- Open Access
Comparison of quantitative computed tomography analysis and single-indicator thermodilution to measure pulmonary edema in patients with acute respiratory distress syndrome
BioMedical Engineering OnLine volume 13, Article number: 30 (2014)
To compare quantitative computed tomography (CT) analysis and single-indicator thermodilution to measure pulmonary edema in patients with acute respiratory distress syndrome (ARDS).
Ten patients with ARDS were included. All underwent spiral CT of the thorax for estimating gas content of lung (GVCT), tissue volume of lung (TVCT), tissue volume index (TVI), mean radiographic attenuation (CTmean) for the whole lung and gas-to-tissue ratio (g/t). Pulmonary thermal volume (PTV) and extravascular lung water index (ELWI) were determined by the PiCCO plus system. CT or single-indicator thermodilution variables were correlated with respiratory system compliance (Crs), PaO2/FiO2, and Acute Physiology And Chronic Health EvaluationII (APACHE II) and Sequential Organ Failure Assessment (SOFA) scores.
1) TVCT and PTV were positively correlated (r =0.8878; P = 0.0006; equation of regression line: PTV = 1.0793 × TVCT + 179.8) as were TVI and ELWI (r =0.9459; P < 0.0001; equation of regression line: ELWI = 1.4506 × TVI-8.7792). The bias between TVCT and PTV as well as TVI and ELWI was -277 ± 217 and 0.62 ± 4.56, respectively. 2) ELWI and CT distribution of lung-tissue compartments were not correlated. 3) CT or single-indicator thermodilution variables were not correlated with Crs, PaO2/FiO2 or APACHE II or SOFA score.
Quantitative CT analysis and single-indicator thermodilution showed good agreement in measuring pulmonary edema.
Interstitial lung edema is the most important pathological character of acute lung injury (ALI) and acute respiratory distress syndrome (ARDS) [1, 2]. Quantitative measurement of lung edema can provide a useful marker of disease severity and prognosis [3, 4]. Quantitative CT analysis and single-indicator thermodilution have been used to quantify lung edema clinically and experimentally.
Quantitative CT analysis provides a measure of lung volume, gas content (GVCT) and tissue volume (TVCT) . It also helps divide the whole lung into 4 compartments by radiographic attenuation values of normally aerated, poorly aerated, nonaerated, and hyperinflated . Although a classification of lung regions according to CT density is often used, various lung components (tissue, vessel, blood, edema) cannot be differentiated by radiologic density [6–11]. Moreover, radiation exposure, the risk of transferring critically ill patients, the relative high costs, and restricted availability all limit obtaining repeated measurements, so CT is not an ideal method to follow the clinical evolution of lung injury.
Extravascular lung water (EVLW) has been widely used to quantify pulmonary edema and guide fluid resuscitation in critically ill patients . Single-indicator thermodilution is a relatively simple and safe method for repeatedly measuring EVLW. Unfortunately, many factors that may affect the accuracy of the single-indicator thermodilution method include vascular obstruction by emboli, thrombosis, severe hypoxic vasoconstriction, and, possibly, high airway pressure, which are common in patients with ALI/ARDS [13–22].
The gravimetric method of estimating EVLW is an experimental method and is the gold standard; it can be used to compare the wet and dry weight of the lung in animals or humans on autopsy, for only one measurement. The accuracy of the single-indicator method in measuring lung water has been confirmed by the gravimetric method in animals and human studies [23–26]. In an experimental animal study, quantitative CT analysis showed good agreement with the gravimetric method .
Measuring lung edema by the thermal indocyanine green-dye double-dilution method has shown good agreement with that by quantitative CT . However, in mainland China, the single-indicator thermodilution method is used more than the thermal indocyanine green-dye double-dilution method. Therefore, we aimed to compare quantitative CT analysis and single-indicator thermodilution for diagnosis measurement of pulmonary edema in patients with ARDS.
Materials and methods
From March to December 2010, we included 10 consecutive patients with ARDS according to the criteria from the European American Consensus Conference on ARDS . Patients had been admitted to the intensive care unit (ICU) of Qilu Hospital, Shandong University, and had undergone lung CT and PiCCO plus system measurement for clinical monitoring. We excluded patients who were < 18 years old, were pregnant, or had chronic obstructive pulmonary disease, recent arrhythmia; unstable angina or myocardial infarction; chest wall trauma; or high intracranial pressure. The Ethics Committee of Qilu Hospital, Shandong University, approved the study protocol and informed consent was obtained from the patient’s next of kin.
Each patient underwent insertion of a 4 F central venous catheter (ARROW Co., USA) in the subclavian vein with an injectate temperature-sensor housing (PV4046, Pulsion Medical Systems, Munich, Germany) and a 5 F femoral arterial catheter with a thermistor tip (PV2015L20, Pulsion Medical Systems, Munich, Germany). Both catheters were connected to the PiCCO plus system (Pulsion Medical Systems V7.1, Munich, Germany). Ventilation was maintained throughout the protocol.
Quantitative CT analysis
Spiral CT of the thorax was performed from the apex to diaphragm (GE Lightspeed, GE Healthcare, USA), with continuous monitoring of heart rate, arterial blood pressure, and oxygen saturation. CT parameters were collimation scan, 20 mm; exposure, 120 kV and 220 mA; table speed, 25 mm/sec; and pitch, 1:1.
When patients were transported for CT scan, the ventilatory settings were maintained as in the ICU ward. The CT scan was performed quickly during an end-expiratory pause. Previous ventilatory settings were promptly reestablished at the end of the scan. CT images were analyzed by use of a home-made dedicated software, ARDS Viewer (Zhang Fan, Wu Dawei, Qilu Hospital, Shandong University, China). The ARDS Viewer allows for drawing the contour of the lung directly on a computer monitor (Figure 1). The mediastinum and pulmonary hila containing the trachea, main bronchi, and hilar blood vessels were excluded from the region of interest. Each element of the 512 × 512 matrix contained the value of the radiographic attenuation expressed in Hounsfield units (HU) corresponding to a volume of tissue (voxel).
Lung volume (gas + tissue) was calculated as the number of voxels present in a given lung region . Despite the complex interface between gas (with radiologic density ~ -1000 HU) and tissue (radiologic density ~ 0 HU [water density]), one can compute the volume of gas, volume of tissue, overall lung volume, and volume distribution of lung aeration for any lung region [5, 6]. The mean CT attenuation of a given lung region (CT lung density) is equivalent to its aeration: for example, -300 HU, the lung region is composed of 70% tissue and 30% gas (lung aeration = 30%). Therefore, the gas volume and tissue volume of each voxel can be calculated as follows [5, 6]:
All voxels in the matrix have the same volume depending on the size of the pixels of the matrix and the thickness of the CT section (0.5 cm). Therefore, the volume of gas (GVCT) and volume of tissue (TVCT) of the overall lung can be obtained. The mean radiographic attenuation value (CTmean) for the whole lung was computed by averaging the attenuation value of all voxels in the region of interest. The gas-to-tissue ratio (g/t) was computed as the ratio between GVCT and TVCT. The tissue volume index (TVI) was obtained by indexing TVCT to predicted body weight of the patient.
Classically, lung aeration is quantified into 4 compartments [7–10]: 1) normal aeration, CT attenuation -900 to -500 HU, corresponding to a normal ventilation perfusion ratio; 2) overinflation, CT attenuation < -900 HU, corresponding to alveolar dead space if the lung region is insufficiently perfused; 3) poor aeration, CT attenuation -100 to -500 HU, which corresponds radiologically to “ground-glass” opacification or reticular pattern and physiologically to venous admixture if the lung region remains perfused; and 4) nonaeration, CT attenuation > -100 HU, corresponding radiologically to consolidation and atelectasis and physiologically to true pulmonary shunt if the lung region remains perfused. We computed the ratio for the tissue volumes of each compartment (Figure 2).
The PiCCO plus system involves a thermal indicator to determine EVLW, cardiac output (CO), and volumetric measures. A central venous catheter and femoral arterial catheter with a thermistor tip are inserted for measurement. Cold saline (8°C), 15 ml volume, is injected into the central venous catheter. Then the thermistor tip on the femoral arterial catheter measures the downstream temperature change within the abdominal aorta. A transpulmonary thermodilution curve can be drawn. The cardiac output is calculated by the modified Stewart Hamilton equation from the area below the transpulmonary thermodilution curve. From the mean transit time (MTt) and the down slope time (DSt) of the exponential thermodilution curve, preload and lung water values are determined.
The intrathoracic thermal volume (ITTV) can be calculated by multiplying the CO by the MTt. The thermal indicator mixes with the largest possible volume of distribution.
The ITTV is composed of the pulmonary thermal volume (PTV) and the global end-diastolic volume (GEDV). PTV can be determined as the product of CO and the exponential DSt as follows.
Sakka and coworkers  examined the relationship between intrathoracic blood volume (ITBV) and GEDV and derived the following equation using linear regression to describe this relationship.
EVLW reflects all fluid, including interstitial and alveolar, that is outside of the pulmonary vasculature during transit of the thermal indicator. EVLW is calculated as the difference between the ITTV and ITBV as follows.
In the PiCCO plus system, ITBV and GEDV are indexed to the body surface area of each patient to obtain the intrathoracic blood volume index (ITBI) and global end-diastolic volume index (GEDI); EVLW is indexed to predicted body weight for the extravascular lung water index (ELWI). All values were obtained just after the patient returned to the ICU after CT scan.
Other related values
Immediately after CT scan, arterial blood was drawn for blood gas analysis, PaO2/FiO2 was calculated; and respiratory system compliance (Crs) was recorded. The Acute Physiology And Chronic Health Evaluation II (APACHE II) and Sequential Organ Failure Assessment (SOFA) scores were obtained from the worse record for each physiological variable within the 24 hr before and after CT scan.
Data are presented as mean ± SD. To evaluate the association of PTV with TVCT and ELWI with TVI, we performed a linear regression analysis. Agreement between values was assessed by the Bland and Altman plot method. Correlation was evaluated between the main values (PTV, ELWI, CTmean, GVCT, TVI, and g/t) and PaO2/FiO2, Crs, and APACHE II and SOFA scores. To assess whether pulmonary edema was associated with any specific tissue compartment, we correlated the ELWI with the relative tissue volume for each compartment. All correlations were assessed by linear regression analysis with the Pearson correlation coefficient r. Statistical significance was set at P<0.05. Statistical analyses involved use of SAS 9.2 (SAS Inst., Cary, NC).
We included 10 patients (mean age 48.60 ± 18.08 years; 8 males); characteristics are in Table 1. Each patient showed a PaO2/FiO2 < 200 mmHg, with an overall mean 103.87 ± 54.42 mmHg. ARDS was due to pneumonia for 6 patients and multiple trauma for 2, with abdominal infection for 2. Half of the patients presented a large amount of pleural fluid.
Quantitative CT analysis
On CT scan, TVCT was 1221.37 ± 383.68 ml, TVI 18.12 ± 6.79 ml/kg, GVCT 916.58 ± 418.08 ml, CTmean -407.97 ± 118.30 HU, and g/t 0.80 ± 0.40 (Table 2). Nonaerated, poorly aerated, normally aerated, and hyperinflated tissue represented 32.55 ± 9.26%, 39.92 ± 13.82%, 27.46 ± 9.29%, and 0.06 ± 0.11% of the TVCT, respectively (Figure 2).
With the PiCCO plus system, PTV and ELWI were 1498.1 ± 466.45 mL and 17.5 ± 10.4 ml/kg, respectively. Individual values for GEDI, ITBI, PTV, and ELWI are in Table 2.
Correlation between quantitative CT analysis and single-indicator thermodilution values
Mean TVCT and PTV were positively correlated (r =0.8878; P = 0.0006; PTV = 1.0793 × TVCT + 179.8) and PTV values overestimated those of TVCT (Figure 3). Mean TVI and ELWI were positively correlated (r =0.9459; P < 0.0001; ELWI = 1.4506 × TVI - 8.7792). The agreement between TVI and ELWI was better than that for TVCT and PTV, although mean TVI was slightly larger than ELWI (Figure 4). The bias between TVCT and PTV, TVI and ELWI was -277 ± 217 and 0.62 ± 4.56, respectively.
We found no correlation between ELWI and proportion of any lung-tissue compartments. The differences between TVI and ELWI were not correlated with proportion of normally aerated, poorly aerated or nonaerated lung compartments.
We found no significant correlation between physiological variables (PaO2/FiO2, Crs, APACHE II and SOFA score) and CT or single-indicator thermodilution variables (Table 3).
In patients with ARDS, lung tissue content measured by quantitative CT analysis was highly correlated with ELWI and PTV measured by the single-indicator thermodilution method.
As a convenient, minimally invasive, and economical technique, PiCCO monitoring has been generally used in ICUs all over the world. As compared with the gravimetric method, single-indicator thermodilution to determine EVLW shows good correlation in animal models of lung injury, although EVLW is slightly higher with the latter method [23–25]. Tagami and colleagues found that measurement of EVLW by the single-transpulmonary system is closely correlated with post-mortem lung weight in humans . Overestimation of EVLW may have several explanations. First, the thermal indicator is also equilibrated with the myocardium and vessel walls, which may lead to a large volume of distribution and a small increase in EVLW measurement. Second, recirculation of cold saline can overestimate lung water . Third, the linear regression equations between ITBV and GEDV may differ among species [25, 30].
An animal experiment indicated that as compared with postmortem gravimetry, TVI estimated by quantitative CT had acceptable accuracy in tracking the extent of pulmonary edema, with a mean bias of 4 ml/kg . Recent experimental and clinical investigations have revealed that CT-derived pulmonary tissue volume is closely associated with EVLW and PTV determined by thermal indocyanine green-dye double-indicator dilution, but TVCT showed a tendency to be larger than PTV or EVLW [27, 28]. This overestimation might be due to the additive effects of several factors. First, CT does not differentiate between lung compartments with interstitial fluid, pulmonary tissue, or residual intra-vascular blood. Second, CT reveals fluids in the alveoli and pleural space that might escape detection with indicator dilution methods.
Many factors may limit the reliability of EVLW determined by single-indicator thermodilution. First, the relationship between GEDV and ITBV may be affected by anatomy (height, weight), mechanics (tidal volume, PEEP), physiologic features (pulmonary edema, hypoxic vasoconstriction), and pharmacologic features (vasoactive drugs) [13, 31]. The equation between GEDV and ITBV differs among species [25, 30]. So EVLW may be affected by all of these factors. Second, the thermodilution method for measuring EVLW relies on heat exchange across the alveolar epithelial and endothelial barriers. If the thermal indicator does not have access to all lung tissue because of decreased perfusion, then EVLW will be consistently underestimated. Several pathophysiological changes in critically ill patients, such as large pulmonary vessel obstruction, pulmonary vascular microembolism, and hypoxic vasoconstriction, can impair the perfusion and result in an inaccurate EVLW [14, 15]. Easley and co-workers found that inactivating hypoxic pulmonary vasoconstriction by endotoxin administration increased pulmonary blood flow to poorly aerated regions, with increased EVLW . Third, PEEP may affect the measured value of EVLW and directly affect the amount of lung water present, thereby indirectly affecting EVLW. High levels of PEEP may lead to pulmonary capillary collapse, thus potentially resulting in underestimating EVLW. However, the use of PEEP may cause a redistribution of pulmonary blood flow toward previously underperfused lung regions, thus leading to a more accurate EVLW. Several trials have given different results for the influence of PEEP on measurement of EVLW [18–21]. Fourth, the degree of lung edema can affect the accuracy of EVLW. However, the exact influence is still controversial [13, 21, 22]. Finally, many anatomic and physiological abnormalities that can alter EVLW include major pulmonary resection, the presence of large aortic aneurysms, arterial catheters placed too far peripherally, and intracardiac shunts .
The results of this study indicate that in patients with ARDS, EVLW and PTV measured by the single-indicator thermodilution method were highly correlated with lung tissue content measured by CT. However, EVLW and PTV determined with single-indicator thermodilution overestimated TVCT perhaps because CT cannot differentiate between lung compartments with interstitial fluid, pulmonary tissue, or residual intra-vascular blood. As well, other factors that can limit the reliability of EVLW measured by single-indicator thermodilution include the level of PEEP, the extent of the shunt in the lungs, and recirculation of cold saline. One trail estimated the CT-based ELWI by two different radiologists without analyzing software . CT-based estimation of ELWI is not accurate for predicting extravascular lung water in critically ill patients when compared to single-indicator thermodilution, because they only use some CT findings to guess the ELWI . But in our study, the quantitative CT analysis is much more accurate. The presence of a large amount of pleural fluid in 5 patients may have affected the reliability of EVLW obtained by single-indicator thermodilution. Two previous studies revealed that fluid in the pleural space did not contribute to the measurement of EVLW [34, 35]. However, one of the studies indicated that pleural fluid contributed to increased PTV . Saugel and coworkers recently found that large-volume thoracentesis significantly increased EVLI measured by single-indicator thermodilution . However, EVLI was higher after removal of pleural fluid; the authors concluded that pleural effusions did not take part in single-indicator transpulmonary thermodilution as part of the pulmonary thermovolume and did not increase EVLI .
ELWI and the differences between TVI and ELWI were not correlated with proportions of any lung compartments. This result agreed with Patroniti and coworkers , but we lack other data for comparison.
We found no correlation between CT or PiCCO variables and Crs or PaO2/FiO2. This result agreed with some previous research [26, 37, 38], although it still remains controversial. PaO2/FiO2 was affected by many factors, such as PEEP and inhaled oxygen concentration. Even for the same patient, the position, sedation state, and airway secretions can lead to changes in PaO2/FiO2 . Dynamic changes in EVLW showed a significant negative correlation with PaO2/FiO2 and Crs changes , which suggests that the dynamic monitoring of EVLW is important for the assessment of severity and treatment effects.
APACHE II and SOFA scores have been widely used to assess the severity of critically ill patients and were closely related to the prognosis of multiple organ dysfunction syndrome [41, 42]. We found no correlation between ELWI or TVI and these scores; APACHE II and SOFA scores may not be specific markers of acute lung injury but they can help in assessing the overall condition of patients.
Because of the small population and the lack of previous data to compare with our findings, further studies are necessary to confirm our results. As well, the time of mechanical ventilation before enrollment and ventilator settings were not the same for each patient, which may have affected our results.
Measuring pulmonary edema by quantiative CT analysis shows good agreement with that by single-indicator thermodilution.
Interstitial lung edema is the most important pathological character of acute respiratory distress syndrome. Quantitative measurement of lung edema can provide a useful marker of disease severity and prognosis.
Measurement of pulmonary edema by quantitative CT analysis showed good agreement with that by single-indicator thermodilution, although extravascular lung water and pulmonary thermal volume values were underestimated by the latter method for tissue volume of the lung.
As a noninvasive method, quantitative CT analysis is a gold standard for evaluating lung aeration, which may have clinical relevance in the mechanical ventilation setting and provide a quantitative “in vivo” measurement to estimate pulmonary edema.
Acute lung injury/acute respiratory distress syndrome
- APACHE II:
Acute Physiology And Chronic Health Evaluation II
Compliance of respiratory system
Compliance of respiratory system
Mean radiographic attenuation value for the whole lung
Exponential downslope decay time
Extravascular lung water
Extravascular lung water index for predicted body weight
Intrathoracic blood volume index for patient surface area
Intrathoracic thermal volume
Intrathoracic blood volume
Global end-diastolic volume index for patient surface area
Global end-diastolic volume
Ratio of gas to tissue volume
Quantitative CT analysis of gas volume
Multiple organ dysfunction syndrome
Mean transit time
Positive end-expiratory pressure
Pulmonary thermal volume
Pulmonary blood volume
Sequential Organ Failure Assessment
Tissue volume index for predicted body weight
Quantitative CT analysis of tissue volume.
Bernard GR, Artigas A, Brigham KL, Carlet J, Falke K, Hudson L, Lamy M, Legall JR, Morris A, Spragg R: The American-European Conference on ARDS: definitions, mechanisms, relevant outcomes, and clinical trial coordination. Am J Respir Crit Care Med 1994, 149: 818–824. 10.1164/ajrccm.149.3.7509706
Howell DCJ, Bellingan GJ: Respiratory Disease and its Management: Acute Lung Injury and Acute Respiratory Distress Syndrome (ALI/ARDS)[M]. London: Springer; 2009.
Sakka SG, Klein M, Reinhart K, Meier-Hellmann A: Prognostic value of extravascular lung water in critically ill patients. Chest 2002, 122: 2080–2086. 10.1378/chest.122.6.2080
Kuzkov VV, Kirov MY, Sovershaev MA, Kuklin VN, Suborov EV, Waerhaug K, Bjertnaes LJ: Extravascular lung water determined with single transpulmonary thermodilution correlates with the severity of sepsis-induced acute lung injury. Crit Care Med 2006,34(6):1647–1653. 10.1097/01.CCM.0000218817.24208.2E
Puybasset L, Cluzel P, Chao N, Slutsky AS, Coriat P, Rouby JJ, and the CT Scan ARDS Study Group: A computed tomography scan assessment of regional lung volume in acute lung injury. Am J Respir Crit Care Med 1998, 158: 1644–1655. 10.1164/ajrccm.158.5.9802003
Puybasset L, Cluzel P, Gusman P, Gremier P, Preteux F, Rouby J-J, CT scan ARDS study group: Regional distribution of gas ang tissue in acute respiratory distress syndrome.I.Consequences for lung morphology. Intensive Care Med 2000, 26: 857–869. 10.1007/s001340051274
Rouby JJ, Puybasset L, Nieszkowska A, Lu Q: Acute respiratory distress syndrome: lessons from computed tomography of the whole lung. Critical care medicine 2003,31(no.4 Suppl):S285-S295.
Malbouisson LM, Muller JC, Constantin JM, Lu Q, Puybasset L, Rouby JJ, and the CT Scan ARDS Study Group: Computed tomography assessment of positive end-expiratory pressure–induced alveolar recruitment in patients with acute respiratory distress syndrome. Am J Respir Crit Care Med 2001, 163: 1444–1450. 10.1164/ajrccm.163.6.2005001
Gattinoni L, Caironi P, Pelosi P, Goodman LR: What has computed tomography taught us about the acute respiratory distress syndrome? Am J Respir Crit Care Med 2001, 164: 1701–1711. 10.1164/ajrccm.164.9.2103121
Puybasset L, Gusman P, Muller JC, Cluzel P, Coriat P, Rouby JJ, and the CT scan ARDS study group: Regional distribution of gas and tissue in acute respiratory distress syndrome.III.Consequences for the effects of positive end-expiratory pressure. Intensive Care Med 2000, 26: 1215–1227. 10.1007/s001340051340
Gattinoni L, Cressoni M: Quantitative CT in ARDS: towards a clinical tool? Intensive Care Med 2010, 36: 1803–1804. 10.1007/s00134-010-2016-0
Brown LM, Liu KD, Matthay MA: Measurement of extravascular lung water using the single indicator method in patients: research and potential clinical value. Am J Physiol Lung Cell Mol Physiol 2009, 297: L547-L558. 10.1152/ajplung.00127.2009
Michard F, Schachtrupp A, Toens C: Factors influencing the estimation of extravascular lung water by transpulmonary thermodilution in critically ill patients. Crit Care Med 2005, 33: 1243–1247. 10.1097/01.CCM.0000164566.23147.35
Noble WH, Kay JC: Effect of emboli, positive pressure ventilation and airway water on lung water measurement. Appl Physiol 1988, 65: 154–164.
Roch A, Michelet P, Lambert D, Delliaux S, Saby C, Perrin G, Ghez O, Bregeon F, Thomas P, Carpentier JP, Papazian L, Auffray JP: Accuracy of the double indicator method for measurement of extravascular lung water depends on the type of acute lung injury. Crit Care Med 2004, 32: 811–817. 10.1097/01.CCM.0000114831.59185.02
Shen JF, Qiu HB, Yang Y, Liu SQ, Chen YM, Li JQ, Wu B, Ding HM: Comparison of single-indicator thermodilution versus gravimetric measurement in determination of extra-vascular lung water in dogs with acute respiratory distress syndrome. Chin Crit Care Med 2006, 18: 327–330.
Michard F: Bedside assessment of extravascular lung water by dilution methods: temptations and pitfalls. Crit Care Med 2007, 35: 1186–1192. 10.1097/01.CCM.0000259539.49339.66
Myers JC, Reilley TE, Cloutier CT: Effect of positive end-expiratory pressure on extravascular lung water in porcine acute respiratory failure. Crit Care Med 1988, 16: 52–54. 10.1097/00003246-198801000-00010
Ruiz-Bailén M, Fernández-Mondéjar E, Hurtado-Ruiz B, Colmenero-Ruiz M, Rivera-Fernández R, Guerrero-López F, Vázquez-Mata G: Immediate application of positive end-expiratory pressure is more effective than delayed positive end-expiratory pressure to reduce extravascular lung water. Crit Care Med 1999, 27: 380–384. 10.1097/00003246-199902000-00046
Carlile PV, Lowery DD, Gray BA: Effect of PEEP and type of injury on thermal-dye estimation of pulmonary edema. Appl Physiol 1986, 60: 22–31.
Schuster DP, Anderson C, Kozlowski J, Lange N: Regional pulmonary perfusion in patients with acute pulmonary edema. J Nucl Med 2002, 43: 863–870.
Fernández-Mondéjar E, Castaño-Pérez J, Rivera-Fernández R, Colmenero-Ruiz M, Manzano F, Pérez-Villares J, de la Chica R: Quantification of lung water by transpulmonary thermodilution in normal and edematous lung. J Crit Care 2003, 18: 253–258. 10.1016/j.jcrc.2003.10.009
Katzenelson R, Perel A, Berkenstadt H, Preisman S, Kogan S, Sternik L, Segal E: Accuracy of transpulmonary thermodilution versus gravimetric measurement of extravascular lung water. Crit Care Med 2004, 32: 1550–1554. 10.1097/01.CCM.0000130995.18334.8B
Kirov MY, Kuzkov VV, Kuklin VN, Waerhaug K, Bjertnaes LJ: Extravascular lung water assessed by transpulmonary single thermodilution and postmortem gravimetry in sheep. Crit Care 2004, 8: R451-R458. 10.1186/cc2974
Rossi P, Wanecek M, Rudehill A, Konrad D, Weitzberg E, Oldner A: Comparison of a single indicator and gravimetric technique for estimation of extravascular lung water in endotoxemic pigs. Crit Care Med 2006, 34: 1437–1443. 10.1097/01.CCM.0000215830.48977.29
Tagami T, Kushimoto S, Yamamoto Y, Atsumi T, Tosa R, Matsuda K, Oyama R, Kawaguchi T, Masuno T, Hirama H, Yokota H: Validation of extravascular lung water measurement by single transpulmonary thermodilution: human autopsy study. Critical Care 2010, 14: R162. 10.1186/cc9250
Kuzkov VV, Suborov EV, Kirov MY, Waerhaug K, Mortensen R, Kuklin VN, Nordhus KC, Bjertnaes LJ: Radiographic lung density assessed by computed tomography is associated with extravascular lung water content. Acta Anaesthesiol Scand 2010, 54: 1018–1026. 10.1111/j.1399-6576.2010.02272.x
Patroniti N, Bellani G, Maggioni E, Manfio A, Marcora B, Pesenti A: Measurement of pulmonary edema in patients with acute respiratory distress syndrome. Crit Care Med 2005,33(11):2547–2554. 10.1097/01.CCM.0000186747.43540.25
Sakka SG, Rühl CC, Pfeiffer UJ, Beale R, McLuckie A, Reinhart K, Meier-Hellmann A: Assessment of cardiac preload and extravascular lung water by single transpulmonary thermodilution. Intensive Care Med 2000, 26: 180–187. 10.1007/s001340050043
Kirov MY, Kuzkov VV, Fernandez-Mondejar E, Bjertnaes LJ: Measuring extravascular lung water: animals and humans are not the same. Critical Care 2006, 10: 415. 10.1186/cc4989
Huber W, Mair S, Gotz SQ, Tschirdewahn J, Siege U, Schmid RM, Saugel B: Extravascular lung water and its association with weight, height, age, and gender: a study in intensive care unit patients. Intensive Care Med 2013, 39: 146–150. 10.1007/s00134-012-2745-3
Easley RB, Mulreany DG, Lancaster CT, Custer JW, Fernandez-Bustamante A, Colantuoni E, Simon BA: Redistribution of pulmonary blood flow impacts thermodilution-based extravascular lung water measurements in a model of acute lung injury. Anesthesiology 2009, 111: 1065–1074. 10.1097/ALN.0b013e3181bc99cf
Saugel B, Holzapfel K, Stollfuss J, Schuster T, Phillip V, Schultheiss C, Schmid RM, Huber W: Computed tomography to estimate cardiac preload and extravascular lung water. A retrospective analysis in critically ill patients. Scand J Trauma Resusc Emerg Med 2011, 19: 31. 10.1186/1757-7241-19-31
Deeren DH, Dits H, Daelemans R, Malbrain ML: Effect of pleural fluid on the measurement of extravascular lung water by single transpulmonary thermodilution. Clin Intensive Care 2004,15(4):119–122. 10.3109/09563070400013333
Blomqvist H, Wickerts CJ, Rosblad PG: Effects of pleural fluid and positive end-expiratory pressure on the measurement of extravascular lung water by the double-indicator dilution technique. Acta Anaesthesiol Scand 1991, 35: 578–583. 10.1111/j.1399-6576.1991.tb03352.x
Saugel B, Phillip V, Ernesti C, Messer M, Meidert AS, Schmid RM, Huber W: Impact of large-volume thoracentesis on transpulmonary thermodilution- derived extravascular lung water in medical intensive care unit patients. J Crit Care 2013, 28: 196–201. 10.1016/j.jcrc.2012.05.002
Küntscher MV, Blome-Eberwein S, Pelzer M, Erdmann D, Germann G: Transcardiopulmonary vs pulmonary arterial thermodilution methods for hemodynamic monitoring of burned patients. J Burn Care Rehabil 2002,23(1):2–26.
Blomqvist H, Wickerts CJ, Berg B, Frostell C, Jolin A, Hedenstierna G: Dose PEEP facilitate the resolution of extravascular lung water after experimental hydrostatic pulmonary oedema? Eur Respir J 1991,4(9):1053–1059.
Thorson SH, Marini JJ, Pierson DJ, Hudson LD: Variability of arterial blood gas values in stable patients in the ICU. Chest 1983, 84: 14–18. 10.1378/chest.84.1.14
Qiu HB, Sun HM, Yang Y, Xu HY, Chen YM: Changes of extravascular lung water in sheep with early acute respiratory distress syndrome. Chin J Tuberc Respir Dis 2004, 27: 537–541.
Vincent JL, de Mendonça A, Cantraine F, Moreno R, Takala J, Suter PM, Sprung CL, Colardyn F, Blecher S: Use of the SOFA score to assess the incidence of organ dysfunction/failure in intensive care units: results of a multicenter, prospective study. Working group on “sepsis-related problems” of the European Society of Intensive Care Medicine. Crit Care Med 1998,26(11):1793–1800. 10.1097/00003246-199811000-00016
Knaus WA, Draper EA, Wagner DP, Zimmerman JE: APACHE II: a severity of disease classification system. Crit Care Med 1985,13(10):818–829. 10.1097/00003246-198510000-00009
We are grateful to Yiqiang Zhan from Dapartment of Periodontology, University of Greifswaid, Greifswald, Germany, for help in statistical analysis. We thank the medical staff from the Department of critical Care Medcine, Qilu Hospital of Shandong University. This study received no specific funding.
The authors declare that they have no competing interests.
FZ, CL and DW designed the study. FZ, CL and JZ planned and collected the data. FZ, CL and HG wrote the manuscript. All authors read and approved the final manuscript.
Fan Zhang, Chen Li contributed equally to this work.
About this article
Cite this article
Zhang, F., Li, C., Zhang, Jn. et al. Comparison of quantitative computed tomography analysis and single-indicator thermodilution to measure pulmonary edema in patients with acute respiratory distress syndrome. BioMed Eng OnLine 13, 30 (2014). https://doi.org/10.1186/1475-925X-13-30
- Acute respiratory distress syndrome
- Pulmonary edema
- Quantitative CT analysis
- Single indicator thermodilution method
- Extravascular lung water