodies in the near field of dipole antennas above 300 MHz. IEEE Transactions on Vehicular Technology 1992, 41: 17–23. 10.1109/25.120141View ArticleGoogle Scholar
- Caputa K, Okoniewski M, Stuchly MA: An Algorithm for Computations of the Power Deposition in Human Tissue. IEEE Antennas and Propagation Magazine 1999, 41: 102–107. 10.1109/74.789742View ArticleGoogle Scholar
- IEEE 1529: Recommended Practice for Determining the Spatial-Peak Specific Absorption Rate (SAR) Associated with the Use of Wireless Handsets – Computational Techniques. Institute of Electrical and Electronics Engineers, New York, draft standard Google Scholar
- Dielectric Properties of Body Tissue in the frequency range 10 Hz – 100 GHz Italian National Research Council, Institute for Applied Physics, Florence, Italy [http://niremf.irac.cnr.it/tissprop]
This article is published under license to BioMed Central Ltd. This is an open-access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.