Hyperspectral venous image quality assessment for optimum illumination range selection based on skin tone characteristics
© Shahzad et al.; licensee BioMed Central Ltd. 2014
Received: 26 May 2014
Accepted: 28 July 2014
Published: 3 August 2014
Subcutaneous veins localization is usually performed manually by medical staff to find suitable vein to insert catheter for medication delivery or blood sample function. The rule of thumb is to find large and straight enough vein for the medication to flow inside of the selected blood vessel without any obstruction. The problem of peripheral difficult venous access arises when patient’s veins are not visible due to any reason like dark skin tone, presence of hair, high body fat or dehydrated condition, etc.
To enhance the visibility of veins, near infrared imaging systems is used to assist medical staff in veins localization process. Optimum illumination is crucial to obtain a better image contrast and quality, taking into consideration the limited power and space on portable imaging systems. In this work a hyperspectral image quality assessment is done to get the optimum range of illumination for venous imaging system. A database of hyperspectral images from 80 subjects has been created and subjects were divided in to four different classes on the basis of their skin tone. In this paper the results of hyper spectral image analyses are presented in function of the skin tone of patients. For each patient, four mean images were constructed by taking mean with a spectral span of 50 nm within near infrared range, i.e. 750–950 nm. Statistical quality measures were used to analyse these images.
It is concluded that the wavelength range of 800 to 850 nm serve as the optimum illumination range to get best near infrared venous image quality for each type of skin tone.
Hyperspectral and multi spectral imaging are well established systems used for remote sensing, satellites imaging, agriculture, physics and military applications. Recent years, these systems grasped attention of researchers in the field of biomedical imaging, especially where the standard imaging techniques fail to provide the desired outcomes . Hyperspectral sensor captures the spectral information of an entire field of view for each band (or wavelength) storing the collected information as a data cube which contains spatial information for each wavelength or band, depending on the characteristics of the system used . Hence, high spectral resolution will provide the ability to analyze data on wavelength or sub-wavelength scale. Furthermore, it also provides the ability to see beyond the visible range in order to acquire more information about the scene. For the case of skin and veins detection, hyper spectral venous imaging allows to look deeper below the skin in near infrared (NIR) range to find the best wavelength range for better contrast between skin and veins. This paper presents the analysis of hyperspectral data for the pre-selection of wavelength ranges suitable for the illumination in the process of veins localization.
Intravenous (IV) catheterization is the first and one of the most important steps in majority of medical treatments. Studies have reported that around 80% of all hospitalized patients and several outpatients need IV catheterization for blood sampling, medication delivery, transfusion and infusion of fluids to the patients [3, 4]. The localization of veins is a tedious process which has to be done by medical practitioners prior to IV catheterization. Veins are usually localized by sight and/or feelings by pressing (with fingers) the targeted area of patient’s body, mostly forearm and hand. These methods of veins localization are approximate, depends on practitioner skills and often fail for the patients whose veins are neither easily visible nor be localized with feelings. The probability that a patient have veins non-visibility problem is high for people with dark skin tone, young children (especially infants), obese and dehydrated patients. The veins non-visibility problem leads to multiple attempts of catheter insertion which cause unnecessary pain and trauma to the patients. Studies have reported that failure to first attempt in general population is around 12 to 26% and it is worse in children 24 to 54% [5–7].
There are many devices available to help medical staff to better visualize the non-visible veins of patients. These devices use one of these techniques: Trans-illumination, Photo-acoustic, Ultrasound and NIR imaging. All these techniques have different advantages and drawbacks but NIR imaging is considered the most suitable for veins localization in catheterization process [8, 9]. Indeed, NIR imaging uses non-ionizing light rays to penetrate deep inside skin tissues to acquire the image of venous structure. In electromagnetic spectrum, there is a low absorption window within NIR region, in which light can penetrate deeper inside skin tissues. This is due to the low absorption spectra of hemoglobin, oxy-hemoglobin and water which are main absorbers of radiations in skin. The span of this low absorption window is from 750 to 950 nm within NIR region . Different NIR imaging devices use different wavelength or range of wavelength out of this low absorption window to illuminate the target site [11–14]. In  a NIR imaging method for subcutaneous veins localization is reported. In this work the illumination range used was spanning the full NIR low absorption window i.e. 740–950 nm (7 wavelengths with 8 LEDs for each). This lighting system can be used on a bench top system which has no power and space limitation but not suitable for a battery powered portable systems. In  no prior classification of skin is done with respect to skin tones. A monochrome camera with CMOS sensor was used to acquire images. For illumination 6 different wavelengths (740,770,810,840 and 910 nm) were used in 63 different combinations. No conclusion on final selection of illumination range was made.
This paper discusses the outcome of the analysis done on hyperspectral images for optimum illumination selection for the four classes of skin tones. A dataset of 80 subjects is made by scanning the forearm of each subject with a hyperspectral camera setup (detailed in next section). As the analysis studies the skin tone, the subjects are separated equally into four different classes based on the luminance measurement of their skin, obtain from a chromameter. To define optimum ranges suitable for illumination, NIR window is divided into four sub-bands (50 nm each) and mean images are formed for each sub-band. These images are analyzed using mean square error (MSE) and universal image quality index (Q) to define the best wavelength ranges where the image quality is highest when compared to the mean reference image.
Hyperspectral image acquisition setup
Spectral DAQ software from Specim® was used to set the camera parameters and data acquisition. The angle for mirror scanner was set in the way that it should scan only the forearm region of the subject, allowing minimizing the data size of each hyper-spectral cube for each patient. The scanning time of hyperspectral image for one subject is about 40s. It depends on the selected scanning range of mirror scanner used with Specim VNIR Image sensor. Our targeted area was the fore arm region of subject, hence around 21° of scanning range (which covers the region from elbow to the tip of subjects fingers) is selected. The distance of mirror scanner is set to be about one meter from the subject’s position. The spectral images acquired are in the form of data cubes which are saved in environment for visualizing images (ENVI) compatible formats to be further processed. MATLAB is used to read the raw data cubes using ENVI toolbox.
A total of 80 volunteers were recruited for the data acquisition process. Most of these volunteers were students having body mass index (BMI) in healthy range .On the basis of BMI, 51 subjects have classified in healthy range, 7 in underweight, 15 in overweight and 7 in obese category .
Skin tone classification
Quality analysis of mean images
A mean NIR image (Im) was created for each subject, by taking mean of bands over a spectral range from 750 - 950 nm. This spans the whole NIR low absorption window. The low absorption window is the range of wavelength from 750 to 950 nm in which main absorbers of skin have low absorption coefficient, which allows the radiations (NIR light) to penetrate deeper in the skin tissues . These mean NIR images serve as reference images since the contrast per pixel (cpp) for these images is higher than the mean images obtained in other spectral regions. Moreover, the noise, present in each band (of 2 nm) is reduced in the mean image due to its highly random nature. The idea is to find the optimum illumination range within NIR region. There are no hard boundaries between bands and the spectrum seems continuous. Four sub-mean images are created by taking mean of bands in the following ranges:
Im: mean 750-950 nm
Image-1: mean 750-800 nm
Image-2: mean 800-850 nm
Image-3: mean 850-900 nm
Human subjects are unable to distinguish between image quality of all sub-mean images since they look quite similar.
Contrast between veins and skin tissues cannot be determined fairly based on the human visual system.
The acquired NIR images are fed to veins detection algorithm which provide an objective assessment observation.
Objective quality measurement is important for machine vision applications. Mathematical measures are used to measure and compare the image quality w.r.t. reference images. The factors, like mean square error (MSE) and Universal Image Quality Index (Q) are widely used to calculate the quality of images with a reference image which is considered the best image of the scene . In this work, these two factors are chosen to find out the best range of wavelengths on which one can have good quality image.
Through this work, it is determined that the image (image-2) which was made with the mean of bands in range 800-850 nm from multispectral data has the best quality. The measure like MSE, PSNR and universal image quality (Q) is found best for this range. Keeping in mind the image acquisition setup, it is concluded that the best quality image is obtained in the spectral range of 800 – 850 nm. These findings will serve the basis of optimum illumination selection for a NIR system for subcutaneous veins localization. For this system, our choice will be the LEDs with a central wavelength lying within 800-850 nm range. With this optimized illumination it is anticipated that better quality and high contrast NIR images can be obtained.
Near infrared imaging is proven as a best technique in veins visualization systems. Optimized illumination is crucial for the low power, portable and wearable systems. A hyperspectral analysis has been done in this work to get the best range of illumination wavelength, for the better image quality and higher contrast. The NIR region is divided into 4 sub regions and means images were formed from these sub-regions. These mean images are analysed with objective quality measurement factors like MSE and Universal Image quality Factor 'Q’ to define best range for illumination. It has been found that the range 800 to 850 nm provides best image quality. For Fair skin, the image quality is slightly better in the range of 750 to 800 nm. For darker skin the overall range considered suitable is 800 to 850 nm. This infers that if we choose the illumination within this range, we can have better image quality and higher contrast for venous images.
Future work is to make the prototype system with optimized illumination and compare the results with a system with generalized illumination.
Sources of funding
This work was supported by Centre for Graduate Studies (CGS), Universiti Teknologi PETRONAS, Malaysia under the Graduate Assistantship (GA) scholarship scheme and Research and Innovation Office (RIO) of Universiti Teknologi PETRONAS under the university internal fund (URIF) scheme.
- Freeman J, Downs F, Marcucci L, Lewis EN, Blume B, Rish J: Multispectral and hyperspectral imaging: applications for medical and surgical diagnostics. Proceedings of IEEE International Conference of Engineering in Medicine and Biology Society: 30 Oct- 2 Nov, 1997; Chicago IL. vol. 2 1997, 700–701.Google Scholar
- Kerekes JP, Schott JR: Hyperspectral imaging systems. In Hyperspectral data exploitation: Theory and applications. Edited by: Chein-I C. Wiley: Wiley; 2007:19–45.Google Scholar
- Jacobson AF, Winslow EH: Variables influencing intravenous catheter insertion difficulty and failure: an analysis of 339 intravenous catheter insertions. Heart Lung 2005, 34: 345–359.View ArticleGoogle Scholar
- Rivera A, Strauss K, Zundert AV, Mortier E: Matching the peripheral intravenous catheter to the individual patient. Acta Anaesthesiol Belg 2006, 58: 19–25.Google Scholar
- Lapostolle F, Catineau J, Garrigue B, Monmarteau V, Houssaye T, Vecci I, Treoux V, Crocheton N, Adent F: Prospective evaluation of peripheral venous access difficulty in emergency care. Intensive Care Med 2007, 33: 1452–1457.View ArticleGoogle Scholar
- Juvin P, Blarel A, Bruno F, Desmonts JM: Is peripheral line placement more difficult in obese than in lean patients? Anesth Analg 2003, 96: 1218–1218.View ArticleGoogle Scholar
- Palefski SS, Stoddard GJ: The infusion nurse and patient complication rates of peripheral-short catheters: a prospective evaluation. JIN 2001, 24: 113–123.Google Scholar
- Paquit V, Price JR, Seulin R, Meriaudeau F, Farahi RH, Tobin KW, Ferrel TL: Near-infrared imaging and structured light ranging for automatic catheter insertion. Proc SPIE Med Imaging 2006, 6141: 61411T.View ArticleGoogle Scholar
- Paquit V, Price JR, Mériaudeau F, Tobin JKW, Ferrell TL: Combining near-infrared illuminants to optimize venous imaging. Proc SPIE Med Imaging 2007, 6509: 65090H.View ArticleGoogle Scholar
- Strangman G, Boas DA, Jeffrey PS: Non-Invasive Neuroimaging Using Near-Infrared Light. Biol Psychiatry 2002, 52: 679–693.View ArticleGoogle Scholar
- Zeman HD, Lovhoiden G, Vrancken C: Prototype vein contrast enhancer. Proc. SPIE Advanced Biomedical and Clinical Diagnostic Systems II 2004, 5318: 39–49.View ArticleGoogle Scholar
- Li X, Liu T, Wang Y, Yuan C, Deng S: Development of an image acquisition system for hand vein based on an embedded system. International Symposium on Photoelectronic Detection and Imaging 2009, 73810A.Google Scholar
- Crisan S, Crisan T: A low cost vein detection system using near infrared radiation. IEEE Sensors Applications Symposium 2007, 1–6.Google Scholar
- Fuksis R, Greitans M, Nikisins O, Pudzs M: Infrared imaging system for analysis of blood vessel structure. Electron Electrical Eng 2010, 1: 45–48.Google Scholar
- About BMI for Adults [http://www.cdc.gov/healthyweight/assessing/bmi/adult_bmi/index.html]
- Nafiu OO, Burke C, Cowan A, Tutuo N, Maclean S, Tremper KK: Comparing peripheral venous access between obese and normal weight children. Pediatr Anesth 2010,20(2):172–176.View ArticleGoogle Scholar
- Hani AFM, Prakasa E, Nugroho H, Asirvadam VS: Implementation of fuzzy c-means clustering for Psoriasis Assessment on lesion erythema. IEEE Symposium on Industrial Electronics and Applications (ISIEA) 2012, 331–335.Google Scholar
- Wang Z, Alan CB Synthesis Lectures on Image, Video, and Multimedia Processing 2.1. Modern image quality assessment 2006, 1–156.Google Scholar
- Wang Z, Alan CB: A universal image quality index. IEEE Signal ProcessLetters 2002,9(3):81–84.View ArticleGoogle Scholar
- Wang Z, Alan CB: Mean squared error: love it or leave it? A new look at signal fidelity measures. IEEE Signal Process Mag 2009,26(1):98–117.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.