 Research
 Open Access
 Published:
Subspacebased technique for speckle noise reduction in ultrasound images
BioMedical Engineering OnLine volume 13, Article number: 154 (2014)
Abstract
Background and purpose
Ultrasound imaging is a very essential technique in medical diagnosis due to its being safe, economical and noninvasive nature. Despite its popularity, the US images, however, are corrupted with speckle noise, which reduces US images qualities, hampering image interpretation and processing stage. Hence, there are many efforts made by researches to formulate various despeckling methods for speckle reduction in US images.
Methods
In this paper, a subspacebased speckle reduction technique in ultrasound images is proposed. The fundamental principle of subspacebased despeckling technique is to convert multiplicative speckle noise into additive via logarithmic transformation, then to decompose the vector space of the noisy image into signal and noise subspaces. Image enhancement is achieved by nulling the noise subspace and estimating the clean image from the remaining signal subspace. Linear estimation of the clean image is derived by minimizing image distortion while maintaining the residual noise energy below some given threshold. The real US data for validation purposes were acquired under the IRB protocol (2002108517) at the University of California Davis, which is also consistent with NIH requirements.
Results
Experiments are carried out using a synthetically generated Bmode ultrasound image, a computer generated cyst image and real ultrasound images. The performance of the proposed technique is compared with Lee, homomorphic wavelet and squeeze box filter (SBF) in terms of noise variance reduction, mean preservation, texture preservation and ultrasound despeckling assessment index (USDSAI). The results indicate better noise reduction capability with the simulated images by the SDC than Lee, Wavelet and SBF in addition to less blurry effect. With the real case scenario, the SDC, Lee, Wavelet and SBF are tested with images obtained from raw radio frequency (RF) data. Results generated using real US data indicate that, in addition to good contrast enhancement, the autocorrelation results shows better preservation of image texture by SDC than Lee, Wavelet and SBF.
Conclusion
In general, the performance of the SDC filter is better than Lee, Wavelet and SBF in terms of noise reduction, improvement in image contrast and preservation of the autocorrelation profiles. Furthermore, the filter required less computational time compared to Lee, Wavelet and SBF, which indicates its suitability for real time application.
Introduction
Ultrasound (US) imaging is one of the most commonly used medical imaging due for diagnostic purposes to its many advantages such as portability, the noninvasive nature, relatively low cost and presents no radiation risk to patient. These features have made the US imaging as the most prevalent diagnostic tool for health practitioners over other more sophisticated imaging techniques such as CT scan, MRI or PET. Unfortunately, like SAR, US images exhibit a speckle pattern and its statistical model is identical to singlelook SAR amplitude signals. Speckle in ultrasound has adverse effect in such a way it causes reduction in image contrast resolution. In [1], Bamber and Daft show that speckle in US images cause reduction of lesion detectability by approximately a factor of eight.
An US machine works by introducing into the body of interest a lowenergy pulse of sound with frequencies typically between 3 and 30MHz by a transducer probe that touches the patients’ skin surface. Upon travelling through the body tissue, some of the pulses get attenuated while some small portion of the pulse energy are scattered back to the probe. The scattered pulse is then received by the same probe to produce echo signals which are processed to form twodimensional images, also known as sonogram. This twodimensional anatomical maps are called Bmode (brightness) images [2].
In principle, US images provide information about internal tissue structures which resulted from interaction between anatomical tissues with the transmitted ultrasound pulse. Due to interaction between ultrasound waves with tissue, backscattered echo signals are produced, in the form of reflection, scattering, interference and absorption. These echo, resulted from coherent summation of ultrasound scatterers, carry information about the tissue under investigation. The nature of coherent summation of such signals gives rise to an interference pattern known as speckle [3].
The despeckling techniques applied in US and SAR imagery can be classified into four main groups, namely, linear and nonlinear filters, adaptive speckle filters, waveletbased filters and anisotropic diffusionbased (AD) approach. In linear filtering technique [4, 5], the multiplicative speckle noise is first converted into an additive noise by applying logarithmic transformation to the speckled image followed by a Wiener filter in order to reject the resultant additive noise. The despeckled image is fully recovered by applying exponential transformation onto the output of Wiener filter. The technique, which convert the multiplicative speckle noise into an additive one, are commonly referred as homomorphic despeckling methods. The Wiener filter is the oldest approach to image denoising, is optimal in the sense of minimum meansquare error (MSE) and is space invariant linear estimator of the signal for images degraded by additive white noise.
The nonlinear filters are possible alternative to the standard linear filters, and the most popular one is the median filter. It has the advantage of preserving edges and is very effective at removing impulsive noise. The median filter sorts the intensities in the neighbourhood window of the reference pixel and calculates the median value of the sorted data. The denoised pixel is obtained by replacing the original reference pixel value by the median value calculated for the particular neigbourhood window [6–8]. The main problem is that the median filter would blur edges and tiny details.
Waveletbased denoising techniques continue to generate great interest among the computer vision and image processing community. Some of the proposed waveletbased speckle filters are presented in [9–15]. The success of the technique is due to the fact that in the wavelet domain, the noise is uniformly spread throughout the coefficients, while most of the image information is concentrated in few significant ones. In other word, the wavelettransformed images tend to be sparse and consequently, noise removal can be achieved by properly suppressing or thresholding the small coefficients that are likely due to noise. The waveletbased denoising techniques involve three major steps, 1) perform a 2D wavelet transform, 2) modify the noisy coefficients using a shrinkage function, and 3) perform a 2D inverse wavelet transform [16, 17]. In general, the most critical step in wavelet denoising techniques is the modification of wavelet coefficients. The classification of the different type of wavelet denoising is typically based on it different approach in modifying the noisy coefficients.
The adaptive speckle reducing filters such as Lee, Kuan and Frost can be applicable to both US and SAR images. The methods are developed based on multiplicative model of speckle noise. The methods are based on two assumptions, 1) the recorded image and the speckle noise are statistical independence [18], and 2) a constant ratio of noise standard deviation to mean throughout the image. The second assumption is valid in homogeneous regions. Each of these filters achieved speckle reduction via spatial filtering in a squaremoving window known as kernel. The filtering is based on the statistical relationship between the centre pixel and its surrounding pixels within a processing window. The typical window size are 3×3, 5×5, and 7×7. With the windowbased techniques, the selection of window will greatly affects the quality of the processed image. If the window is too small, the noise filtering algorithm is not effective, where as if the window is too large, subtle details of the image will be lost in the filtering process.
The squeeze box filter (SBF) which can be classified as an iterative technique, reduces speckle noise by suppressing outliers as a local mean of its neighborhood [19, 20]. Based on the fact that speckle is a stochastic process where outliers inevitably occurs, the proposed SBF achieves noise reduction by iteratively removes the outliers. Specifically, the image pixel outliers are defined to be local minimums and local maximums determined from a 3×3 window. Each outlier will be replaced by a local mean determined from a window centered on the outlying pixel. The outlier pixel value is not used in computing the local mean. After all the outliers are replaced by the local means, the process is repeated until a predetermined number of iteration is reached or until convergence is attained. In [19], experimental results showed that the SBF improves the image quality in terms of contrast enhancement, structural similarity and segmentation result. Although an effective speckle reduction, the SBF however still has artifacts in the form of blurred edges and irregular intensity pattern around edges [21].
In this paper, a subspacebased technique to reduce the speckle noise in US images, is proposed. Fundamentally, the proposed technique is an extension of the original work of Ephraim and Van Trees [22], in speech enhancement towards 2dimensional signals. The underlying principle is to decompose the vector space of the noisy image into a signalplusnoise subspace and the noise subspace. The noise removal is achieved by nulling the noise subspace and controlling the noise distribution in the signal subspace. For white noise, the subspace decomposition can theoretically be performed by applying the KarhunenLoeve transform (KLT) to the noisy image. Linear estimator of the clean image is performed by minimizing image distortion while maintaining the residual noise energy below some given threshold. For colored noise, a prewhitening approach prior to KLT transform, or a generalized subspace for simultaneous diagonalization of the clean and noise covariance matrices, can be used. The fundamental signal and noise model for subspace methods is additive noise uncorrelated with the signal. But, in US images the noise is multiplicative in nature, so a homomorphic framework takes advantage of logarithmic transformation, in order to convert multiplicative noise into additive noise.
The paper is organized as follows. Firstly, the statistic of speckle noise in US images is described. Secondly, the principle of subspace and how it can be extended to speckle noise removal is presented. In specific, this second section covers the proposed subspace technique and its implementation in speckle noise filtering followed by experimental results to determine optimum value of Lagrange multiplier. The subsequent section presents the experimental results to validate and evaluate the performance of the proposed filter. The performance evaluation of the proposed technique is divided into three main categories, 1) using simulated Bmode US images 2) using Field II generated images and 3)using real US images in comparison to Lee filter, wavelet filter [23, 24] in homomorphic framework and SBF technique [19]. The final section concludes this paper.
For clarity, an attempt has been made to adhere to a standard notational convention. Lower case boldface characters will generally refer to vectors. Upper case characters will generally refer to matrices. Vector or matrix transposition will be denoted using (.)^{T} and {\mathbb{R}}^{m\times m} denotes the real vector space of m×m dimensions.
Signal and noise model in ultrasound images
Consider matrix G to be the noisy observation of the original image, W. Let denote ξ _{ m } and ξ _{ a } as the set of corrupting multiplicative and additive speckle noise components, respectively. The noisy US image can be expressed as [4, 9, 11, 25]
Generally, in medical US images, the effect of the additive speckle noise (such as sensor noise) is considerably less significant than the multiplicative component [4, 9, 11, 25]. Taking the assumption that the speckle is fully developed and the additive term can be neglected, equation (1) can be expressed as
Applying the logarithmic function to both side of (2), we get
Expression (3) can be rewritten as
where Y,X and N are the logarithms of G,W and ξ _{ m } respectively.
The statistical theory to describe US speckle are drawn from the literature of laser optic by Goodman in [26]. Goodman mathematically models speckle as an accumulation of a large number of complex phasors z, to be denoted as z=a+j b, also known as complex random walk. These complex phasors, z can have either constructive or destructive relationship with each other. Applying central limit theorem to the random walk will results in a signal having twodimensional Gaussian probability density function (PDF) in the complex plane,
where v^{2} is the variance of the Gaussian distributed inphase/quadrature (IQ) components. Equation (5) is simply the product of two independent Gaussian density functions with zero mean and variance v^{2} and referred to as a circular Gaussian probability density function. Using the law of conservation of probability, the PDF of speckle phasors magnitude, A=\sqrt{{a}^{2}+{b}^{2}} is given by
For the intensity format, I=A^{2}, the PDF is given by [27]
The equation in (6) and (7) are respectively, known as Rayleigh PDF and exponentional PDF. In Bmode US signal, the magnitude A is the quantity of interest since the image is form using envelope detection, in which the phase components are removed. The histogram of the pixels in homogeneous area marked as “A” is shown in Figure 1 which shows a distribution consistent with Rayleigh distribution.
The subspacebased techniques for noise reduction
In this section, we derive the linear spatialdomain constraint (SDC) estimator, which minimizes the image distortion while constraining the energy of residual noise. The fundamental principle is to decompose the vector space of the noisy image into a signal subspace and noise subspace. The decomposition of the space into two subspaces can be done using either the singular value decomposition (SVD) or the eigenvalue decomposition (EVD). The noise removal is achieved by nulling the noise subspace and controlling the noise distribution in the signal (signal + noise) subspace. We begin with derivation of time (spatial) domain constraints estimator which minimizes the image distortion while constraining the energy of residual noise. Using the signal X and an additive noise model N, the noisy image matrix can be expressed as Y=X+N. In this case, the error signal ε obtained from the linear estimation, \widehat{X}=\mathit{\text{HY}} is given by
where ε _{ X } represents the image distortion, and ε _{ N } represents the residual noise [22]. Defining the energy of the image distortion {\stackrel{\u0304}{{\epsilon}_{X}}}^{2}, and the energy of the residual noise {\stackrel{\u0304}{{\epsilon}_{N}}}^{2} as
where E[·] is the expected value, the optimum linear estimator can be obtained by solving the following spatialdomain constrained optimization problem [22, 28]
where σ is a positive constant.
The optimum estimator is the sense of Eq. (11) can be found using the KuhnTucker necessary conditions for constrained minimization [29]. It involves solving a constrained minimization problem by applying the method of Lagrange multipliers [30]. Specifically, H is a stationary feasible point, if it satisfies the gradient equation of the Lagrangian,
where λ≥0 is the Lagrange multiplier, and
The solution to Eq. 12 is a stationary feasible point that satisfies the gradient equation, ∇_{ H } L(H,λ)=0, thus we obtain
thus,
Since the noise is assumed to be white, then {R}_{N}={v}_{n}^{2}I where {v}_{n}^{2} is the noise variance and I is the identity matrix. Hence, the solution for the optimum estimator H _{ SDC } is given as
Before the final form of the optimal estimator H _{ SDC } is considered, it is worthy to note that there is a strong empirical evidence indicating that the transformed covariance matrix of most images by the eigenvectors of the R _{ X } have some eigenvalues small enough to be considered as zeros. This means that the number of basis vectors for the pure image is smaller than the dimension of its vectors.To verify this key statement, we plot the eigenvalues of two ultrasound images of captured from a patient, as shown in Figure 2. The images shown in Figure 2 correspond to malignant and benign tumor obtained from biopsyverified studies. The image size is 1556×360 pixels where the xaxis giving the lateral sizes and the yaxis giving the axial sizes. Specifically, for the malignant tumor, the patient was diagnosed with IDC (Invasive Ductal Carcinoma) and for the benign tumor, the patient was diagnosed with fibroadenoma. The RF frames are recorded at 17 frame/second and a total of 12 seconds of data are acquired using a linear transducer array from the Antares®; System. In order to obtain the Bmode ultrasound images, the URI Offline Processing Tools (URIOPT) run on MATLAB platform is used to convert the RF data to the Bmode images as shown in Figure 2.
The eigenvalue plot in Figure 3, it shows that some of the eigenvalues of matrix R _{ X } are close to zero, which indicates that the energy of the clean image is distributed among a subset of its coordinates and the signal is confined to a subspace of the noisy Euclidean space. Since all noise eigenvalues are strictly positive, the noise fills in the entire vector space of the noisy image. In other word, the vector space of the noisy image is composed of a signalplusnoise subspace and a complementary noise subspace. The signalplusnoise subspace or simply the signal subspace comprises vectors of the clean image as well as of the noise process. The noise subspace contains vectors of the noise process only. Using eigendecomposition of R _{ X }=U Δ _{ X } U^{T}, Eq. (16) can be expressed as
The link between the maximal oriented energy and the signal subspace as well as between the minimal energy and the noise subspace were established in [31]. Using the eigendecomposition analysis [31], in which the {\Delta}_{X,i}={\Delta}_{Y,i}{v}_{n}^{2}, we can improve the form of model matrix H _{ SDC } in Eq. (17) by removing the noise subspace and estimating the clean image from the remaining principal signal subspace
In the implementation of SDC, a proper selection of signal subspace dimension r and Lagrangian multiplier, λ are critical in order to achieve the best noise reduction technique. For subspace dimension, a method based on eigenvalues is proposed in [31, 32] whereas the Lagrangian multiplier is to be empirically determined. As with any other noise filtering technique, the value noise variance needs to be estimated. In this case, the noise variance can be estimated using the last trailing end of the smallest singular value as outlined in [31].
When dealing with ultrasound data, the SDC is implemented in homomorphic framework where the noisy image is first logtransformed prior to SDC filtering. This transformation will convert the multiplicative nature of the speckle to an additive on. The final form of the despeckled image is recovered by performing antilog on the output of the SDC filter. The implementation detail of SDC are given as follows,

1.
Apply the homomorphic transformation to the noisy image, Y= log(G).

2.
Estimate the noise variance, {v}_{n}^{2}.

3.
Compute the dimension of signal subspace, r.

4.
Using the estimated r in step 3, apply eigendecomposition on {R}_{{Y}_{l}}, then extract the basis vectors of signal subspace U _{1}, and their related eigenvalues {\Delta}_{X}^{\left(i\right)}={\Delta}_{Y}^{\left(i\right)}{v}_{n}^{2}.

5.
Select the best value of λ, then compute the optimum linear estimator,
{H}_{\mathit{\text{SDC}}}={U}_{1}{\Delta}_{X1}{\left({\Delta}_{X1}+\lambda {v}_{n}^{2}I\right)}^{1}{U}_{1}^{T}.(19) 
6.
Compute the clean image,
\widehat{X}={H}_{\mathit{\text{SDC}}}\xb7\mathrm{Y.} 
7.
Reverse the homomorphic effect by taking the exponential of the \widehat{X} as follows
\u0174=1{0}^{\widehat{X}}.(20)
In essence, reversing the homomorphic effect in step 7 converts the logarithmic form of the filtered image to a linear form prior to image display.
Optimum value of the lagrange multiplier
To find the best λ value for SDC, a test image made up is created as shown in Figure 4. The test image is made up of synthetic patterns, specimens from Broadatz texture set, geometrical shapes, and some alphabets with different size. In particular, the bright and dark strips on the upper left corner closely resemble clinical ultrasound images of carotid artery at the far wall [33]. The test image is selected as it combines different critical features of typical US images. The Broadatz texture is to assess on how well the filter can preserves the texture of the original image. Besides, the different geometrical shapes and alphabets of different sizes are included in order to evaluate the filter capability in preserving edges and fine details of the image. Lastly, the selection of bright and dark strips that closely resemble clinical US images of carotid artery is to assess the filter capability in preserving the artery wall and its edges. The experiment is conducted by corrupting the test image speckle noise of variance extends from 0.03 to 0.05 and λ ranging between 1 and 105. The signaltonoise value (SNR) calculated as
where MSE represents the meansquare error, given by
is used to indicate the denoising effect of the SDC. The results are shown in Figure 5.
The results in Figure 5 show that the SDC is not too sensitive to the selected value of the Lagrange multiplier. Notably, the results in Figure 5 show that for high noise level, ({v}_{n}^{2}>0.04) the despeckle effect of the SDC, measured in terms of the SNR, shows improvement by 1 dB to 1.5 dB, as the Lagrange multiplier varies from 1 to 40. For lower value noise level, ({v}_{n}^{2}\le 0.04) the SNR improvement is around 0.3 dB as the Lagrange multiplier varies from 1 to 10. In general, the results in Figure 5 show better SNR values for higher values of the Lagrange multiplier. However, it should be noted that high value of λ may results in oversmoothed images and cause loss of details. Consequently, the rule of selecting λ is that for noise variance less than 0.04, λ should be selected to be around 10 and with noise variance greater than 0.04 it should be selected to be less than 40.
Results and discussions
The experimental results presented in this section can be divided into 2 parts. In the first part, the performance of the proposed SDC technique is compared with Lee [34], homomorphic wavelet filter [35] and SBF technique [19] using a simulated speckle image. With a known noisefree image, the performance of SDC is measured in terms of Peak SignaltoNoise Ratio (PSNR) defined as
The value of 255 in Eq. (23) corresponds to the maximum possible pixel value and MSE is defined as in (22).
In the second part, the performance of the proposed SDC technique is investigated using a computer generated image and real US images. Here, the Lee filter is implemented with 7×7 window size, the homomorphic wavelet is used with Daubechies lengtheight filter and a 7×7 window and the SBF technique is implemented according to the set up given in [19]. The SDC is implemented as in The subspacebased techniques for noise reduction section. The rank values and the noise variance of the different images are calculated using the method outlined in [31]. As for the Lagrange multiplier, the value is selected using the rule set in the previous section.
When using computer generated US or real US images, the noisefree image is not available which is the practical scenario of denoising applications of US images. Therefore, referencefree methods are used to quantitatively assess the denoising performance. The referencefree methods in this work are mean preservation, normalized variance, autocorrelation [36] and USDSAI [37]. Details on each assessment metric are as follows;

1.
Mean Preservation: A good speckle filter will maintains the mean intensity within a homogenous region.

2.
Normalized Variance: The normalized variance indicates the performance of the filter in homogeneous areas. This metric is given by
\frac{\mathit{\text{var}}}{\mathit{\text{mea}}{n}^{2}}=\frac{\frac{1}{\mathit{\text{mn}}}\sum _{i=1}^{m}\sum _{j=1}^{n}{\left(X\left(i,j\right)\stackrel{\u0304}{X}\right)}^{2}}{{\stackrel{\u0304}{X}}^{2}},(24)
where \stackrel{\u0304}{X} corresponds to the mean value of the pixel. In general, lower normalized variance values in the filtered image indicate better speckle suppression.

3.
Autocorrelation: is another method of filter assessment in homogeneous area where close autocorrelation profile to the original image indicates better texture preservation. The autocorrelation for m×n image X is given as [36]
\rho (x,y)=\frac{\frac{1}{\left(m\leftx\right\right)\left(n\lefty\right\right)}\sum _{i}\sum _{j}X(i,j)X(i+x,j+y)}{\frac{1}{\mathit{\text{mn}}}\sum _{i=1}^{m}\sum _{j=1}^{n}X{(i,j)}^{2}},(25)
where X(i,j) is the grey value of pixel (i,j).

4.
Ultrasound Despeckling Assessment Index (USDSAI): is a modified Fisher discriminant contrast metric [37]. USDSAI gives an indication on how well a despeckling algorithm reduces variances in homogeneous classes while keeping the distinct classes well separated. The metric is defined as
\mathit{\text{USDSAI}}=\frac{\sum _{k\ne l}\left(\mathit{\text{mea}}{n}_{{C}_{k}}\mathit{\text{mea}}{n}_{{C}_{l}}\right)}{\sum _{k=1}^{K}\mathit{\text{varianc}}{e}_{{C}_{k}}},(26)
where C _{ k } denotes the number of pixels in class C _{ k }. If a despeckling filter produces classes that are well separated then the numerator in 26 will be large. Conversely, if the intraclass variance is reduced, then the denominator will be small giving large value of USDSAI indicating desirable image restoration and enhancement.
Evaluation of SDC performance in simulated speckle noise scenario
In this experiment, the capability of the SDC technique in reducing the speckle noise is tested and compared with Lee, homomorphic wavelet and SBF technique. The performances of the noise reduction techniques are measured in terms of PSNR values as tabulated in Table 1.
The results in Table 1 show clearly the better reduction of noise achieved by SDC to Lee, Wavelet and SBF as the noise variance as the noise varies from 0.02 to 0.1. In average, the PSNR value of the SDC is improved by more 3dB followed by SBF(2.9dB), Lee (2.68dB) and Wavelet (1.1dB). However, in order to gain more insight into the performances of the SDC, the denoised images of Figure 4 by SDC, Lee, Wavelet and SBF are shown in Figure 6. Visual inspection of the denoised image by Lee in Figure 6 clearly shows the blurring effect of Lee filter. The wavelet on the other hand shows very close performance to the SDC except for some ringing effect which is visible in the homogeneous part of the image. The SBF exhibits some blurred edges with some noise are not removed around edges. In summary, SDC shows better noise reduction capability and less blurring effect in comparison to Lee and SBF and comparable performance to Wavelet, but with significantly less artifacts and better details preservations.
Evaluation of SDC performance using a Field II simulated image
In this experiment, the computer model of a cyst phantom is generated using the MATLAB Field II simulation [38, 39]. The phantom contains five point targets; 6, 5, 4, 3, 2 mm diameter waterfilled cysts, and 6, 5, 4, 3, 2 mm diameter high scattering regions. The resulted Bmode US image is shown in Figure 7. The “Cyst” phantom in Figure 7 is composed of 3 constant classes and the filters ability to reduce speckle noise while keeping the distinct classes well separated will be evaluated using normalized variance, mean preservation, preservation of autocorrelation [31] and USDSAI assessment metric. Prior to despeckling, the cyst image is converted into an 8bit image of size 512×512 pixels.
In the first experiment, the normalized variance and mean preservation for the cyst image are calculated over two selected regions labeled as A and B as in Figure 8. The normalized variances of the two regions calculated before and after denoising for SDC, Lee, Wavelet and SBF are presented in Table 2. The results in Table 2 show clearly the better reduction of noise achieved by SDC compared to Lee, Wavelet and SBF over the two homogeneous regions. In order to further verify the better better performance by the SDC, the denoised images of Figure 7 by SDC, Lee, Wavelet and SBF are shown in Figure 8. Visual inspection of the denoised images in Figure 8 clearly shows far less introduced blurring effect, better noise reduction, and better contrast enhancements by the SDC in comparison to the Lee, Wavelet and SBF. On the other hand, Figure 8 also shows that the SBF introduces relatively similar blurring effect to Lee and Wavelet though it gives better contrast enhancement values, measured in terms of USDSAI as tabulated in Table 3.
In addition to variance reduction, the values of mean preservation for the two regions calculated before and after denoising for SDC, Lee, Wavelet and SBF filter are also evaluated and included in Table 4. The results in Table 4 indicate the better capability by Lee to Wavelet, SBF and SDC in preserving the mean value in the computer generated cyst image in Figure 7. The better mean preservation by Lee is highly expected because of the averaging scheme of Lee filter which tends to maintain the mean value in the image.In order to assess the capability of the different algorithms in texture preservation in the denoised image, the autocorrelation in region A and B of the cyst image in Figure 7 are calculated before and after speckle filtering and depicted in Figure 9. The autocorrelation profiles in Figure 9 clearly show the better details preservation by the SDC in comparison to Lee, Wavelet and SBF. Notably, the profiles by Lee, Wavelet and SBF exhibit wider profiles in the neighbourhood of zero lag and largely deviated from the original at other lags. On the contrary, the SDC shows close autocorrelation profile of the denoised image to the original one in terms of shape and better preservation of the unit impulse structure at zero lag value than Lee, Wavelet and SBF.
Evaluation of SDC performance using real US images
In this experiment, the performance of the proposed SDC is analyzed and compared with Lee and Wavelet using ultrasound images captured from a patient as shown in Figure 2. The images are biopsyverified studies and presented with nonpalpable tumors initially detected by mammography [40]. These images are shown in Figure 2 for malignant and benign tumor. In Figure 2, the patient with malignant tumor was diagnosed with invasive ductal carcinoma whereas the patient with benign tumor was diagnosed with fibroadenoma. The image size is 1536×256 pixels with the xaxis and the yaxis giving lateral sizes and axial sizes of the image, respectively. The RF frames are recorded at 17 frame/second and a total of 12 seconds of data are acquired using a linear transducer array from the Antares^{®;} System. In order to obtain the Bmode ultrasound images, the URI Offline Processing Tools (URIOPT) run on MATLAB platform is used to convert the RF data to the Bmode images as shown in Figure 2.
In the first part of this experiment, two homogeneous areas are selected and marked as region A and B Figure 10. In order to assess the capability of the filters in reducing noise in image, variances are calculated over these two regions before and after denoising the image in Figure 2. The values of normalized variance are tabulated in Table 5. The results in Table 5 indicate the better noise reduction capability by the Wavelet in comparison to Lee, SBF and SDC which show a relatively comparable performance. However, in order to gain more insight into the performance of the Wavelet and to aid the interpretation of the results in Table 5, the denoised images by Lee, Wavelet, SBF and SDC are shown in Figure 11. The results in Figure 11 clearly show that the main reason for the high noise reduction values by the Wavelet in Table 5 is the intensive appearance of wavelet artifacts in its denoised image. On the other hand, though the SDC gives approximately similar values to Lee and SBF in Table 5, the denoised images in Figure 11 show clearly better noise reduction and image details preservation.
In addition to the noise reduction capability addressed by the normalized variance, the mean preservation capability is also tested and presented in Table 6. The results show the better performance of Lee in preserving mean value and this performance is very close to SDC. Notably, the mean value of Lee and SDC only differs by no more than 0.03. On the other hand, the result on Wavelet and SBF indicates poor preservation of mean by the two filters. In terms of contrast enhancement, given by the USDSAI values as shown in Table 7, the SDC gives better contrast enhancement to both Lee and Wavelet but a comparable performance to SBF.To gain more insight into the performance of the three considered techniques, their capability in preserving the characteristics of the original image is tested in terms of autocorrelation profiles of the selected region, A and B as shown in Figure 12. The results in Figure 12 give clear indication on the better preservation of the texture of the original image by SDC in comparison to Lee, Wavelet and SBF. In fact, the SDC shows close autocorrelation profile of the denoised image to the original one especially in term of shape and better preservation of the unit impulse structure at zero lag than Lee, Wavelet and SBF. Moreover, the autocorrelation profiles produced by Lee and Wavelet shows widened profiles at zero lag and largely deviated profiles from the original at other lags.
In the third experiment, the required computational time by Lee, Wavelet, SBF and SDC to process the ultrasound images of Figure 2 are calculated and included in Table 8. The filters are implemented on MATLAB platform using a computer with Intel(R) Xeon(R) 5607 @ 2.27 GHz processor and 8GB RAM. The results in Table 8 shows that the computational times of both SDC and wavelet are almost similar and less by nearly 3 times SBF and 10 times than Lee (7 × 7).
Conclusions
A subspacebased denoising technique for US images is presented and tested. The proposed technique, SDC is based on linear estimator and rank reduced subspace model to estimate the clean image from the corrupted one with speckle noise. The performance of the SDC is tested with simulated and real data, and compared with Lee and wavelet. The results indicate better noise variance reduction capability with the simulated images by the SDC than Lee, Wavelet and SBF in addition to less blurry effect. With the real case scenario, the SDC, Lee, Wavelet and SBF are tested with images obtained from raw RF data. The performances are calculated in terms of noise reduction, improvement in image contrast and preservation of the autocorrelation profiles. The results indicate that SDC offer better texture preservation, measured in terms of autocorrelation profiles and good contrast enhancement, measure in terms of USDSAI value. Finally, the computational complexity of the algorithms is compared and the results show that SDC required the least computational time compared to Lee, Wavelet and SBF.
References
Bamber JC, Daft C: Adaptive filtering for reduction of speckle in ultrasonic pulseecho images. Ultrasonics 1986, 24: 41–44. 10.1016/0041624X(86)900727
Insana MF: Ultrasonic Imaging. New Jersey: John Wiley & Sons, Inc; 2006.
Burckhardt C: Speckle in ultrasound Bmode scans. IEEE Trans Sonics Ultrason 1978, 25: 1–6.
Jain AK: Fundamentals of Digital Image Processing. New Jersey: Prentice Hall; 1989.
Abbott JG, Thurstone FL: Acoustic speckle: Theory and experimental analysis. Ultrasonic Imaging 1979,1(4):303–324. 10.1177/016173467900100402
Wang M, Lai CH: A Concise Introduction to Image Processing using C++. UK: Chapman & Hall/CRC; 2008.
Gonzalez RC, Wintz P: Digital Image Processing. Reading, MA: AddisonWesley; 2009.
Petrou M, Petrou C: Image Processing: The Fundamentals. New Jersey: Wiley; 2010.
Xuli Z, Laine AF, Geiser EA: Speckle reduction and contrast enhancement of echocardiograms via multiscale nonlinear processing. IEEE Trans Med Imag 1998,17(4):532–540. 10.1109/42.730398
Hao X, Gao S, Gao X: A novel multiscale nonlinear thresholding method for ultrasonic speckle suppressing. IEEE Trans Med Imag 1999,18(9):787–794. 10.1109/42.802756
Achim A, Bezerianos A, Tsakalides P: Novel Bayesian multiscale method for speckle removal in medical ultrasound images. IEEE Trans Med Imag 2001,20(8):772–783. 10.1109/42.938245
Solbo S, Eltoft T: Homomorphic waveletbased statistical despeckling of SAR images. IEEE Trans Geosci Remote Sens 2004,42(4):711–721.
Rabbani H, Nezafat R, Gazor S: Waveletdomain medical image denoising using bivariate laplacian mixture model. IEEE Trans Biomed Eng 2009,56(12):2826–2837.
Parrilli S, Poderico M, Angelino CV, Verdoliva L: A Nonlocal SAR image denoising algorithm based on LLMMSE wavelet shrinkage. IEEE Trans Geosci Remote Sens 2012,50(2):606–616.
Ranjani JJ, Thiruvengadam SJ: Dualtree complex wavelet transform based SAR despeckling using interscale dependence. IEEE Trans Geosci Remote Sens 2010,48(6):2723–2731.
Donoho DL, Johnstone IM: Ideal spatial adaptation via wavelet shrinkage. Biometrika 1994, 81: 425–455. 10.1093/biomet/81.3.425
Donoho DL, Johnstone IM: Adapting to unknown smoothness via wavelet shrinkage. J Am Stat Assoc 1995,. 90
Baraldi A, Parmiggiani F: An alternative form of the Lee filter for speckle suppression in SAR images. Graph Models Image Process 1995, 57: 75–78. 10.1006/gmip.1995.1008
Tay PC, Garson CD, Acton ST, Hossack JA: Ultrasound despeckling for contrast enhancement. IEEE Trans Image Process 2010,19(7):1847–1860.
Tay P, Shen H: A novel background subtraction method to detect microcalcifications. Image Analysis and Interpretation (SSIAI), 2012 IEEE Southwest Symposium on 2012, 141–144.
Kim DH, Plataniotis K, Ro YM: Denoising 3D ultrasound volumes using sparse representation. Engineering in Medicine and Biology Society (EMBC), 2013 35th Annual International Conference of the IEEE 2013, 4034–4037.
Ephraim Y, Trees HLV: A signal subspace approach for speech enhancement. IEEE Trans Speech Audio Process 1995,3(4):251–266. 10.1109/89.397090
Sendur L, Selesnick IW: Bivariate shrinkage functions for waveletbased denoising exploiting interscale dependency. IEEE Trans Signal Process 2002,50(11):2744–2756. 10.1109/TSP.2002.804091
Sendur L, Selesnick IW: Bivariate shrinkage with local variance estimation. IEEE Signal Process Lett 2002,9(12):438–441.
AbdElmoniem KZ, Youssef ABM, Kadah YM: Realtime speckle reduction and coherence enhancement in ultrasound imaging via nonlinear anisotropic diffusion. IEEE Trans Biomed Eng 2002,49(9):997–1014. 10.1109/TBME.2002.1028423
Goodman JW: Statistical Optics. New York: WileyInterscience; 1985.
Goodman JW: Statistical Properties of Laser Speckle Patterns. Heidelberg, Germany: SpringerVerlag; 1980.
Rezayee A, Gazor S: An adaptive KLT approach for speech enhancement. IEEE Trans Speech Audio Process 2001,9(2):87–95. 10.1109/89.902276
Luenberger D: Linear and nonlinear programming. Reading, MA: AddisonWesley; 1984.
Haykin S: Adaptive Filter Theory. Englewood Cliffs, NJ: PrenticeHall; 1991.
Yahya N, Kamel N, Malik A: Subspacebased technique for speckle noise reduction in SAR images. IEEE Trans Geosci Remote Sens 2014,52(10):6257–6271.
Rahmat R, Kamel N, Yahya N: Principle subspacebased signature verification technique. Innovative Technologies in Intelligent Systems and Industrial Applications, 2009. CITISIA 2009 2009, 317–321.
Loizou CP, Murray V, Pattichis MS, Pantziaris M, Nicolaides AN, Pattichis CS: Despeckle filtering for multiscale amplitudemodulation frequencymodulation (AMFM) texture analysis of ultrasound images of the intimamedia complex. Int J Biomed Imaging 2014, 2014: 13.
Lee JS: Digital image enhancement and noise filtering by use of local statistics. IEEE Trans Pattern Anal Mach Intell 1980,PAMI2(2):165–168.
Lei Z, Bao P, Xiaolin W: Multiscale LMMSEbased image denoising with optimal wavelet selection. IEEE Trans Circuits Syst Video Technol 2005,15(4):469–481.
Petrou M, Sevilla P: Image Processing: Dealing with Texture. New Jersey: Wiley; 2006.
Tay P, Acton S, Hossack J: Ultrasound despeckling using an adaptive window stochastic approach. Image Processing, 2006 IEEE International Conference on 2006, 2549–2552.
Jensen J, Svendsen N: Calculation of pressure fields from arbitrarily shaped, apodized, and excited ultrasound transducers. IEEE Trans Ultrason Ferroelectr Freq Control 1992,39(2):262–267.
Jensen J: Field: A program for simulating ultrasound systems. Med Biol Eng Comp 10th NordicBaltic 1996, 4: 351–353.
The RF ultrasound data 2014. [http://ultrasonics.bioengineering.illinois.edu/data_patient.asp ]
Acknowledgements
The authors would like to thank the Ultrasonic Imaging Laboratory at University of Illinois at UrbanaChampaign for providing validation data and the anonymous reviewers for their effort and constructive comments which helped in properly addressing the different issues and resulted in stronger proof of the proposed approach. This research work is funded under the ShortTerm Internal Research Fund (STIRF) Grant Scheme (0153AAC68), awarded by the Universiti Teknologi Petronas.
Author information
Authors and Affiliations
Corresponding author
Additional information
Competing interests
The authors declare that they have no competing interests.
Authors’ contributions
NY and NSK contributed to development and formulation of the SDC as well as the experiment design for validation and assessment of the SDC. AMS participated in statistical analyses of the results and critical revision of the manuscript for intellectual content. All authors read and approved the final manuscript.
Authors’ original submitted files for images
Below are the links to the authors’ original submitted files for images.
Rights and permissions
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
About this article
Cite this article
Yahya, N., Kamel, N.S. & Malik, A.S. Subspacebased technique for speckle noise reduction in ultrasound images. BioMed Eng OnLine 13, 154 (2014). https://doi.org/10.1186/1475925X13154
Received:
Accepted:
Published:
DOI: https://doi.org/10.1186/1475925X13154
Keywords
 Speckle reduction
 Denoising
 Signal subspace
 USDSAI