- Open Access
An active electrode for biopotential recording from small localized bio-sources
© Valchinov and Pallikarakis; licensee BioMed Central Ltd. 2004
- Received: 02 June 2004
- Accepted: 22 July 2004
- Published: 22 July 2004
Laser bio-stimulation is a well-established procedure in Medical Acupuncture. Nevertheless there is still a confusion as to whether it works or the effect is just placebo. Although a plethora of scientific papers published, showing positive clinical results, there is still a lack of objective scientific proofs about the bio-stimulation effect of lasers used in Acupuncture. The objective of this work was to design and build a body surface electrode and an amplifier for biopotential recording from acupuncture points, considered here as small localized bio-sources (SLB). The design is aimed for studying SLB potentials provoked by laser stimulus, in search for objective proofs of the bio-stimulation effect of lasers used in Medical Acupuncture.
The active electrode presented features a new adjustable anchoring system and fractionation of the biopotential amplifier between the electrode and the cabinet's location. The new adjustable electrode anchoring system is designed to reduce the electrode-skin contact impedance, its variation and motion artifacts. That is achieved by increasing the electrode-skin tension and decreasing its relative movement. Additionally the sensing element provides local constant skin stretching thus eliminating the contribution of the skin potential artifact. The electrode is attached to the skin by a double-sided adhesive pad, where the sensing element is a stainless steel, 4 mm in diameter. The fractionation of the biopotential amplifier is done by incorporating the amplifier's front-end op-amps at the electrodes, thus avoiding the use of extra buffers. The biopotential amplifier features two selectable modes of operation: semi-AC-mode with a -3 dB bandwidth of 0.32–1000 Hz and AC-mode with a bandwidth of 0.16–1000 Hz.
The average measured DC electrode-skin contact impedance of the proposed electrode was 450 kΩ, with electrode tension of 0.3 kg/cm2 on an unprepared skin of the inner forearm. The peak-to-peak noise voltage measured at the amplifier output, with input terminals connected to common, was 10 mVp-p, or 2 μVp-p referred to the input. The common-mode rejection ratio of the amplifier was 96 dB at 50 Hz, measured with imbalanced electrodes' impedances. The prototype was also tested practically and sample records were obtained after a low intensity SLB laser stimulation. All measurements showed almost a complete absence of 50 Hz interference, although no electrolyte gel or skin preparation was applied.
The results showed that the new active electrode presented significantly reduced the electrode-skin impedance, its variation and motion artifact influences. This allowed SLB signals with relatively high quality to be recorded without skin preparation. The design offers low noise and major reduction in parts, size and power consumption. The active electrode specifications were found to be better or at least comparable to those of other existing designs.
- Active Electrode
- Contact Impedance
- Amplifier Circuit
- Medical Acupuncture
- Biopotential Amplifier
The non-invasive nature of laser bio-stimulation have made lasers an attractive alternative in Medical Acupuncture at the last 25 years. Unfortunately there is still a confusion as to whether they work or their effect is just placebo. Although a plethora of scientific papers published, showing positive clinical results, there is still a lack of objective scientific proofs about the bio-stimulation effect of lasers used in Acupuncture. The objective of this work was to design and built a body surface electrode and an amplifier for biopotential recording from acupuncture points. The latter are considered here as small localized bio-sources (SLB). As discussed by other authors, SLB are small area body regions with specific electrical, physiological and anatomical properties (e.g. high density of gap junctions, relatively low impedance etc.) [1–4]. They appear to be highly sensitive to mechanical, thermal, electrical or electromagnetic stimulation and are found to take place from the epidermis (stratum granulosum) to a maximum depth of 2 cm [5–8].
The active electrode is aimed for studying SLB potentials provoked by laser stimulus, in search for objective proofs of the bio-stimulation effect of lasers used in Medical Acupuncture.
The attempt to define the optimal parameters of the active electrode was based on a set of preliminary measurements performed in our laboratory. Anatomical, physiological and electrical characteristics of the signal source were considered. The SLB AC signal level, after stimulation, varied from subject to subject, but did not exceed 1 mV peak-to-peak (p-p). Additionally SLB occasionally manifested a high DC potential up to 200 mV, implying the use of differential amplifier with optional DC coupling and wide DC input range. The frequency band of the signal of interest was found to be in the range 0–200 Hz. The preliminary experiments showed that SLB potentials were best recorded with small pasteless electrodes although their contact impedance depends strongly on sweat gland secretion. The application of electrolytic gel resulted in significant reduction of the SLB signal amplitude, probably due to smoothing of the potential caused by saturation of the epidermis with electrolyte. Moreover, potentials between closely spaced SLB might be shortened by the application of excessive gel or large surface electrodes. An additional difficulty is that the SLB are often situated at convex or concave body surface areas where large flat electrodes could not be easily affixed. Skin abrasion with sandpaper is also not recommended since it can cause skin irritation and SLB potential changes. However, the use of small passive dry electrodes on an unprepared skin results in high electrode-skin impedance, motion artifacts, high power-line cable interference and noise. When the electrodes are DC coupled to the amplifier, a motion induced interfering signal appears at the amplifier input, mainly due to:
Electrokinetic effect – the disturbance of the double layer of charge at the electrode-skin interface causes variations of the DC polarization potential .
Skin potential or skin stretch artifact – stretching of the skin causes a change of the potential of the barrier layer between the epidermis and the dermis .
Variation in the electrode-skin contact resistance – caused by the amplifier input bias current and the current flowing due to the polarization potential.
V mot = ΔV pol + ΔV skin + (ΔR d + ΔR s ) (V POL / R i + i b ) (1)
where ΔVskin is the skin stretch artifact and ib is the amplifier input bias current. It was deduced that in order to keep the resistive interfering component less than 10 μV when DC coupling is employed and with both currents contributing equally to it, then ib<50 pA and Ri>1 GΩ . If an AC coupling is used then the resistive component of the motion artifact is eliminated.
Basic amplifier circuit
where Ad1(s) is the differential-mode (DM) gain and Ac1(s) is the common-mode (CM) gain of op-amp A1. If we take the usual definitions for the DM input signal, Vd=(E1-E0), and for the CM input signal, Vc=(E1 + E0)/2, then the output voltage can be also written as
U 1(s) = A d (s)V d + A c (s)V c (3)
It can be shown that the respective expressions for the DM gain Ad(s), and the CM gain Ac(s), are given by
Considering (4) and (5), then the common-mode rejection ratio CMRR(s) is
Assuming op-amps A0 and A1 are ideal then the only factor contributing to the CMRR is the mismatching of the resistors. Thus we can define a common-mode rejection ratio for the resistors, CMRRR. By taking 1/Ad0(s) = Ac(s) = 0 in (6) we obtain
Therefore CMRRR(s) approaches infinity if the relevant impedances are chosen according to
If the condition in (8) is fulfilled and op-amps A0 and A1 are ideal, then (4) simplifies to
Considering (7), equation (6) can be written as
where CMRRA1(s) is the CMRR of op-amp A1. Further if we assume that Z1, R2, R3 and ZE have tolerance t, then from (7) and (8) we can deduce that the worst case condition will be when Z1 = Z10(1-t), R2 = R20(1+t), R3 = R30(1-t) and ZE = ZE0(1+t) where Z10, R20, R30, and ZE0, are the respective nominal values. Equation (7) then can be written as
The CMRRA1 has the form
where ωr is the frequency where CMRRA1 has decreased by 3 dB and is usually between 100 Hz and 1 kHz. The open loop gain Ad0(s), also decreases at higher frequencies with a corner frequency ω0 = 1/τ0 which is usually lower than ωr, if A0 and A1 are of the same type. Therefore the CMRR is mainly determent at low frequency by the matching of the resistors and the DM gain, and at high frequencies by the open loop response of op-amp A0, rather than its CMRR. If we take the advantage of the fact implicit in (10), and achieve
then theoretically the CMRR becomes infinite. In part this can be achieved by the use of a capacitor and resistor in parallel for the impedance Z1 (Fig. 3), and then trimming R2. Thus the need of low-tolerance components is eliminated. Therefore Z1(s) will have the form
It has been shown  that a good approximation for the optimal value of the capacitor C1 is
where GBPA0 is the gain bandwidth product of op-am A0. Trimming R2 is a good solution for achieving an ultra high CMRR for demanding application, however it is not practical since the trimmer has to be incorporated in the electrode. Alternatively, ZE or R3 can be trimmed, which however will alter also the amplifier DM gain. Considering equation (12) it can be shown that for application with relatively high DM gain and proper op-amp selection, both trimming and compensation (C1) can be omitted, without significantly degrading the CMRR. For example, if the usual 1% tolerance resistors are used and op-amps with CMRR of 100 dB and DM open loop gain of 120 dB at 50 Hz, then for an amplifier with DM gain of 5000, a CMRR of 96 dB can be achieved without trimming.
In the amplifier circuit shown in Fig. 3, ZE is replaced with an active DC rejection/suppression circuit . It includes an integrator (A2, Ri, Ci) and two potential dividers (R6, R5 and R4, R3). The amplifier can operate in AC-mode or in semi-AC-mode. The two modes are selectable by the switch S1: AC-mode with S1 open and semi-AC-mode with S1 closed. In AC-mode the DC signals are rejected, where in semi-AC-mode they are suppressed. If R6 = R6 *, R5 = R5 * and Ri = Ri *, then the respective expressions for the equivalent impedance ZE(s) for the two modes are given by
where τi = RiCi is the time constant of the integrator, and τ2 are the respectively the DM open loop gain and the time constant of the first pole of op-amp A2. Whenever Ri>>R5 then k ≈ (R6/R5+1), which is true with the time constants and voltage gains, typical in biopotential recordings. For signals bellow the amplifier high-pass cut-off frequency, ZE(s) decreases due to the active DC rejection/suppression circuit. For DC signals equation (8) is maximally imbalanced and thus CMRRR(0) ≈ Ad(0). Since for biopotential amplifiers Ad(0) is much lower than CMRRA1(0) and Ad0(0), therefore CMRR(0) ≈ Ad(0), which represents the worst case.
If we consider only the -3 dB bandwidth and assume that op-amp A2 is ideal, then (17) and (18) simplify to
Therefore, in this case (9) can be written as
which represents the mid band DM gain for both modes. After substituting (17) and (18) in (9), it can be shown that the respective DC differential gains for the two modes are given by
where 2k is approximately the DC gain of the stopped integrator (A2, Ri, Ci, Ri *, R5 *, R6 *) in semi-AC-mode. Thus DC signals meet lower gain, in order to prevent saturation from large electrode offsets or other high DC potentials.
The active electrodes' input resistances RiE0 and RiE1, are not equal due to the different closed loop gains of op-amps A0 and A1, and can be expressed as
where RiA0 and RiA1 are the input resistances of op-amps A0 and A1. However, at higher frequencies, the electrodes' input impedances are much lower and about the same (assumed that A0 and A1 are of the same type), due to the op-amps' input and additional stray capacitance, being in parallel to the high op-amps' input resistance.
The output noise spectral density for the -3 dB bandwidth is approximately the same for both modes and can be written as
where en0, en1 and en2 are the respective voltage noises of op-amps A0, A1 and A2.
Assuming E0 is connected to common (Fig. 3), then the amplifier transfer function H(s) is given by
After substituting ZE(s) and Ad1(s) in (24), it can be shown that H(s) has three poles and two zeros for both modes. However, with the time constants and voltage gain used in the current application, one pole almost coincides with one zero. Therefore, H(s) can be approximated very well by a transfer function with two poles and one zero. The respective approximations for AC-mode and semi-AC-mode are given by
The circuits described by the transfer functions HAC(s) and HsAC(s) are stable because all the poles are situated in the left half of the complex s-plane and there are no resonance effects as the poles are on the real s-axis.
Practical amplifier circuit
Because of the large integrator's time constant, the amplifier has a very slow response after overloads (≤ 10τi), caused by large signal disturbances. Thus a deblocking circuit was added at the cabinet's location, for temporary reduction of the time constant during overload . It is controlled by the output voltage U1 through the low pass filter (R13, C2). The filter output controls two threshold triggers (A3, A4), which through D1,D2 control the MOS transistor T1, acting like a switch. When the output signal reaches its range limits (defined by R14, R15, R16), T1 opens and the new reduced time constant τi* = (R7//R11)C2, pulls the output signal to the zero level. This state is maintained for additional hundred milliseconds (R13C2) and then is switched back to its original value.
The connection between the amplifier common and the signal source is implemented by a driven right-leg (DRL) circuit. The CM voltage at the output of A0 is reduced by a factor equal to the DRL circuit gain (ADRL = 314 at 50 Hz), which theoretically should give a 50 dB extra CMRR at 50 Hz. In addition, in case of a faulty op-amp, the DRL circuit will limit the maximum patient current to a safe level of 50 μA.
Active electrode specifications
Bandwidth (-3 dB)
AC mid band gain
Differential mode AC input range
Differential mode DC input range
± 370 mV
Common mode input range
± 2 V
Input noise current
1 pArms @ 0.1–200 Hz
Input bias current
Input impedance, Active Electrode
320 MΩ @ 50 Hz (1000 GΩ //10 pF)
96 dB @ 50 Hz
Input noise voltage
2 μVp-p (0.33 μVrms) @ 0.1–200 Hz
11 mW @ one channel
The best solution for an active electrode would be to perform the entire analog signal processing at the electrode site. This could be achieved with a custom made integrated circuit, but the cost would be much higher. We found a good alternative in using SMD technology and integrating only the front-end of the amplifier into the electrode.
The ultra high input resistance of the electrode is degraded at higher frequencies by the op-amp's input capacitance in parallel with the stray capacitance due to the electrode Printed Circuit Board (PCB). Nevertheless, combining an op-amp with low input capacitance and a proper PCB design, allowed a relatively high input impedance to be achieved at 50 Hz. That decreased the amplifier sensitivity to high electrode-skin impedance imbalances, by reducing the transformation of the CM interference signal into unwanted DM signal. Unfortunately, most data sheets do not properly specify op-amp's input capacitance, neither DM nor CM.
The active electrode presented is not suitable for applications requiring a low differential gain and large signal bandwidth due to the decreasing CMRR at higher frequencies, if not properly compensated. On the other hand, below the high-pass cut-off frequency, the CMRR is degraded by the active feedback circuit, and reaches its minimum value for DC signals, equal to the DM gain. The circuit can accept high value input filter resistances, which will also limit the patient auxiliary current in case of fault condition of op-amps A0 and A1. Because of the limited electrode space, it is preferable that the front-end op-amps feature internal electrostatic discharge protection circuitry, rather than building an external one.
The new electrode anchoring system significantly reduced the electrode-skin impedance, its variation and motion artifact influences. The proposed amplifier fractionation resulted in lower noise and less parts. Moreover splitting the amplifier between the electrodes and the cabinet's location allowed the use of an automatic DC deblocking system and mode switching. The prototype tests showed that with the active electrode presented, SLB signals with relatively high quality could be recorded without skin preparation. The 50 Hz interference pickup by the electrode leads was practically eliminated. Because high electrode-skin impedances are tolerated, no electrolytic gel is needed. This allows fast application of the electrodes, minimizes patient discomfort and eliminates the risk of infection.
With proper op-amps selection, the active electrode specifications were found to be better or at least comparable to those of other existing designs. The design offers low noise and major reduction in parts, size and power consumption. It is currently used in studying laser provoked SLB potentials and their propagation, aiming to gain a better insight into the bio-stimulation effect of lasers in Medical Acupuncture.
- Mashansky VF, Markov UV: Topography of the gap junctions in the human skin and their possible role in the non-neural signal transduction. Arch Anat Histol Embryol 1983, 84: 53–60.Google Scholar
- Reichmanis M, Marino AA, Becker RO: Electrical correlates of acupuncture points. IEEE Trans Biomed Eng 1975, 22: 533–535.View ArticleGoogle Scholar
- Reichmanis M, Marino AA, Becker RO: Laplace plane analysis of transient impedance between acupuncture points Li-4 and Li-12. IEEE Trans Biomed Eng 1977, 24: 402–405.View ArticleGoogle Scholar
- Johng HM, Cho JH, Shin HS, Soh KS, Koo TH, Choi SY, Koo HS, Park MS: Frequency dependence of impedances at the acupuncture point Quze (PC3). IEEE Eng Med Biol Mag 2002, 21: 33–36. 10.1109/MEMB.2002.1000183View ArticleGoogle Scholar
- Litscher G: Effects of acupressure, manual acupuncture and Laserneedle acupuncture on EEG bispectral index and spectral edge frequency in healthy volunteers. Eur J Anaesthesiol 2004, 21: 13–19.View ArticleGoogle Scholar
- Litscher G: Cerebral and peripheral effects of laser needle-stimulation. Neurol Res 2003, 25: 722–728. 10.1179/016164103101202237View ArticleGoogle Scholar
- Niboyet J: Traite d'acupuncture Maisonneuve: Sainte-Ruffine 1979.Google Scholar
- Stux G, Pomeranz B: Basics of Acupuncture New York: Springer-Verlag 1998.View ArticleGoogle Scholar
- Khan A, Greatbatch W: Physiologic electrodes. In Medical Engineering (Edited by: Ray CD). Chicago: Year Book Medical Publishers 1974, 1073–1082.Google Scholar
- Travis C: Nonlinear aspects of the bioelectrode-electrolyte interface. In Biomedical electrode technology (Edited by: Miller HA). New York: Academic Press 1974, 143–159.Google Scholar
- Zipp P, Ahrens H: A model of bioelectrode motion artifact and reduction of artifact by amplifier input stage design,. J Biomed Eng 1979, 1: 273–276.View ArticleGoogle Scholar
- Neuman MR: Biopotential electrodes. In Medical Instrumentation, Application and Design 2 Edition (Edited by: Webster JG). New York: John Wiley and Sons 1995, 227–287.Google Scholar
- Searle A, Kilrup L: A direct comparison of wet, dry and insulating bioelectric recording electrodes. Physiol Meas 2000, 21: 271–283. 10.1088/0967-3334/21/2/307View ArticleGoogle Scholar
- Godin DT, Parker PA, Scott RN: Noise characteristics of stainless-steel surface electrodes. Med Biol Eng Comput 1991, 29: 585–590.View ArticleGoogle Scholar
- Richardson PC: Some new electrode techniques for long-term physiological monitoring. Aerosp Med 1968, 39: 745–750.Google Scholar
- Ko WH, Hynecek J: Dry electrodes and electrode amplifiers. In Biomedical electrode technology (Edited by: Miller HA, Harrison DC). New York: Academic Press 1974, 169–181.Google Scholar
- MettingVanRijn AC, Kuiper AP, Dankers TE, Grimbergen CA: Low cost active electrode improves the resolution in biopotential recordings. In Proceedings of the 18th Ann Int Conf IEEE Eng Med Biol, Amsterdam 1: 101–102. 31 October-3 November 1996Google Scholar
- Hagemann B, Luhede G, Luczak H: Improved active electrode for recording bioelectric signals in work physiology. Eur J Appl Physiol Occup Physiol 1985, 54: 95–98.View ArticleGoogle Scholar
- Padmadinata FZ, Veerhoek JJ, Van Dijk GJA, Huijsing JH: Microelectronic skin electrode. Sensors and Actuators 1990, B1: 491–494.View ArticleGoogle Scholar
- MettingVanRijn AC, Peper A, Grimbergen CA: Amplifiers for bioelectric events: a design with a minimal number of parts,. Med Biol Eng Comput 1994, 32: 305–310.View ArticleGoogle Scholar
- Bergey GE, Squires RD, Sipple WC: Electrocardiogram recording with pasteless electrodes. IEEE Trans Biomed Eng 1971, 18: 206–211.View ArticleGoogle Scholar
- Geddes LA: Electrodes and the measurement of bioelectric events New York: Wiley-Interscience 1972.Google Scholar
- Rosell J, Colominas J, Riu P, Pallas-Areny R, Webster JG: Skin impedance from 1 Hz to 1 MHz. IEEE Trans Biomed Eng 1988, 35: 649–651. 10.1109/10.4599View ArticleGoogle Scholar
- MettingVanRijn AC, Kuiper AP, Honsbeek RH, Speijer K, Peper A: DC rejection and deblocking in multichannel bioelectric recordings. In Procceedings of the 17th Ann Int Conf IEEE Eng Med Biol, Montreal 2: 1665–1666. 20–23 September 1995Google Scholar
This article is published under license to BioMed Central Ltd. This is an open-access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.