Review | Open | Published:
Neurovascular coupling: in vivo optical techniques for functional brain imaging
BioMedical Engineering OnLinevolume 12, Article number: 38 (2013)
Optical imaging techniques reflect different biochemical processes in the brain, which is closely related with neural activity. Scientists and clinicians employ a variety of optical imaging technologies to visualize and study the relationship between neurons, glial cells and blood vessels. In this paper, we present an overview of the current optical approaches used for the in vivo imaging of neurovascular coupling events in small animal models. These techniques include 2-photon microscopy, laser speckle contrast imaging (LSCI), voltage-sensitive dye imaging (VSDi), functional photoacoustic microscopy (fPAM), functional near-infrared spectroscopy imaging (fNIRS) and multimodal imaging techniques. The basic principles of each technique are described in detail, followed by examples of current applications from cutting-edge studies of cerebral neurovascular coupling functions and metabolic. Moreover, we provide a glimpse of the possible ways in which these techniques might be translated to human studies for clinical investigations of pathophysiology and disease. In vivo optical imaging techniques continue to expand and evolve, allowing us to discover fundamental basis of neurovascular coupling roles in cerebral physiology and pathophysiology.
Cerebral blood flow (CBF) is vital to the normal brain. The average CBF in humans is approximately 55 ml per 100 g of brain tissue per minute. Between 700 and 800 ml of the normal cardiac output of 5 l/min is used to maintain membrane potentials and reverse ion fluxes from action potentials in the brain. More blood flow is needed when active neural processes increase energy requirements. Blood flow correlates precisely with the regional energy utilization in the brain matter. That is, the cerebral circulation must maintain a constant and adequate blood flow for the brain to function. The cerebrovascular system has several mechanisms to ensure the correct blood circulation to meet specific energy demands and prevent harmful fluctuations due to changes in arterial pressure. This autoregulation of the cerebrovascular system involves several different blood vessels with contractile elements, such as layers of smooth muscle cells, which may control the diameter of the vessel and the CBF. In contrast to other organs, the parenchymal flow in the brain is controlled entirely outside of the organ. Two-thirds of the vascular resistance in the brain is due to large cerebral arteries and pial vessels; non-pial vessels are responsible for the remaining one-third of vascular resistance.
Furthermore, the brain capillary system is highly heterogeneous. The brain uses an additional mechanism, hyperaemia, to increase the flow of blood to the regions in which neurons are active. According to this mechanism, neurons and astrocytes directly regulate the local blood flow within the capillaries, resulting in local neurovascular coupling. Initially, it was proposed that the local increase in blood flow upon neuronal activation was caused by a metabolic signal, such as a decrease in either the O2 or glucose concentration or accumulation of CO2. However, recent experiments have shown that a neurotransmitter-mediated mechanism that involves astrocytes is responsible for local flow regulation. According to this, upon glutamate activation, nitric oxide (NO) and arachidonic acid derivatives are released from neurons and astrocytes, respectively, upon glutamate activation. Although this hypothesis was initially debated, there are extensive studies showing that the final effect of these substances depends on the local O2 concentration and differs between brain regions[6, 7]. Furthermore, this mechanism of local vascular flow regulation may affect not only smooth-muscle-dependent arteriole contraction but also the intrinsic contraction of the pericytes, which are supporting cells of capillaries, playing a key role in both neurovascular coupling and the maintenance of the blood–brain-barrier (BBB)[6, 7].
Neurovascular coupling and the correlation between neural activity and the CBF can be exploited to examine both normal brain function and the pathophysiology of the cerebrovascular system. Optical imaging techniques provide the necessary tools to accurately detect and analyze regional hyperemic changes. This review will present several important optical imaging techniques that are used for in vivo measurement of functional hyperemia, including 2-photon laser scanning microscopy (TPLSM), laser speckle contrast imaging (LCSI), voltage-sensitive dye imaging (VSDi), functional near-infrared spectroscopy (fNIRS) and functional photoacoustic microscopy (fPAM). Due to space constrictions, we refer to other reviews addressing functional magnetic resonance imaging (fMRI), intrinsic signal optical imaging (IOS)[9, 10] and other techniques[11, 12]. In the conclusion of this review, we discuss the possible applications of these in vivo techniques in basic and clinical human studies, especially in techniques that can be administered intraoperatively.
Examination of cerebral neurovascular and neurometabolic functions using optical imaging techniques
The question of how to examine the relationship between neural, vascular, hemodynamic and metabolic responses using optical imaging techniques has not been answered comprehensively. The relationship between optical signals of cerebral blood oxygenation (SO2) and neuronal activity is complex. The response depends on the coupling of neurovascular functions to the corresponding cerebral neurometabolic functions of tissues and cells. Optical techniques to probe functional brain activity are known to produce reliable results. Furthermore, the specific characteristics of these optical signals are sensitive to the corresponding cerebral vascular physiology and structure.
Different kinds of optical signals reflect different neuronal activity in different parts of the brain: changes in blood oxygenation, local blood circulation, tissue concentration of oxy- and deoxyhemoglobin. Neural and glial[13–15] cell volume changes are frequently associated with alterations in different components of the optical signals. Since brain metabolism, local blood circulation and cell volume (swelling) reflect neural activity, we can reasonably assume that the different kinds of brain optical imaging can be applied to help understand the localization of neural activity.
Figure 1 provides a schematic representation of the neurovascular coupling model[16, 17] most commonly used to characterize the vasomotion and metabolic responses in the brain. This model assumes that the optical signals directly correspond to changes in the cerebral metabolic rate of oxygen (CMRO2). Several researchers have proposed a set of equations to describe the response of the arteries and capillaries in neurovascular coupling[18–20]. Neurons can be triggered by unique stimulation events to produce corresponding evoked potential signals (EPSs). These EPSs are directly correlated with the responses of the vascular and oxygen transport models.
The response of the vascular model also affects the CBF response, which drives changes in cerebral blood volume (CBV) and total of hemoglobin (HbT) (includes oxyhemoglobin (HbO) and deoxyhemoglobin (HbR)). Existing optical imaging techniques for probing hemodynamic changes are based on either the absorption model (i.e., TPLSM, fNIRS and fPAM), the speckle model (i.e., LCSI) or activity-dependent fluorescence model (VSDi) (Table 1). The optical absorption model can be applied to experimental measurements to predict HbR and HbO directly via the oxygen transport model because of the unique contrast witnessed in optical signals. In addition, the oxygen level in the blood vessels can be modeled using the principles of mass balance based on the CBF changes predicted from the vascular model for oxygen consumption metabolic coupling. On the other hand, the speckle model is directly related to changes in CBF because speckles are the result of interaction of coherent light with the scatters (mainly red blood cells (RBCs)) in the blood vessels. In this section, we will describe how each of these imaging techniques works, as shown in Table 1, and as well as elaborate upon their applications.
2-photon laser scanning microscopy
Several advances in microscopy techniques have allowed examination of capillary flow based on the gaps between RBCs during their movement along the blood vessel. These techniques initially involved the use of a CCD detector, then fluorescence microscopy and, finally, confocal laser scanning microscopy[50, 51]. However, the strong light-scattering associated with brain tissue that occurs when using these techniques prevented cellular and subcellular resolution in the intact brain. High-resolution imaging of brain cell morphology and function in vivo was not possible until TPLSM (Figure 2A) was developed by Denk et al.. The description of the physical properties of the two-photon absorption phenomenon is beyond the scope of this review. We thus refer to previous literature for a detailed description[52, 53].
TPLSM has been employed to measure vascular changes in single vessels in vitro[54–59] and in vivo[60–64]. Neuronal activity can be imaged simultaneously using calcium indicators. The ability to image both neuronal and glial calcium transients is important for neurovascular researches because both neurons and glia have been implicated in vasculature communication[54, 56, 57, 63, 66]. Furthermore, functional changes in calcium levels can also be detected in arteriolar smooth muscle, and these changes appear to coincide with active changes in vascular diameter. Therefore, TPLSM can be used to observe neurons, glia and the vessels which control blood flow through a combination of calcium imaging and direct blood flow measurements (Figure 2B). This approach is expected to aid in the identification of cellular elements which control vascular responses under normal conditions and, possibly, the mechanisms which lead to failure under pathological conditions.
As well known, the essence is that a fluorophore – Ca2+ − or voltage-sensitive dye, fluorescent proteins, or other sort of fluorescence markers is excited by the absorption of two photons at the same time, each of which brings only part of the energy required for fluorophore excitation. Different fluorophores may have different emission wavelength that is allow to generate separated 2D and 3D images for the different sorts of cells.
Two of the main advantages of TPLSM over single-photon microscopy techniques are its higher resolution and ability to image greater depths (up to 600 μm) below the cortical surface[62, 68]. In the 2-photon absorption process, the fluorescence signal depends nonlinearly on the amount of excitation by light. Therefore, fluorescence generation is localized to the focal spot. In contrast to confocal microscopy, resolution is achieved by limiting the excitation volume at the focal point, instead of limiting the detection of a photon with a pinhole. This technique minimizes tissue damage caused by photochemical interactions. Furthermore, the excitation light required to combine two photons to excite a fluorophore has a longer wavelength (700–1,000 nm) compared with the light used in single-photon microscopes. Therefore, the resulting scattering is significantly reduced, and greater depths can be reached. However, because the resolution scales inversely with the wavelength, the resolution decreases progressively as depth increases.
Recent advances in TPLSM have allowed simultaneous measurement of the activity of single, visually identified blood vessels and visualization of neuronal and glial activity using calcium indicators[11, 12, 23]. TPLSM can be employed to calculate vascular dynamics at the single-vessel level by measuring the diameter of the vascular lumen and RBC speed (Figure 2C). A common technique to visualize blood vessels utilizes space-occupying high-contrast molecules, such as fluorescent dye-dextran conjugates (e.g., fluorescein isothiocyanate, FITC). Continuous line scans perpendicular to the vessel axis can be conducted to measure changes in vessel diameter over time, and RBC speed can be calculated from repeated line scans along the axis of the vessel lumen. The movement of RBCs over the background fluorescence of dextran-containing fluid generates streaks in the space-time stack image. The blood flow can then be calculated as a product of the vessel cross-section area and RBC speed.
Laser speckle contrast imaging (LSCI): a powerful tool for probing cerebral blood flow
LSCI is a very practical and economic imaging tool used to visualize fine microvasculature and perfusion dynamics (Figure 3A). Physically, LSCI is a powerful imaging method based on light absorption and scattering and focuses on interpreting the speckle pattern phenomenon - intensity patterns produced by the interference of a set of light waves.
LSCI is a non-scanning imaging modality that is capable of monitoring a wide field of view with a high temporal resolution. LSCI utilizes RBCs to induce speckle contrast of laser light, which is inversely related to CBF. The details of the theory and methodology underlying the LSCI technique can be found in previous reports[26, 28, 29]. However, LSCI can only be used to generate information from single vessels that are located on or close to the cortical surface due to its limited penetration depth and poor depth resolution[26, 28, 29]. Nevertheless, LSCI is a popular imaging tool because of its ability to directly assess cerebral flow velocities with an excellent resolution, without the need for an exogenous contrast agent.
The LSCI technique has been applied by many researchers to study the spatiotemporal evolution of cortical hemodynamic patterns in response to functional stimulation. Li et al. developed an LSCI system with a high resolution (6.7 μm × 6.7 μm) to characterize the CBF and corresponding vasomotor response that occurred in response to electrical stimulation of the rat peripheral trigeminal nerve (Figure 3B). The trigeminal neural system is an important factor in the pathogenesis of migraines, and understanding the relationship between this system and migraine via the LSCI technique will be useful for further examining disorders of the neurovascular system, especially for exploring their underlying cellular and molecular basis. Moreover, Bouchard et al. proposed an LSCI system with a lower overall cost compared with the LSCI system currently employed in laboratory settings. This low-cost LSCI system enables simultaneous visualization of HbT, HbO2 and Hb dynamics within single vessels in response to forepaw stimulation. LSCI has proven to be an important tool for neuroscience research, presenting excellent spatial and temporal resolutions and capabilities that extend beyond the visualization of cerebral functional and structural hemodynamic patterns.
There is great interest in studying neurovascular responses of animals in an awake state because recent studies have shown that anesthesia may alter the responses of neural circuits. Most of the LSCI systems that have been developed to monitor CBF can be used only on anesthetized animals. Therefore, a novel LSCI device that can examine CBF changes in small anesthetized animals is needed. However, in designing such a device for use in freely moving animals, there are several major challenges: 1) the imaging device must be lightweight, and the equipment response must be rapid; 2) anti-motion artifacts need to be considered in freely moving animals; and 3) the wireless charge and signal transmission must be robust, especially for behavioral studies. How to overcome the above-mentioned issues using technology to minimize LSCI systems into a reliable device for use in freely moving animals is an issue of great interest and importance. Recently, Miao et al. proposed a miniature LSCI imager that weighs approximately 20 g and can image CBF changes in freely moving animals. This miniature LSCI imager includes an image sensor, a light source, an optical lens and a data acquisition board and can generate real-time CBF images with a high spatiotemporal resolution in freely moving animals.
Murari et al. proposed an integrated miniature LSCI microscope that allows several optical techniques to be applied (i.e., reflectance, spectroscopy, speckle and fluorescence imaging) (Figure 3C-3D). This novel lightweight LSCI microscope can image cortical CBF changes in mobile rats. In the future, widespread use of the LSCI technique to image awake and active animals will provide critical information about currently unknown cerebral neuroscience issues in these animals (Figure 3D). Moreover, this technique will offer the advantage of allowing physiological processes and structures to be studied in freely moving animals.
In vivo VSDi: direct imaging of neural activity and neurovascular coupling
Over the past several decades, various imaging approaches have been developed to investigate both individual and network neuronal properties in living organisms. Imaging techniques such as fMRI and IOS are used to detect changes in brain tissue oxygenation levels in response to neural activity. VSDi (Figure 4A) can achieve a much higher spatial and temporal resolution compared with these techniques. This approach is particularly useful for generating functional images of neural circuit dynamics in superficially located brain structures such as the neocortex. VSDi is based on the electrochemical properties of the neural membrane. Voltage-sensitive dye molecules change the level of light absorbance or fluorescence in proportion to the membrane potential. Therefore, they allow visualization of neural activity with a high temporal (less than 1 millisecond) resolution.
In VSDi, the optical signal is determined by two components: 1) the amplitude of the changes in the membrane potential and 2) the size of the membrane surface stained with the voltage-sensitive dye. In the neocortex, the dendrites cover a larger area than the neuronal soma; therefore, the fluorescence signal mostly reflects postsynaptic potentials, rather than action potentials. The response time of the voltage-sensitive dye molecules is quite short but is longer than the time over which a membrane potential changes. For some probes, the response time is in the microseconds and shows a linear correlation with the membrane voltage. The two main classes of voltage-sensitive dyes can be differentiated based on their chemistry: the first class of dyes undergoes changes in absorbance and the second class changes in fluorescence.
The excitation and emission wavelengths of the two main classes of voltage-sensitive dyes are also different. For in vivo research, these wavelengths should show minimal overlap with hemoglobin absorption; however, this consideration is not critical for in vitro imaging. During in vitro imaging, the sample is illuminated by a monochromatic excitation light, and the fluorescent signal is recorded by a CCD camera or a photodiode array through appropriate optic filters. The temporal resolution of VSDi is limited by the technical properties of the imaging setup, whereas the spatial resolution is determined by the optical scattering of the brain tissue and usually cannot exceed a few tens of microns. Spatial resolution can be improved by applying VSDi in combination with either confocal laser scanning microscopy for in vitro experiments or multiphoton microscopy for both in vivo and in vitro experiments.
Due to chemical toxicity, VSDi is not suitable for use in clinical practice. However, VSDi is a powerful technique for animal experiments, including visual, auditory and somatosensory[48, 81] research (Figure 4B). This field of brain imaging will undoubtedly benefit from the development of genetically encoded fluorescent voltage-sensitive probes, such as voltage-sensitive dye proteins (VSDP). These genetically engineered proteins allow the analysis of rapid electrical signals in neural populations in vitro as well as in vivo. Encoded fluorescent probes can also be targeted to selected neurons in different parts of the brain.
Neurovascular photoacoustic microscopy
Photoacoustic (PA) imaging is an optical absorption-based hybrid imaging technique that combines the advantages of optical and ultrasound techniques to provide a high optical absorption contrast and ultrasonic spatial resolution at a penetration depth of up to several centimeters (Figure 5A). This technique visualizes the high optical absorption contrast of biological tissues, instead of the low acoustic scattering contrast. The spatial resolution of PA imaging is determined primarily by the characteristics of the ultrasound transducer, such as center frequency, bandwidth, and numerical aperture, rather than by optical diffusion (as in optical imaging). Ultrasonic detection improves the spatial resolution of PA imaging up to few microns and can also improve it up to even one micron. Moreover, PA imaging is uniquely sensitive to blood in vivo due to the high intrinsic optical absorption of blood relative to other biological tissues. PA imaging of in vitro and in vivo blood samples has shown that this technique can assess relative changes in the concentrations of both oxyhemoglobin and deoxyhemoglobin. Using appropriate and distinct PA excitation wavelengths, it has been demonstrated that changes in HbT, CBV and SO2 can be probed independently. For example, photoacoustic signals generated at 560 or 600 nm are sensitive to SO2 changes, whereas the signals at 570 nm, an isosbestic point of the molar extinction spectra of oxy- and deoxy-hemoglobin, are instead insensitive to SO2 but on the other hand can be used to image changes in HbT and CBV.
A reflection-mode PA microscopy (PAM) technique involving dark-field illumination has been proposed by Maslov, et al. (Figure 5A). PAM can be further divided into optical- (OR-PAM) and acoustic-resolution PAM (AR-PAM), depending on the resolution and penetration preference for an optical or ultrasonic focus in experiments, respectively. OR-PAM provides an imaging resolution at the cellular level (i.e., ranging from a few hundred nanometers to a few micrometers). At depths beyond the optical diffusion limit and up to a few millimeters, AR-PAM has achieved a lateral resolution of 45 μm and an imaging depth of 3 mm. Dark-field PAM is able to monitor brain activity through the intrinsic blood hemodynamic response and can distinguish between increased blood oxygenation and a decreased blood volume via spectroscopic techniques. The ability of PAM to track blood oxygenation in the mouse brain was shown using a technique that controls hypoxic and hyperoxic changes. Recently, the PAM technique has been applied to subcutaneous vasculature imaging, breast tumor detection, drug delivery[90, 91], tissue temperature monitoring and oxygenation monitoring in blood vessels[36, 85].
Based on the advantages of PAM and excitation wavelength optimization, Liao et al. have successfully applied acoustic-resolution functional photoacoustic microscopy system (fPAM) to probe changes in CBV (Figure 5B) and SO2 (Figure 5C) without a labeling agent under electrically stimulation. This technique is ideal for in vivo imaging of changes in HbT, CBV and SO2 in single cerebral cortex vessels in small animals (Figure 5D). The fPAM technique has a high resolution and can achieve a satisfactory imaging quality in small animals with good penetration depth. fPAM can be used visualize dynamic and functional activities in the brain because of the intrinsic optical absorption contrast of blood in brain tissues and the ability to distinguish between increased blood oxygenation and a decreased blood volume via spectroscopic techniques. Details of the method for selecting the proper wavelength for the intrinsic contrasting of optical absorptions have been described and discussed in a previous report. This technique has been used to visualize 1) cerebral hemodynamic responses to neuronal activities induced by electrical stimulation in rats[40, 86] (Figure 5E), 2) cerebral functions based on changes in HbT, CBV and SO2 in the rat brain and 3) transcranial (through the cranium, through the skull) functional cerebral hemodynamic changes in single blood vessels[37, 39]. In addition, Yao et al. proposed a method to measure transverse blood flow speed (Figure 5F) and velocity (Figure 5G) with directions that employs PA Doppler broadening of the signal bandwidth.
These authors confirmed that their method reliably measures the blood flow in the microvasculature of the mouse ear. These results suggest that all of the parameters which are needed for CMRO2 estimation (i.e., the functional changes in cerebral HbT, CBV, SO2 and blood flow) are measurable via the current fPAM technique in one single setup. One future challenge is the exact quantification CMRO2via fPAM, as this will be very useful in clinical applications. Thus, transcranial monitoring of cerebral blood oxygenation by multi-wavelength PA system was reported recently. It was demonstrated that the oxygenation-related PA signal is detectable through the sheep’s intact scalp and skull, so the same technology can be performed in humans. The skull strongly absorbs and scatters light, but the photon recycler increases the light transmittance through the bone. This promising technology had demonstrated the feasibility of photoacoustic tomography through the adult human skull.
Another future challenge is to design a “portable real-time fPAM system” for freely moving animals. Such a technique would be safe and would provide both a high contrast and high spatial resolution. Zemp’s group recently designed a portable PAM with an optical-level resolution and demonstrated the performance of this device in both in vitro and in vivo studies. This handheld PAM retains the original merits of the previous OR-PAM system and allows real-time imaging to be performed. This fiber based handheld PAM design has the potential to be used in applications of freely moving animals.
Functional near-infrared spectroscopy
The fNIRS technique relies on specific laser wavelengths (usually in the range of 700 – 1,700 nm) penetrating through the scalp to enable non-invasive measurement of changes in brain activity[43, 44]. Although the spatial resolution of fNIRS is lower than that of other optical imaging techniques, fNIRS offers distinct advantages over fMRI, such as much lower cost, portability, ability to make continuous and long term measurements, and the ability to separate the relative concentrations of SO2, Hb and HbT when several proper wavelengths are simultaneously used. Previous studies have applied NIRS in functional imaging and cortical brain mapping based on the observation that regional CBF increases in response to a stimulus event. A variety of methods have been developed for fNIRS instruments, including the 1) continuous wave (CW), 2) time domain (TD) and 3) frequency domain (FD) methods, which have been applied in functional imaging to map motor, visual and resting-state connectivity in small animal models.
The CW method is used to measure changes in the incident light amplitude. The laser light source is emitted and illuminates a selected scalp surface position. After the light has passed through the skull near the brain surface, highly scattered and attenuated light can be observed. Therefore, light intensity changes can be measured from a few centimeters below the skull. The intensity changes at the selected light wavelengths can be used to calculate the concentrations of HbO and HbR based on the Beer-Lambert Law (BLL). Because a CW system can only detect the amplitude decay of light, the exact volume of tissue that the light has illuminated is unknown; therefore, the absolute values of the hemodynamic concentrations cannot be provided. In contrast, TD- and FD-fNIRS systems can estimate the path length of the NIR light from multi-distance sensors to derive the absolute values of hemodynamic concentrations. For example, time-resolved near-infrared spectroscopy (TR-NIRS) systems employ short, picosecond, laser pulses and a fast photon-counting detector to detect the flight time of photons as light is attenuated through various cerebral tissue layers (i.e., the skull). In addition, the TR-NIRS technique is comparable to IOS because of its’ ability to distinguish changes in absorption from changes in scattering.
A future challenge is to improve the fNIRS technique for use in clinical settings and studies, such as for rehabilitation or for brief evaluations before and after a trial of a particular therapy[98, 99]. For example, although examining an individual stroke survivor’s unique brain response might be one method to determine the appropriate rehabilitation program, the use of fNIRS may provide a better understanding of the brain reorganization and motor recovery that have occurred and could provide a new avenue for designing therapeutic rehabilitation strategies that are tailored to the individual. We expect many unique and significant discoveries to be made in the area of brain-based rehabilitation research. Due to its many advantages, including cost, convenience of noninvasive and real-time imaging, etc., clinical studies applying fNIRS are likely to find use and applications in the very near future.
Multimodal imaging techniques
The use of only one imaging technique to study cerebral functions provides a descriptive view of a single aspect of the underlying multi-faceted physiological processes. Multimodal imaging techniques provide a better way to study complex cerebral functions by allowing simultaneous measurement of multiple relevant physiological and biophysical parameters. Several multimodal fPAM techniques for neuroimaging have been proposed recently. For example, Wang et al. described a dual-modality microscope that combines PAM and fluorescence confocal microscopy (PA-FCM), designed for use in in vivo SO2 imaging. Tan et al. proposed a multimodal imaging system that combines laser-scanning optical-resolution photoacoustic microscopy (LSOR-PAM) with spectral-domain optical coherence tomography (SD-OCT) to obtain better structural information along with that of blood vessels. They demonstrated the feasibility of measuring in vivo metabolic changes in small animals. To achieve molecular-specific imaging with a high spatial resolution in deep tissue, Yakovlev, et al. combined molecularly specific stimulated Raman excitation with photoacoustic detection. They found that unique stimulated Raman photoacoustic waves can be triggered and detected using a traditional ultrasonic transducer to form an image, which follows the design of the established PAM technique.
The combination of different imaging techniques in the same session seems very promising. For example, PA that is agent–free imaging method, was successfully indicates glucose metabolism level. Fluorescently-labeled deoxyglucose analog, called 2-(N-(7-nitrobenz-2-oxa-1,3-diazol-4-yl)amino)-2-deoxyglucose (2-NBDG), which penetrates across the BBB, provided significant contrast for the photoacoustic imaging.
VSDi was successfully used for a neurovascular coupling studies in the epileptic research. Thus, simultaneous monitoring of the cerebral blood volume and VSDi demonstrated that an epileptic seizure consist of multiple dynamic multidirectional waves of membrane potential change that spread through a widespread network. It seems extremely promising that VSDi recently has been used in combination with LSCI to directly visualize cortical spread depression. LSCI was employed for the monitoring of the CBF changes while VSD was used for direct recording of the neural activity.
Another powerful imaging method is the combination of VSDi with 2-photon excitation effect. This combination allows depth-resolved optical recordings with high temporal resolution not only from anesthetized but also from awake animals. Theoretically, 2-photon excitation can be combined with many types of fluorescence probes, but it is possible to use such combination only in animal experiments due to the effects of fluorescence agents such as bleaching and cytotoxicity.
The fluorescent optical biomarker, indocyanine green (ICG), has been employed in combination with fNIRS. This method not only brings new data about neurovascular coupling but also may be of wide clinical interest, since ICG is non-toxic as and generally considered safe (and approved for human use by the Food and Drug Administration). ICG is already used in clinical applications, such as retinal imaging, and hence in combination with various imaging modalities, it may prove to be valuable in performing imaging studies at the bedside in patients.
Translation to human studies
Transcranial optical imaging remains challenging since optical scattering and absorption degrade the spatial resolution as well as the signal-to-noise ratio. Brain imaging based on fluorescence probes was first conducted in 1948 but was not considered an established method until recently. Fluorescence angiography provides a method to visualize perfusion during neurovascular operations, such as the treatment of angioma and the clipping of aneurysms. The fluorescent dye employed in this technique; usually ICG binds to plasma proteins and remains intravascular. It was successfully employed during neurosurgery for the vascular imaging. However, this dye fails to reveal relative microcirculatory flows or tissue perfusion. Other dyes, such as aminolevulinic acid (ALA)-induced protoporphyrin IX (PpIX), bind selectively to tumor cells to facilitate the resection of brain tumors. Angiogenesis is essential for brain tumor development therefore the changes in the local neurovascular coupling play important role in the tumor localization and anticancer therapy. Brain glucose metabolism, one of the most important factors in the neurovascular coupling, can be studied using specific fluorescent glucose substitutes. In clinical practice, 2-deoxy-2-[(7-nitro-2,1,3-benzoxadiazol-4-yl)amino]-D-glucose (2-NBDG) has been used frequently due to its strong fluorescence and low toxicity.
Advances in brain imaging can also contribute to the understanding of cognitive phenomena and neurological diseases, as these may correlate well with the changes in the neurovascular –coupling. Cerebral blood measurements are of great importance in the clinical setting especially in identification of progressive hypoperfusion before the onset of permanent brain damage. But the imaging of the CBF as well as local blood and tissue oxygenation did not exist in the clinical practice twenty years ago. Regional changes in CBF, local oxygenation and metabolism have been reported in epilepsy, depression, bipolar disorder, brain trauma and other types of the brain pathology. These measurements are not only useful in a variety of surgical procedures, such as during clipping of aneurysms or vessel bypasses to assess baseline blood flow but also during neural or glial tumor resection to assess postsurgical tissue viability. Such measurements provide information allowing the identification of motor, sensory and speech activation centers in the cortex in procedures that require functional localization. However, the application of optical imaging techniques in humans presents several difficulties.
Due to the spatial resolution limitations imposed by the thickness of the human skull and the dura mater, beyond the first few years of life, the use of transcranial optical imaging can be performed only on the exposed brain during the course of therapeutic neurosurgical interventions. Because calcium- and voltage-sensitive dyes are not approved for use in humans, most studies have relied on intrinsic signals. The IOS is based on the light absorption and reflection property changes that occur in neural tissue when activated. Although the spatial resolution (up to ~50 μm) of the IOS is pretty good for the human studies, IOS lucks the depth resolution in contrast with fMRI. Anyway, it is suitable for many functional studies that can utilize functional optical changes arising at the cortical surface. The temporal resolution of the IOS is better than that of fMRI in many cases. Furthermore, the affordable IOS setup is more affordable as it requires no additional modification of the operating room (OR). It may also be safe as no physical contact with tissue is required, thus avoiding potential risk of infection. Because light scattering and absorption are determined by the wavelength of the photon involved, the use of infrared and near-infrared light allows imaging at greater depths. For this reason, NIRS has been utilized extensively in clinical situations, including in non-invasive studies in infants with thin skulls. Because the NIRS apparatus is portable and tolerable to movement by the subject during image acquisition, this technique has proven to be successful for many applications, such as the localization of epileptic seizures with increasing local cortical blood circulation and in cognitive neuroscience research. Based on the light-scattering pattern produced upon neural activation, which is likely due to optical changes caused by ionic movements across the neuronal membrane, a similar imaging technique referred to as the event-related optical signal (EROS) has been used in numerous functional studies. Intraoperative imaging using EROS in awake patients during brain surgery has allowed characterization of stimulation-evoked epileptiform activity and the activation of Wernicke’s and Broca’s areas during language tasks[112, 115]. Based on initial reports as well as its high temporal resolution and real-time capabilities, LSCI might be an efficient method to image CBF intraoperatively, with minimal interference to the ongoing surgery.
Although PA imaging presents many advantages[116, 117] application of this technique is also faces several difficulties. For example, because physical contact is needed for efficient ultrasound transmission, use of a sterile sleeve to cover the transducer and sterile saline solution as a coupling medium are imperative. Nevertheless, because of its high resolution and ability to carry out multifunctional analysis of brain structure and metabolism, PAM appears to be a promising tool for measuring cerebral hemodynamic responses during neurovascular procedures.
It may take a while before a definitive technique for intraoperative neurovascular imaging is developed. Ideally, a suitable neurovascular imaging technique should have both high spatial and temporal resolution and should allow itself to be incorporated harmoniously within surgical techniques. The alternatives presently available allow compatible integration within the OR setting, precise anatomical resolution, or real-time accuracy, but not all three simultaneously. However, several technological advances suggest that an ideal system is possible in the future. Features such as fiber-optic and microscope miniaturization allow imaging techniques, previously exclusive to the bench top, to be made portable. The use of a single optical fiber as both the illumination point source and detection pinhole, in combination with tissue-specific fluorophores and handheld confocal systems, is currently being tested in neuroimaging and brain tumor surgery. A synergy between these high-resolution optical systems and the other systems reviewed above may provide definitive surgical imaging tools in the near future.
Electroencephalography (EEG) is often considered the "golden standard" in neurology. However, EEG monitoring is inadequate for questions regarding co-localization of electrical activity and changes in oxygenation and blood flow resulting from metabolic activity[120–122]. Thus, for such applications and measurements optical recordings may actually be more adequate and also applicable to monitoring or localizing foci of particular neurological disorders. The goal of the current brain optical imaging studies will be to translate recent findings into clinical practice.
We have reviewed a wide range of in vivo optical imaging techniques and explored their many important applications in studying the cerebral neurovascular functions. We have further provided insight into the fundamental basis of many in vivo optical imaging techniques and highlighted important considerations for their practical implementation. We hope that this detailed review will aid researchers who are interested in either using or developing in vivo optical imaging techniques to understand the key aspects that should be considered when acquiring measurements or analyzing data. We have surveyed the huge body of literature that discusses studies in which in vivo optical imaging technologies and systems have been used successfully in ground-breaking and fundamentally important research and applications in cerebral neuroscience. The development of in vivo optical imaging techniques represents a rapidly expanding field that is continually evolving to embrace new technologies and clinical applications.
Iadecola C: Neurovascular regulation in the normal brain and in Alzheimer's disease. Nat Rev Neurosci 2004, 5: 347–360. 10.1038/nrn1387
Attwell D, Laughlin SB: An energy budget for signaling in the grey matter of the brain. J Cereb Blood Flow Metab 2001, 21: 1133–1145.
Klein B, Kuschinsky W, Schrock H, Vetterlein F: Interdependency of local capillary density, blood flow, and metabolism in rat brains. Am J Physiol 1986, 251: H1333-H1340.
Faraci FM, Heistad DD: Regulation of large cerebral arteries and cerebral microvascular pressure. Circ Res 1990, 66: 8–17. 10.1161/01.RES.66.1.8
Orozco-Cabal L, Pollandt S, Liu J, Shinnick-Gallagher P, Gallagher JP: Regulation of synaptic transmission by CRF receptors. Rev Neuroscience 2006, 17: 279–307.
Rossi DJ: Another BOLD role for astrocytes: coupling blood flow to neural activity. Nat Neurosci 2006, 9: 159–161. 10.1038/nn0206-159
Attwell D, Buchan AM, Charpak S, Lauritzen M, Macvicar BA, Newman EA: Glial and neuronal control of brain blood flow. Nature 2010, 468: 232–243. 10.1038/nature09613
Zhang X, Toronov VY, Fabiani M, Gratton G, Webb AG: The study of cerebral hemodynamic and neuronal response to visual stimulation using simultaneous NIR optical tomography and BOLD fMRI in humans. Proc Soc Photo Opt Instrum Eng 2005, 5686: 566–572.
Hudetz AG: Blood flow in the cerebral capillary network: A review emphasizing observations with intravital microscopy. Microcirculation 1997, 4: 233–252. 10.3109/10739689709146787
Tsytsarev V, Premachandra K, Takeshita D, Bahar S: Imaging cortical electrical stimulation in vivo: fast intrinsic optical signal versus voltage-sensitive dyes. Opt Lett 2008, 33: 1032–1034. 10.1364/OL.33.001032
Ohki K, Chung S, Ch'ng YH, Kara P, Reid RC: Functional imaging with cellular resolution reveals precise micro-architecture in visual cortex. Nature 2005, 433: 597–603. 10.1038/nature03274
Stosiek C, Garaschuk O, Holthoff K, Konnerth A: In vivo two-photon calcium imaging of neuronal networks. Proc Natl Acad Sci USA 2003, 100: 7319–7324. 10.1073/pnas.1232232100
Kucheryavykh LY, Kucheryavykh YV, Rolon-Reyes K, Skatchkov SN, Eaton MJ, Cubano LA, Inyushin M: Visualization of implanted GL261 glioma cells in living mouse brain slices using fluorescent 4-(4-(dimethylamino)-styryl)-N-methylpyridinium iodide (ASP+). Biotechniques 2012, 0: 1–4.
Inyushin M, Kucheryavykh LY, Kucheryavykh YV, Nichols CG, Buono RJ, Ferraro TN, Skatchkov SN, Eaton MJ: Potassium channel activity and glutamate uptake are impaired in astrocytes of seizure-susceptible DBA/2 mice. Epilepsia 2010, 51: 1707–1713. 10.1111/j.1528-1167.2010.02592.x
Benedikt J, Inyushin M, Kucheryavykh YV, Rivera Y, Kucheryavykh LY, Nichols CG, Eaton MJ, Skatchkov SN: Intracellular polyamines enhance astrocytic coupling. Neuroreport 2012, 23: 1021–1025. 10.1097/WNR.0b013e32835aa04b
Boas DA, Jones SR, Devor A, Huppert TJ, Dale AM: A vascular anatomical network model of the spatio-temporal response to brain activation. Neuroimage 2008, 40: 1116–1129. 10.1016/j.neuroimage.2007.12.061
Huppert TJ, Allen MS, Benav H, Jones PB, Boas DA: A multicompartment vascular model for inferring baseline and functional changes in cerebral oxygen metabolism and arterial dilation. J Cereb Blood Flow Metab 2007, 27: 1262–1279. 10.1038/sj.jcbfm.9600435
Devor A, Dunn AK, Andermann ML, Ulbert I, Boas DA, Dale AM: Coupling of total hemoglobin concentration, oxygenation, and neural activity in rat somatosensory cortex. Neuron 2003, 39: 353–359. 10.1016/S0896-6273(03)00403-3
Franceschini MA, Nissila I, Wu WC, Diamond SG, Bonmassar G, Boas DA: Coupling between somatosensory evoked potentials and hemodynamic response in the rat. Neuroimage 2008, 41: 189–203. 10.1016/j.neuroimage.2008.02.061
Huttunen JK, Grohn O, Penttonen M: Coupling between simultaneously recorded BOLD response and neuronal activity in the rat somatosensory cortex. Neuroimage 2008, 39: 775–785. 10.1016/j.neuroimage.2007.06.042
Fazl M, Houlden DA, Weaver K: Correlation between Cerebral Blood-Flow, Somatosensory Evoked-Potentials, Ct Scan Grade and Neurological Grade in Patients with Subarachnoid Hemorrhage. Can J Neurol Sci 1991, 18: 453–457.
Denk W, Strickler JH, Webb WW: Two-photon laser scanning fluorescence microscopy. Science 1990, 248: 73–76. 10.1126/science.2321027
Garaschuk O, Milos RI, Konnerth A: Targeted bulk-loading of fluorescent indicators for two-photon brain imaging in vivo. Nat Protoc 2006, 1: 380–386. 10.1038/nprot.2006.58
Kuhn B, Denk W, Bruno RM: In vivo two-photon voltage-sensitive dye imaging reveals top-down control of cortical layers 1 and 2 during wakefulness. Proc Natl Acad Sci USA 2008, 105: 7588–7593. 10.1073/pnas.0802462105
Shih AY, Driscoll JD, Drew PJ, Nishimura N, Schaffer CB, Kleinfeld D: Two-photon microscopy as a tool to study blood flow and neurovascular coupling in the rodent brain. J Cereb Blood Flow Metab 2012, 32: 1277–1309. 10.1038/jcbfm.2011.196
Peng M, Rege A, Nan L, Thakor NV, Shanbao T: High resolution cerebral blood flow imaging by registered laser speckle contrast analysis. IEEE Trans Biomed Eng 2010, 57: 1152–1157.
Briers JD: Laser speckle contrast imaging for measuring blood flow. Optica Applicata 2007, 37: 139–152.
Li N, Jia X, Murari K, Parlapalli R, Rege A, Thakor NV: High spatiotemporal resolution imaging of the neurovascular response to electrical stimulation of rat peripheral trigeminal nerve as revealed by in vivo temporal laser speckle contrast. J Neurosci Methods 2009, 176: 230–236. 10.1016/j.jneumeth.2008.07.013
Miao P, Li N, Thakor NV, Tong S: Random process estimator for laser speckle imaging of cerebral blood flow. Opt Express 2010, 18: 218–236. 10.1364/OE.18.000218
Miao P, Lu H, Liu Q, Li Y, Tong S: Laser speckle contrast imaging of cerebral blood flow in freely moving animals. J Biomed Opt 2011, 16: 090502. 10.1117/1.3625231
Hecht N, Woitzik J, Dreier JP, Vajkoczy P: Intraoperative monitoring of cerebral blood flow by laser speckle contrast analysis. Neurosurg Focus 2009, 27: E11.
Parthasarathy AB, Weber EL, Richards LM, Fox DJ, Dunn AK: Laser speckle contrast imaging of cerebral blood flow in humans during neurosurgery: a pilot clinical study. J Biomed Opt 2010, 15: 066030. 10.1117/1.3526368
Rege A, Murari K, Seifert A, Pathak AP, Thakor NV: Multiexposure laser speckle contrast imaging of the angiogenic microenvironment. J Biomed Opt 2011., 16:
Li N, Downey JE, Bar-Shir A, Gilad AA, Walczak P, Kim H, Joel SE, Pekar JJ, Thakor NV, Pelled G: Optogenetic-guided cortical plasticity after nerve injury. Proc Natl Acad Sci USA 2011, 108: 8838–8843. 10.1073/pnas.1100815108
Zhang HF, Maslov K, Wang LV: In vivo imaging of subcutaneous structures using functional photoacoustic microscopy. Nat Protoc 2007, 2: 797–804. 10.1038/nprot.2007.108
Wang LV: Multiscale photoacoustic microscopy and computed tomography. Nat Photonics 2009, 3: 503–509. 10.1038/nphoton.2009.157
Liao LD, Lin CT, Shih YYI, Duong TQ, Lai HY, Wang PH, Wu R, Tsang S, Chang JY, Li ML: Transcranial imaging of functional cerebral hemodynamic changes in single blood vessels using in vivo photoacoustic microscopy. J Cereb Blood Flow Metab 2012, 32: 938–951. 10.1038/jcbfm.2012.42
Liao LD, Lin CT, Shih YYI, Lai HY, Zhao WT, Duong TQ, Chang JY, Chen YY, Li ML: Investigation of the cerebral hemodynamic response function in single blood vessels by functional photoacoustic microscopy. J Biomed Opt 2012, 17: 061210–1-061210–10.
Thakor NV: Highlights: Transcranial imaging of functional cerebral hemodynamic changes in single blood vessels. J Cereb Blood Flow Metab 2012, 32: 936–937. 10.1038/jcbfm.2012.41
Liao LD, Li ML, Lai HY, Shih YYI, Lo YC, Tsang S, Chao PCP, Lin CT, Jaw FS, Chen YY: Imaging brain hemodynamic changes during rat forepaw electrical stimulation using functional photoacoustic microscopy. Neuroimage 2010, 52: 562–570. 10.1016/j.neuroimage.2010.03.065
Wang LHV, Hu S: Photoacoustic tomography: in vivo imaging from organelles to organs. Science 2012, 335: 1458–1462. 10.1126/science.1216210
Yao J, Wang LV: Transverse flow imaging based on photoacoustic Doppler bandwidth broadening. J Biomed Opt 2010, 15: 021304–021305. 10.1117/1.3339953
Mesquita RC, Franceschini MA, Boas DA: Resting state functional connectivity of the whole head with near-infrared spectroscopy. Biomed Opt Express 2010, 1: 324–336. 10.1364/BOE.1.000324
Bunce SC, Izzetoglu M, Izzetoglu K, Onaral B, Pourrezaei K: Functional near-infrared spectroscopy - An emerging neuroimaging modality. IEEE Eng Med Biol Mag 2006, 25: 54–62.
Izzetoglu M, Izzetoglu K, Bunce S, Ayaz H, Devaraj A, Onaral B, Pourrezaei K: Functional near-infrared neuroimaging. IEEE Trans Neural Syst Rehabil Eng 2005, 13: 153–159. 10.1109/TNSRE.2005.847377
Hoshi Y: Functional near‒infrared spectroscopy: potential and limitations in neuroimaging studies. Int Rev Neurobiol 2005, 66: 237.
Suzurikawa J, Tani T, Nakao M, Tanaka S, Takahashi H: Voltage-sensitive-dye imaging of microstimulation-evoked neural activity through intracortical horizontal and callosal connections in cat visual cortex. J Neural Eng 2009, 6: 066002. 10.1088/1741-2560/6/6/066002
Tsytsarev V, Pope D, Pumbo E, Yablonskii A, Hofmann M: Study of the cortical representation of whisker directional deflection using voltage-sensitive dye optical imaging. Neuroimage 2010, 53: 233–238. 10.1016/j.neuroimage.2010.06.022
Wayland H, Johnson PC: Erythrocyte velocity measurement in microvessels by a two-slit photometric method. J Appl Physiol 1967, 22: 333–337.
Dirnagl U, Villringer A, Gebhardt R, Haberl RL, Schmiedek P, Einhaupl KM: Three-dimensional reconstruction of the rat brain cortical microcirculation in vivo. J Cereb Blood Flow Metab 1991, 11: 353–360. 10.1038/jcbfm.1991.74
Villringer A, Them A, Lindauer U, Einhaupl K, Dirnagl U: Capillary perfusion of the rat brain cortex. An in vivo confocal microscopy study. Circ Res 1994, 75: 55–62. 10.1161/01.RES.75.1.55
Helmchen F, Denk W: Deep tissue two-photon microscopy. Nat Methods 2005, 2: 932–940. 10.1038/nmeth818
Svoboda K, Yasuda R: Principles of two-photon excitation microscopy and its applications to neuroscience. Neuron 2006, 50: 823–839. 10.1016/j.neuron.2006.05.019
Cauli B, Tong XK, Rancillac A, Serluca N, Lambolez B, Rossier J, Hamel E: Cortical GABA interneurons in neurovascular coupling: relays for subcortical vasoactive pathways. J Neurosci 2004, 24: 8940–8949. 10.1523/JNEUROSCI.3065-04.2004
Filosa JA, Bonev AD, Straub SV, Meredith AL, Wilkerson MK, Aldrich RW, Nelson MT: Local potassium signaling couples neuronal activity to vasodilation in the brain. Nat Neurosci 2006, 9: 1397–1403. 10.1038/nn1779
Mulligan SJ, MacVicar BA: Calcium transients in astrocyte endfeet cause cerebrovascular constrictions. Nature 2004, 431: 195–199. 10.1038/nature02827
Rancillac A, Rossier J, Guille M, Tong XK, Geoffroy H, Amatore C, Arbault S, Hamel E, Cauli B: Glutamatergic Control of Microvascular Tone by Distinct GABA Neurons in the Cerebellum. J Neurosci 2006, 26: 6997–7006. 10.1523/JNEUROSCI.5515-05.2006
Simard M, Arcuino G, Takano T, Liu QS, Nedergaard M: Signaling at the gliovascular interface. J Neurosci 2003, 23: 9254–9262.
Zonta M, Angulo MC, Gobbo S, Rosengarten B, Hossmann KA, Pozzan T, Carmignoto G: Neuron-to-astrocyte signaling is central to the dynamic control of brain microcirculation. Nat Neurosci 2003, 6: 43–50. 10.1038/nn980
Devor A, Tian PF, Nishimura N, Teng IC, Hillman EMC, Narayanan SN, Ulbert I, Boas DA, Kleinfeld D, Dale AM: Suppressed neuronal activity and concurrent arteriolar vasoconstriction may explain negative blood oxygenation level-dependent signal. J Neurosci 2007, 27: 4452–4459. 10.1523/JNEUROSCI.0134-07.2007
Faraci FM, Breese KR: Nitric oxide mediates vasodilatation in response to activation of N-methyl-D-aspartate receptors in brain. Circ Res 1993, 72: 476–480. 10.1161/01.RES.72.2.476
Kleinfeld D, Mitra PP, Helmchen F, Denk W: Fluctuations and stimulus-induced changes in blood flow observed in individual capillaries in layers 2 through 4 of rat neocortex. Proc Natl Acad Sci USA 1998, 95: 15741–15746. 10.1073/pnas.95.26.15741
Takano T, Tian GF, Peng W, Lou N, Libionka W, Han X, Nedergaard M: Astrocyte-mediated control of cerebral blood flow. Nat Neurosci 2006, 9: 260–267. 10.1038/nn1623
Zhang SX, Murphy TH: Imaging the impact of cortical microcirculation on synaptic structure and sensory-evoked hemodynamic responses in vivo. PLoS Biol 2007, 5: 1152–1167.
Chaigneau E, Tiret P, Lecoq J, Ducros M, Knopfel T, Charpak S: The relationship between blood flow and neuronal activity in the rodent olfactory bulb. J Neurosci 2007, 27: 6452–6460. 10.1523/JNEUROSCI.3141-06.2007
Chuquet J, Hollender L, Nimchinsky EA: High-resolution in vivo imaging of the neurovascular unit during spreading depression. J Neurosci 2007, 27: 4036–4044. 10.1523/JNEUROSCI.0721-07.2007
Filosa JA, Bonev AD, Nelson MT: Calcium dynamics in cortical astrocytes and arterioles during neurovascular coupling. Circ Res 2004, 95: E73-E81. 10.1161/01.RES.0000148636.60732.2e
Theer P, Hasan MT, Denk W: Two-photon imaging to a depth of 1000 microm in living brains by use of a Ti:Al2O3 regenerative amplifier. Opt Lett 2003, 28: 1022–1024. 10.1364/OL.28.001022
Helmchen F: Two-Photon Functional Imaging of Neuronal Activity. ed: CRC Press; 2009:37–58.
Smith LE, Wesolowski E, McLellan A, Kostyk SK, D'Amato R, Sullivan R, D'Amore PA: Oxygen-induced retinopathy in the mouse. Invest Ophthalmol Vis Sci 1994, 35: 101–111.
Devor A, Shih A, Tsai P, Blinder P, Tian P, Teng I, Kleinfeld D: Two-Photon Laser Scanning Microscopy as a Tool to Study Cortical Vasodynamics Under Normal and Ischemic Conditions. In Imaging the Brain with Optical Methods. Edited by: Roe AW. Springer New York: LA - English; 2010:245–261.
Murari K, Li N, Rege A, Jia X, All A, Thakor N: Contrast-enhanced imaging of cerebral vasculature with laser speckle. Appl Opt 2007, 46: 5340–5346. 10.1364/AO.46.005340
Murari K, Greenwald E, Etienne-Cummings R, Cauwenberghs G, Thakor N: Design and characterization of a miniaturized epi-illuminated microscope. Conf Proc IEEE Eng Med Biol Soc 2009, 2009: 5369–5372.
Arulmozhi DK, Veeranjaneyulu A, Bodhankar SL: Migraine: current concepts and emerging therapies. Vascul Pharmacol 2005, 43: 176–187. 10.1016/j.vph.2005.07.001
Bouchard MB, Chen BR, Burgess SA, Hillman EMC: Ultra-fast multispectral optical imaging of cortical oxygenation, blood flow, and intracellular calcium dynamics. Opt Express 2009, 17: 15670–15678. 10.1364/OE.17.015670
Grinvald A, Hildesheim R: VSDI: a new era in functional imaging of cortical dynamics. Nat Rev Neurosci 2004, 5: 874–885. 10.1038/nrn1536
Chemla S, Chavane F: Voltage-sensitive dye imaging: Technique review and models. J Physiol Paris 2010, 104: 40–50. 10.1016/j.jphysparis.2009.11.009
Tsytsarev V, Bernardelli C, Maslov KI: Living brain optical imaging: technology, methods and applications. J Neurosci Neuroeng 2012., 1:
Shoham D, Glaser DE, Arieli A, Kenet T, Wijnbergen C, Toledo Y, Hildesheim R, Grinvald A: Imaging cortical dynamics at high spatial and temporal resolution with novel blue voltage-sensitive dyes. Neuron 1999, 24: 791–802. 10.1016/S0896-6273(00)81027-2
Tsytsarev V, Fukuyama H, Pope D, Pumbo E, Kimura M: Optical imaging of interaural time difference representation in rat auditory cortex. Front Neuroeng 2009, 2: 2.
Devor A, Sakadzic S, Srinivasan VJ, Yaseen MA, Nizar K, Saisan PA, Tian P, Dale AM, Vinogradov SA, Franceschini MA, Boas DA: Frontiers in optical imaging of cerebral blood flow and metabolism. J Cereb Blood Flow Metab 2012, 32: 1259–1276. 10.1038/jcbfm.2011.195
Barnett L, Platisa J, Popovic M, Pieribone VA, Hughes T: A fluorescent, genetically-encoded voltage probe capable of resolving action potentials. PLoS One 2012, 7: e43454. 10.1371/journal.pone.0043454
Wang LV: Tutorial on photoacoustic microscopy and computed tomography. IEEE J Sel Top Quantum Electronics 2008, 14: 171–179.
Maslov K, Stoica G, Wang LV: In vivo dark-field reflection-mode photoacoustic microscopy. Opt Lett 2005, 30: 625–627. 10.1364/OL.30.000625
Zhang HF, Maslov K, Stoica G, Wang LV: Functional photoacoustic microscopy for high-resolution and noninvasive in vivo imaging. Nat Biotechnol 2006, 24: 848–851. 10.1038/nbt1220
Tsytsarev V, Hu S, Yao JJ, Maslov K, Barbour DL, Wang LV: Photoacoustic microscopy of microvascular responses to cortical electrical stimulation. J Biomed Opt 2011., 16:
Yao J, Maslov KI, Shi Y, Taber LA, Wang LV: In vivo photoacoustic imaging of transverse blood flow by using Doppler broadening of bandwidth. Opt Lett 2010, 35: 1419–1421. 10.1364/OL.35.001419
Stein EW, Maslov K, Wang LV: Noninvasive, in vivo imaging of blood-oxygenation dynamics within the mouse brain using photoacoustic microscopy. J Biomed Opt 2009, 14: 020502. 10.1117/1.3095799
Ermilov SA, Khamapirad T, Conjusteau A, Leonard MH, Lacewell R, Mehta K, Miller T, Oraevsky AA: Laser optoacoustic imaging system for detection of breast cancer. J Biomed Opt 2009, 14: 024007. 10.1117/1.3086616
Jokerst JV, Thangaraj M, Kempen PJ, Sinclair R, Gambhir SS: Photoacoustic imaging of mesenchymal stem cells in living mice via silica-coated gold nanorods. ACS Nano 2012, 6: 5920–5930. 10.1021/nn302042y
Laufer J, Johnson P, Zhang E, Treeby B, Cox B, Pedley B, Beard P: In vivo preclinical photoacoustic imaging of tumor vasculature development and therapy. J Biomed Opt 2012, 17: 056016. 10.1117/1.JBO.17.5.056016
Prost A, Funke A, Tanter M, Aubry JF, Bossy E: Photoacoustic-guided ultrasound therapy with a dual-mode ultrasound array. J Biomed Opt 2012, 17: 061205. 10.1117/1.JBO.17.6.061205
Petrov IY, Petrov Y, Prough DS, Cicenaite I, Deyo DJ, Esenaliev RO: Optoacoustic monitoring of cerebral venous blood oxygenation though intact scalp in large animals. Opt Express 2012, 20: 4159–4167. 10.1364/OE.20.004159
Nie L, Cai X, Maslov K, Garcia-Uribe A, Anastasio MA, Wang LV: Photoacoustic tomography through a whole adult human skull with a photon recycler. J Biomed Opt 2012, 17: 110506–110506. 10.1117/1.JBO.17.11.110506
Hajireza P, Shi W, Zemp RJ: Real-time handheld optical-resolution photoacoustic microscopy. Opt Express 2011, 19: 20097–20102. 10.1364/OE.19.020097
Khan B, Chand P, Alexandrakis G: Spatiotemporal relations of primary sensorimotor and secondary motor activation patterns mapped by NIR imaging. Biomed Opt Express 2011, 2: 3367–3386. 10.1364/BOE.2.003367
Aletti F, Re R, Pace V, Contini D, Molteni E, Cerutti S, Maria Bianchi A, Torricelli A, Spinelli L, Cubeddu R, Baselli G: Deep and surface hemodynamic signal from functional time resolved transcranial near infrared spectroscopy compared to skin flowmotion. Comput Biol Med 2012, 42: 282–289. 10.1016/j.compbiomed.2011.06.001
Bozkurt A, Rosen A, Rosen H, Onaral B: A portable near infrared spectroscopy system for bedside monitoring of newborn brain. Biomed Eng Online 2005, 4: 29. 10.1186/1475-925X-4-29
Coyle SM, Ward TE, Markham CM: Brain-computer interface using a simplified functional near-infrared spectroscopy system. J Neural Eng 2007, 4: 219–226. 10.1088/1741-2560/4/3/007
Wang Y, Hu S, Maslov K, Zhang Y, Xia Y, Wang LV: In vivo integrated photoacoustic and confocal microscopy of hemoglobin oxygen saturation and oxygen partial pressure. Opt Lett 2011, 36: 1029–1031. 10.1364/OL.36.001029
Liu T, Wei Q, Wang J, Jiao SL, Zhang HF: Combined photoacoustic microscopy and optical coherence tomography can measure metabolic rate of oxygen. Biomedical Optics Express 2011, 2: 1359–1365. 10.1364/BOE.2.001359
Yakovlev VV, Zhang HF, Noojin GD, Denton ML, Thomas RJ, Scully MO: Stimulated Raman photoacoustic imaging. Proc Natl Acad Sci USA 2010, 107: 20335–20339. 10.1073/pnas.1012432107
Yao J, Xia J, Maslov KI, Nasiriavanaki M, Tsytsarev V, Demchenko AV, Wang LV: Noninvasive photoacoustic computed tomography of mouse brain metabolism in vivo. Neuroimage 2012.
Ma H, Zhao M, Schwartz TH: Dynamic Neurovascular Coupling and Uncoupling during Ictal Onset, Propagation, and Termination Revealed by Simultaneous In Vivo Optical Imaging of Neural Activity and Local Blood Volume. Cereb Cortex 2012.
Obrenovitch TP, Chen SB, Farkas E: Simultaneous, live imaging of cortical spreading depression and associated cerebral blood flow changes, by combining voltage-sensitive dye and laser speckle contrast methods. Neuroimage 2009, 45: 68–74. 10.1016/j.neuroimage.2008.11.025
Keller E, Ishihara H, Nadler A, Niederer P, Seifert B, Yonekawa Y, Frei K: Evaluation of brain toxicity following near infrared light exposure after indocyanine green dye injection. J Neurosci Methods 2002, 117: 23–31. 10.1016/S0165-0270(02)00044-4
Moore GE, Peyton WT, French LA, Walker WW: The Clinical Use of Fluorescein in Neurosurgery - the Localization of Brain Tumors. J Neurosurg 1948, 5: 392–398. 10.3171/jns.1948.5.4.0392
Kim EH, Cho JM, Chang JH, Kim SH, Lee KS: Application of intraoperative indocyanine green videoangiography to brain tumor surgery. Acta Neurochir 2011, 153: 1487–1495. 10.1007/s00701-011-1046-x
Tsytsarev V, Maslov KI, Yao J, Parameswar AR, Demchenko AV, Wang LV: In vivo imaging of epileptic activity using 2-NBDG, a fluorescent deoxyglucose analog. J Neurosci Methods 2012, 203: 136–140. 10.1016/j.jneumeth.2011.09.005
Reichold J, Stampanoni M, Lena Keller A, Buck A, Jenny P, Weber B: Vascular graph model to simulate the cerebral blood flow in realistic vascular networks. J Cereb Blood Flow Metab 2009, 29: 1429–1443. 10.1038/jcbfm.2009.58
Hill DK, Keynes RD: Opacity changes in stimulated nerve. J Physiol 1949, 108: 278–281.
Lenkov DN, Volnova AB, Pope A, Tsytsarev V: Advantages and limitations of brain imaging methods in the research of absence epilepsy in humans and animal models. J Neurosci Methods 2012.
Ferrari M, Quaresima V: A brief review on the history of human functional near-infrared spectroscopy (fNIRS) development and fields of application. Neuroimage 2012, 63: 921–935. 10.1016/j.neuroimage.2012.03.049
Gratton G, Fabiani M: Fast optical imaging of human brain function. Front Hum Neurosci 2010, 4: 52.
Haglund MM, Hochman DW: Imaging of intrinsic optical signals in primate cortex during epileptiform activity. Epilepsia 2007, 48 Suppl 4: 65–74.
Beard P: Biomedical photoacoustic imaging. Interface focus 2011, 1: 602–631. 10.1098/rsfs.2011.0028
Wang L: Photoacoustic Imaging and Spectroscopy vol. 144. CRC Press; 2009.
Flusberg BA, Nimmerjahn A, Cocker ED, Mukamel EA, Barretto RP, Ko TH, Burns LD, Jung JC, Schnitzer MJ: High-speed, miniaturized fluorescence microscopy in freely moving mice. Nat Methods 2008, 5: 935–938. 10.1038/nmeth.1256
Sanai N, Eschbacher J, Hattendorf G, Coons SW, Preul MC, Smith KA, Nakaji P, Spetzler RF: Intraoperative confocal microscopy for brain tumors: a feasibility analysis in humans. Neurosurgery 2011, 68: 282–290. discussion 290
Kozma R, Davis JJJ, Freeman WJ: Synchronized minima in ECoG power at frequencies between beta-gamma oscillations disclose cortical singularities in cognition. J Neurosci Neuroeng 2012., 1:
Liao LD, Lin CT: Novel trends in biosensors used for electroencephalography measurements in neurocognitive engineering applications. J Neurosci Neuroeng 2012, 1: 32–41. 10.1166/jnsne.2012.1006
Liao LD, Lin CT, McDowell K, Wickenden AE, Gramann K, Jung TP, Ko LW, Chang JY: Biosensor technologies for augmented brain-computer interfaces in the next decades. Proc IEEE 2012, 100: 1553–1566.
The authors thank the National University of Singapore for supporting the In-Vivo Optical Neuroimaging Group at the Singapore Institute for Neurotechnology (SINAPSE) under grant number R-711-000-026-133. The authors also thank those who kindly provided permission for the reproduction of the figures found within this review paper.
The authors declare that they have no competing interests.
Study concept and design (LDL, VT and NT); drafting of the manuscript (LDL and VT); critical revision of the manuscript for important intellectual content (IDM, MLL, RE, AV, JO, NT and LDL); obtained funding (NT, LDL and YRL); administrative, technical, and material support (HYL, YYC and NT, LDL); study supervision (LDL and NT). All authors read and approved the final manuscript.
Lun-De Liao, Vassiliy Tsytsarev contributed equally to this work.
About this article
- Neurovascular coupling
- Cerebral neuroimaging
- 2-photon microscopy
- Laser speckle contrast imaging
- Voltage sensitive dye imaging
- Functional photoacoustic microscopy
- Functional near-infrared spectroscopy