Coronary artery bypass grafting hemodynamics and anastomosis design: a biomedical engineering review
BioMedical Engineering OnLine volume 12, Article number: 129 (2013)
In this paper, coronary arterial bypass grafting hemodynamics and anastomosis designs are reviewed. The paper specifically addresses the biomechanical factors for enhancement of the patency of coronary artery bypass grafts (CABGs). Stenosis of distal anastomosis, caused by thrombosis and intimal hyperplasia (IH), is the major cause of failure of CABGs. Strong correlations have been established between the hemodynamics and vessel wall biomechanical factors and the initiation and development of IH and thrombus formation. Accordingly, several investigations have been conducted and numerous anastomotic geometries and devices have been designed to better regulate the blood flow fields and distribution of hemodynamic parameters and biomechanical factors at the distal anastomosis, in order to enhance the patency of CABGs. Enhancement of longevity and patency rate of CABGs can eliminate the need for re-operation and can significantly lower morbidity, and thereby reduces medical costs for patients suffering from coronary stenosis. This invited review focuses on various endeavors made thus far to design a patency-enhancing optimized anastomotic configuration for the distal junction of CABGs.
Coronary artery disease (CAD) is the leading cause of death globally, and is expected to account for 14.2% of all deaths by 2030 . According to the statistics from the American Heart Association, mortality data show that cardiovascular disease, as the underlying cause of death, accounted for 34.3% of all (1 of every 2.9) deaths; in particular, coronary heart disease caused approximately 1 of every 6 deaths in 2006 in the United States .
Several alternative treatments exist for CAD, including medical therapy, rotablation, endarterectomy, percutaneous coronary intervention (PCI) or balloon angioplasty, stenting, and coronary arterial bypass grafting (CABG). Depending on the severity, number and position of atherosclerotic lesions and the clinical history of the patient, any of the above mentioned treatments may be chosen. For high-risk patients, such as those with left main coronary artery (LMCA) disease, severe three-coronary vessel disease, severe ventricular dysfunction (i.e., low ejection fraction), and diabetes mellitus, CABG is the preferred treatment . In general, the greater the extent of coronary atherosclerosis and its diffuseness, the more compelling the choice of CABG, particularly if the left ventricle function is depressed .
CABG is a surgical procedure performed to graft arteries or veins from the patient’s body or synthetic conduits to the occluded coronary arteries in order to bypass the atherosclerotic narrowing and improve the blood supply to the coronary circulation, for nourishing the myocardium. Figure 1 illustrates both arterial and venous grafts, each bypassing a coronary blockage (stenosis) formed by cholesterol build-ups.
Although CABG is extremely effective for symptomatic relief and prognostic improvement in CAD and is the preferred remedy for high-risk patients, it is not devoid of complications and the long-term benefits are directly related to continuing conduit patency. Approximately 10–15% of vein grafts occlude during the first year after operation . About half of the vein grafts are only effective for a period of 5 to 10 years [6, 7]. By 10 years after surgery, about 60% of vein grafts are patent, only 50% of which remain free of significant stenosis .
Early graft failure (within 30 days) of bypass grafts is attributable to surgical technical errors and resulting thrombosis, while late graft failures are mainly caused by progression of atherosclerosis and intimal hyperplasia (IH) . Various studies have found IH to be the major cause of graft stenosis . IH is the abnormal, continued proliferation and overgrowth of smooth muscle cells (SMCs) in response to endothelial injury or dysfunction. Although the exact mechanism and pathophysiology of IH remains an enigma, there are indications that both biological and biomechanical factors are involved, which include endothelial injury , platelet activation , disturbed local hemodynamics [13, 14], compliance mismatch between the graft and host vessel , and interactions between blood and graft material .
Among the abovementioned factors, hemodynamic parameters (HPs) are believed to be highly important [17, 18] in the genesis and development of IH. It has been shown that in end-to-side (ETS) graft–artery configurations, IH develops predominantly at the toe and heel of the anastomosis and on the artery bed across the junction where disturbed flow patterns and hemodynamic factors are observed [17, 19]. On the basis of this focal distribution of intimal thickening (IT), disturbed flow patterns and the associated hemodynamic factors have been correlated with the onset and progression of atherosclerosis and distal anastomotic intimal hyperplasia (DAIH) [19, 20]. Among the important hemodynamic factors are wall shear stress (WSS), spatial and temporal gradients of WSS, and oscillatory shear index (OSI).
Accordingly, several investigations have been conducted and different anastomotic geometries and devices have been designed to improve the flow fields and HPs distribution at ETS anastomosis, in order to enhance the graft patency. These investigations include studies on the effects of geometrical factors, such as anastomotic angle [21–27], modified configuration of distal anastomosis [28–31], graft-to-host artery diameter ratio [32–34], and out-of-plane graft [35–37], and effects of stenosis severity and proximal artery flow [38, 39], irregularities of venous graft wall (due to venous valve sinus) , and distance of grafting (i.e., the distance of anastomosis from the occluded site) . Considerable efforts towards attaining an optimal patency-enhancing CABG anastomotic configuration have been made, and continue to be made by investigators. This is because enhancement of the longevity and patency rate of CABGs (by means of an optimal anastomotic configuration) can result in considerable improvement in the left ventricular contractility index and ejection fraction of patients with CAD , elimination of the need for re-operation, reduced medical costs for patients suffering from coronary stenosis, and significantly lower morbidity.
Accordingly, this paper reviews the theories on bypass graft failure and its causative biological and biomechanical factors, followed by various attempts to design an optimal anastomotic configuration for the distal CABG anastomosis. This review illuminates the impact of CABG layout designs towards obtaining higher long-term graft patency rates and the benefit of superior anastomotic designs for the improvement of global ejection fraction of patients with coronary artery disease.
The first section of this review covers the studies correlating different biological and biomechanical factors to initiation and progression of atherosclerosis and IH. The majority of the investigations on biological factors are in vivo studies along with complementing in vitro investigations. On the other hand, computational simulations of blood flow along with numerous in vitro and in vivo investigations constitute the studies of the biomechanical factors and hemodynamic parameters, and have provided strong evidence on the influence of these factors on initiation and onset of IH.
The second section of this review elaborates on the various attempts to design an optimal anastomotic configuration for CABG. These attempts include adjustments of anastomotic angle and graft-to-host artery ratio, design of different cuffed and patched anastomotic configurations, and design and development of other novel configurations (such as the coupled sequential anastomoses) and synthetic devices.
Finally, the current state of the art of CABG anastomotic configurations is discussed and future directions are suggested.
Theories on bypass graft failure
According to angiography and histological examinations, common graft failure modes include acute thrombosis, IH, and onset of progressive atherosclerosis [8, 43, 44]. Thrombosis is the formation of a blood clot (thrombus) in arteries or veins, as a result of low blood velocity within a graft caused by a flow-limiting stenosis or high-shear rates caused by a jet-type flow . Atherosclerosis is a disease of the arteries in which fatty material and plaque are deposited in the wall of an artery, resulting in narrowing of the arterial lumen and eventual impairment of blood flow. The predominant theory suggests that atherosclerosis develops in response to injury ; however, there is no consensus as to whether this injury is mechanical, chemical, immunological, or a combination of these and other factors.
IH, the major cause of graft failure, is the abnormal increase in the number of endothelial cells (ECs) and thickening of the tunica intima of a blood vessel. IH was first described by Carrel and Guthrie , who observed that within a few days after arterial vein graft implantation, the anastomotic stitches happened to be covered with a material similar to the normal endothelium.
Although the pathological mechanisms responsible for the development of IH and atherosclerosis have not been fully elucidated, many theories have been hypothesized, indicating the involvement of both biological and biomechanical factors, some of which are briefly discussed in the following sections.
The endothelial cells sense the mechanical forces acting on them through mechanotransducers . The presence of abnormal mechanical forces on the endothelium leads to some biological effects, which can trigger excessive release of mitogens and growth factors and lead to subsequent SMC proliferation and excessive platelet aggregation . Monocytes and cholesterol accumulate in the intima, creating foam cells. Monocytes become macrophages within the intima, resulting in formation of an atherosclerotic plaque .
Older plaques can develop into a calcified lesion or nodule. The origin or mechanism of the calcification is not precisely known, but it appears to be associated with healed plaque. The artery with a rigid matrix is unable to remodel, causing further cellular proliferation to push the fibrous cap out into the lumen. The rupture of this fibrous cap exposes tissue factors and collagen to the blood, forming a nidus for thrombus formation and consequent restenosis.
During the surgery, endothelium and SMCs of the recipient artery and the autogenous graft are injured in several ways: grasping of the instruments used to harvest the graft; routinely applied high intra-luminal pressure to check for the graft leakage; and sutures for the construction of the anastomoses. In addition, due to implantation of the vein graft in the arterial circulation, the grafted vein is exposed to high pressures and high blood flow causing further damage to the venous endothelium .
The injured endothelium is less capable of producing anti-proliferative products , and the physiological balance in the vessel wall is further disturbed by the release of intracellular growth factors from injured medial SMCs [6, 58]. It has been demonstrated that acute injury to the intima and media can produce hyperplasia and SMCs proliferation, which occur at a rate proportional to the degree of the injury [11, 49]. It is also observed that IH forms around the injury sites after balloon angioplasty  (Figure 2). These suggest the involvement of a wound healing process in IH formation.
The ability of a blood vessel to expand and contract passively with changes in pressure is an important function of large arteries and veins. Compliance is a measure of the tendency of a hollow organ to resist recoil toward its original dimensions upon removal of a distending or compressing force. In other words, the ability of a vessel to distend and increase volume with increasing transmural (inside minus outside) pressure is the vessel compliance (C), which is defined as:
where ∆V is change in volume and ∆P is change in transmural pressure. It is usually assumed that an increase in the volume of a blood vessel segment is almost exclusively caused by an increase in its radius since elongation is negligible in-vivo . Thus, compliance can be estimated through the variation in cross-sectional area, and the above relation can be rewritten as :
where D d is diastolic diameter, D s is systolic diameter, and P P is the pulse pressure. Veins have a higher compliance than arteries due to their thinner walls. A schematic volume-pressure relationship for an artery and vein is depicted in Figure 3. Two important characteristics stand out. Firstly, the curves are not linear (slopes are not constant); this is because the blood vessel wall is a heterogeneous tissue. Therefore, compliance decreases at higher pressure and volume (i.e., vessels become stiffer at higher pressures and volumes). Secondly, although at low pressure the compliance of a vein is about 10 to 20 times greater than that of an artery, at high pressure and volume, venous compliance (slope of volume-pressure curve) becomes smaller than arterial compliance. In case of arterial bypass, a mismatch in compliance between the vascular graft and the host artery is regarded as a mechanical factor which is detrimental to graft patency.
It has been shown in vivo that a higher degree of compliance mismatch between the graft and the artery results in a greater amount of suture-line IH formation in ETS anastomoses [17, 66, 67], but not in end-to-end (ETE) junctions . Following these observations, Ballyk et al.  performed a large-strain finite-element analysis of vascular wall mechanics to compare the influence of compliance mismatch on intramural stresses in ETE versus ETS anastomoses. They found out that increased compliance mismatch increases stresses in ETS junctions (Figure 4), but, it has little influence on stresses in ETE junctions, suggesting that the proliferative influence of increased compliance mismatch on suture-line hyperplasia in ETS anastomoses can be explained by the resulted increase in intramural stresses. As high intramural stresses have generally been observed at suture-lines in both ETS and ETE junctions, it has been suggested that elevated suture-line intramural stresses might be an important proliferative stimulus for IH formation in vascular reconstructions , and that IH might be a response to the mechanical injury caused by stretch and high intramural stresses as a result of compliance mismatch .
Although the detailed mechanism for stress-induced IT is not completely understood, it has been shown that endothelial cells respond to stretch stimuli by producing transcription factors which stimulate their replication  and chemotactic factors and mitogens which cause smooth muscle cells to migrate and proliferate .
Compliance mismatch is not only between the stiffer bypass graft and the artery, but also at the anastomosis itself between the suture and the graft on one side and between the suture and the artery on the other side, forming a para-anastomotic hyper-compliant zone and elevating intramural stresses. Different surgical techniques have been suggested to reduce the level of intramural stresses (elevated by compliance mismatch) and the amount of consequent IH (e.g., by geometric compliance matching (Figure 5)  or by using an external stent/sheath [72, 73]). However, it seems that compliance mismatch alone, without trauma caused by suturing, causes only limited amounts of IH formation . Thus, in order to alleviate the trauma in anastomosis, alternatives to sutures have been suggested, including biological glues, clips, clamps, magnetic vascular positioners, and laser generated solders, some of which have shown promising results [75–78]; yet, further work is required before they can become applicable for routine use.
Although compliance mismatch has been reported to cause trapping of micro-particles near the wall downstream the anastomosis  and to disturb the protein transport process  in ETE junctions, computational studies of ETS anastomosis models have illustrated that compliance mismatch has only minor effects on local hemodynamic factors [18, 81].
In vivo observations indicate that in an ETS graft-artery anastomotic configuration, IH occurs preferentially around the suture-line (especially at the toe and heel of the anastomosis), and also develops on the bed of the host artery across the junction [17, 19, 26, 67, 82, 83] (Figure 6).
Arterial floor IT is attributable to altered flow conditions . Although it has been suggested that suture-line IT might be related to vascular healing, an in vivo study by Sottuirai  has shown opposing results. He investigated the role of anastomotic configuration, using autogenous femoro-femoral bypass with ETS configuration and ETE interposition graft in canine models. Since compliance mismatch is not an issue in the autogenous femoro-femoral bypass, for DAIH which exclusively occurred in the ETS (and not in the ETE) distal anastomoses, the geometry of the distal anastomosis has been concluded to be the logical causal factor. Moreover, creation of an arbitrary stricture on an extended hood of the arterial graft, to function as the “physiologic toe”, resulted in transferring DAIH to the site of the graft stricture (Figure 7). This attests to the fact that it is the ETS anastomotic configuration and its unnatural flow conditions, and not the trauma along the suture-line, which contributes to DAIH formation (more details are given in the next section on “Disturbed flow patterns associated with distal ETS anastomosis”).
Several investigations have been conducted to better understand the relationship between blood flow-based stresses acting on the walls and IT in bypass grafts [13, 17, 19, 85–88]. Subsequently, multiple hemodynamic parameters have been associated with occlusive formations in arterial bypass grafts and other branching blood vessel configurations, as follows.
Safe bandwidth of WSS
Endothelial cells are constantly exposed to shear stress, induced by blood flow. Endothelial shear stress (i.e., WSS or τ w ) is the product of dynamic viscosity (μ) and shear rate () of blood at the vessel wall:
where U zɵ is the velocity component parallel to the vessel wall, r is the radial axis, and R is the radius of the blood vessel.
The existence of a safe bandwidth of WSS has been suggested by Kleinstreuer et al.  to explain the localization of atherosclerotic plaques and IH, based on two contradictory hypotheses: (i) high shear stress theory and (ii) low shear stress theory.
By experimental exposure of endothelium to high shear stresses, Fry [90, 91] showed that a sufficiently high shear stress level would induce endothelial injury and promote the development of lesions, which were postulated to increase the permeability of endothelium and to alter the transport of molecular species across the endothelial barrier into the arterial wall, resulting in plaque formation.
On the other hand, Caro et al. , observed that atherosclerotic lesions occur along the inner wall of arterial curvature, where low shear stress exists. Hence, they proposed that, due to low WSS and enhanced particle residence time in flow separation and flow recirculation zones, excess cholesterol is deposited on the surface of the lumen, initiating atheroma growth, while in the regions of moderately high WSS, more cholesterol is washed away by the blood flow.
Hence, combining the abovementioned opposing theories, it is suggested that there exists a safe bandwidth of WSS and the wall shear that falls outside of this range will result in plaque formation. This hypothesis has successfully determined the sites and growth patterns of atherosclerotic lesions and IH in several arterial bifurcations and bypass graft configurations, respectively [93, 94]. Moreover, study of numerical results of simulation of blood flow in the human aortic arch has suggested preferential development of early atherosclerotic lesions in regions of extreme (either maxima or minima) WSS and pressure . In addition, it has been reported that the ECs in both low and high shear regions experience structural and functional abnormalities , thereby supporting the hypothesis of “safe bandwidth of WSS”.
Low-magnitude high-oscillatory WSS
Based on the theory of Caro et al. , it is the “shear-dependent mass transfer” which is responsible for atheroma development and IT. Low shear stress acting on the vessel wall has been introduced as the key hemodynamic factor involved in the localization of IT, due to significant correlations found between the preferred sites of IT and the regions of slow recirculation flow (i.e., long particle residence time) with low WSS [14, 88, 97, 98].
Morinaga et al.  investigated IT occurrence in autogenous vein grafts in dogs, by comparing the conditions of high flow rate and low WSS with low flow rate and high WSS. A comparatively significant intimal thickness was observed in high flow rate and low WSS condition, revealing that WSS, and not the rate of flow, is the essential hemodynamic factor related to IH.
Ku et al.  found a positive correlation between plaque location and low, oscillating shear stress, indicating that marked oscillations in the direction of wall shear may enhance atherogenesis. Consequently, they put forward the concept of “oscillatory shear index” to quantify the oscillatory nature of WSS. Based on its modified definition (Equation 4) , the OSI value varies between 0 and 0.5, where 0 corresponds to the regions experiencing no reverse flow, and 0.5 is for the case of fully oscillatory flow without net forward flow.
where T is the time period of a cardiac cycle and is the WSS vector.
Li and Rittgers  compared the mechanical factors, obtained from in vitro study of pulsatile flow in a model of the distal ETS anastomosis of an arterial bypass graft, with histological findings of IH formation from earlier canine studies. Theeir results suggest that regions exposed to a combination of low-mean WSS and high-OSI may be most prone to IH formation. The same conclusion was obtained using in vitro preconditioned human umbilical vein ECs . Besides, similar correlations of the hemodynamic parameters and sites of IT formation were observed by Zhang et al.  in a computational investigation of blood flow in a complete coronary artery bypass model.
Both low-mean shear and oscillatory shear stress contribute to an increased near-wall particle residence time, which may alter the mass transport of atherogenic substances to the vessel wall and increase the probability of deposition of platelets and macrophages, resulting in IT.
High gradients of WSS
In an ETS graft-artery anastomosis, the floor typically experiences low oscillating WSS, due to the presence of a moving stagnation point during the cardiac cycle, as shown in Figure 8. However, this low-WSS–high-OSI hypothesis does not explain IH formation at the toe of the anastomosis (as the WSS is neither low nor oscillating at this location). Consequently, it has been postulated that the large spatial WSS gradient (WSSG), which is mainly observed at the toe of the anastomosis, induces morphological and functional changes in the endothelium that contribute to elevated wall permeability and hence possible atherosclerotic lesions . The spatial gradient of WSS represents the non-uniformity of the force distribution on the endothelium and implies a stretching force applied on the ECs, which can create local deformation of ECs and increase the wall permeability, leading to IH [28, 84].
Moreover, the local WSSG is suggested as the single best indicator of non-uniform flow fields leading to atherogenesis . Based on the biological evidence that non-uniform hemodynamic factors trigger an increase in wall permeability, Lei et al.  introduced an equation for wall permeability as a function of local WSSG magnitude; by employing the aorto-celiac junction of rabbits as a representative atherosclerotic model, their experimentally validated computer simulation model for enhanced LDL transport into the arterial wall showed that the WSSG is a reliable predictor of critical atherogenic sites in branching arteries . Besides, it has been observed that IH tends to develop at sites having high spatial and temporal gradients in WSS .
Disturbed flow patterns associated with distal ETS anastomosis
The configuration of distal ETS anastomosis is not naturally present in the arterial system (except for the patent ductus arteriosus). Although an ETS anastomosis is basically a bifurcation, it is different from naturally occurring blood vessel bifurcations. The angle between the daughter vessels is effectively obtuse and the flow division between the daughter branches in anastomoses can vary widely which has a significant impact on the hemodynamics . A distal ETS anastomosis is characterized by abnormal flow conditions, including flow oscillation at the heel, impact on the artery floor, and flow separation at the toe.
Typically, there is a low-WSS region at the heel, where a vortex forms due to the interaction of the flow from the graft with the relatively slow flow in the occluded proximal artery, whose size changes with the flow phase (see Figure 8b). The presence of a slow recirculation flow (i.e. a vortex) increases the near-wall residence time and results in platelet activation  and fibrin thrombus formation [110, 111], which leads to IH development [14, 17, 83, 88, 97, 98].
Along with this vortex, there is a stagnation point on the artery bed, where the graft flow impinges the floor (Figure 8b), whose location oscillates (with the size of the vortex) during the cardiac cycle. This moving stagnation point provides a low-magnitude‒high-oscillatory WSS condition on the artery bed which is prone to enhancement of atherogenesis  and IH formation [102, 112]. In addition, the flow impact on the artery floor is known to be injurious to the endothelium and is believed to be a contributing factor to the graft failure, as there is evidence of change in the flow character once it impacts against the junction floor .
In a conventional ETS configuration, there are high flow shear rates at the toe of the anastomosis (causing high WSS at the toe), usually with a flow separation region just distal to the toe at the inner wall of the coronary artery (causing flow recirculation and low WSS at this area), as shown in Figure 9a. This results in a high spatial gradient of WSS at the toe of the anastomosis, which induces morphological and functional changes in the endothelium that contribute to elevated wall permeability and consequent atherosclerotic lesions [105, 107, 108] and IH development . Moreover, in flow separation and flow recirculation zones, due to low WSS and enhanced particle residence time, excess cholesterol is deposited on the surface of the lumen, initiating atheroma growth  and IH [23, 109].
Although an ETS anastomosis is essentially a bifurcation, being manmade, surgically created anastomoses can be modified (in contrast to arterial bifurcations) to yield a flow environment that improves graft longevity . As reviewed above, investigations of blood flow and HPs and their comparison with focal locations of IT and IH formation in CABGs have resulted in correlation of some HPs with initiation and progression of IH. Consequently, HPs can, in turn, be utilized as indicators to show susceptible sites of IT and favorable conditions for thrombi and IH formation. Accordingly, using these indicators (i.e., by modification of HPs), extensive efforts have been put to obtain an optimal graft design, which is an end point for the study of correlations between hemodynamics and graft failure.
Attempts to design an optimal anastomotic configuration
The first efforts towards attaining an optimal anastomosis have been made by changing the anastomotic angle. It has been shown that tissue remodeling at ETS arterial anastomoses is highly sensitive to graft angle , and graft patency rates vary according to anastomotic angle. The anastomotic angle affects the flow regime and shear stress [21, 25]. A smaller anastomotic angle reduces (i) the peaks and gradients of WSS , (ii) the flow separation and disturbances at the toe [22, 23, 26, 27], (iii) secondary flow components , and (iv) size of recirculation area (i.e., reversed flow) downstream of the anastomosis [25–27], as shown in Figure 9. Hence, a smaller distal ETS anastomotic angle (≤ 30°) seems to bring about a less disturbed and more uniform, smooth flow from the graft into the coronary artery.
The effect of graft caliber (i.e., graft-to-host diameter ratio) on the hemodynamics of CABGs has also been examined by investigators. It is observed that larger graft-to-host diameter ratios (5:3) have better hemodynamic performance than smaller ones (1:1) , as they can bring about relatively large positive longitudinal velocity, uniform and large WSS , and small WSSG [21, 33]. Likewise, results of a computational study, using mesh-less CFD and genetic algorithms optimization, indicate that the graft caliber should always be maximized, in order to minimize the spatial and temporal gradients of WSS .
Besides, smaller grafts typically present an increased risk of early graft failure due to thrombosis . Several clinical studies have demonstrated that small caliber (<3.5 mm) of vein grafts is the only independent risk factor for vein graft stenosis [115, 116]. Idu et al.  suggested that a small caliber is a greater risk factor for graft failure than the use of arm or composite vein grafts, and that these alternative veins should be preferred if the saphenous vein graft is less than 3.5 mm in diameter.
It has been observed that a small-diameter (<3 mm) saphenous vein graft is associated with a 2.1-fold increased risk of early failure , and such conduits have a higher rate of occlusion in the perioperative (0‒30 day) interval . Moreover, observations from a large multi-centre trial suggest that small size of vein graft is the dominant technical determinant of early graft failure .
In addition, a smaller graft diameter increases the graft resistance against the flow, which can elevate the flow portion through the native (partially stenosed) coronary and escalate the competitive flow problem  that eventually results in graft thrombosis and failure.
The effect of competitive flow (i.e., flow through a bypassed native coronary artery with low degree of stenosis) on the graft patency has been extensively investigated, but still is somewhat controversial. Many studies have demonstrated that the patency of bypass grafts on functionally significant lesions is considerably higher than the patency of bypass grafts on non-significant lesions [120–127]. They have confirmed the existence of a critical value for stenosis severity, below which the graft failure is expected, and above that, the recipient artery will be progressively occluded [119, 128].
Although competitive flow (from patent native coronary vessels) is implicated in the failure of internal mammary artery (IMA) grafts, it is not thought to affect the patency of saphenous vein grafts (SVGs) [120, 124, 129]. This is because non-muscular SVGs cannot adjust their lumens in response to metabolic requirements as much as arterial grafts. Thus, the response of vein grafts to low flow is limited .
On the other hand, some studies have demonstrated that despite of significant correlation between (low) degree of proximal stenosis of the recipient coronary artery (i.e., presence of competitive flow) and occurrence of a string sign (where the graft conduit is patent but with only a thread of antegrade flow, due to narrowing of the graft), chronic native competitive flow does not significantly affect midterm graft status  and that the flow rates of the IMA grafts are comparable with and without stenosis or string phenomenon . Also, limited studies have reported that competitive flow from a moderately stenotic coronary artery has not predisposed patients toward the string sign of the IMA graft in the presence of substantial diastolic IMA flow .
The effects of stenosis severity and distance of grafting on the hemodynamics of distal anastomosis have also been investigated. Computational simulations have shown that in the case of bypass grafting of partially stenosed coronary artery, the flow through partially occluded host artery interacts with the bypass graft flow at the anastomotic junction and that this combined flow can cause adverse hemodynamic effects, particularly when the distance of grafting is short [39, 133–135]. The jet flow from a partially stenosed artery can increase the peak value of the axial velocity, if the stenosis is close to the anastomosis . Also, interaction between the flows from the graft and the partially occluded artery results in steep variations of WSS near the heel and toe of the anastomosis, which can facilitate intimal proliferation and thrombogenesis around the suture-line when combined with flow recirculation in these regions . Thus, it has been recommend that anastomosis be sutured with a sufficient distance of grafting, to enable the velocity profile to fully reattach before the heel so as to minimize the risks of intimal hyperplasia at the anastomosis .
The influence of out-of-plane graft curvature has been studied by several investigators . These investigations have revealed reductions in magnitudes of the peak time-averaged WSS  and mean oscillatory shear  in the non-planar models as compared to the planar configurations, which imply a corresponding reduction in the spatial extent of wall regions exposed to physiologically unfavorable flow conditions . Accordingly, in order to induce non-planar flow effects, the use of grafts with intrinsic helical axis was suggested . In vitro flow visualizations have shown significantly increased cross-plane mixing for the helical grafts (Figure 10a and b), and preliminary in vivo studies in arteriovenous bypass grafts have indicated that helical grafts offer the potential for improved patency (Figure 10c) [136, 137]. This can be attributed to the swirling flow effects induced by helical grafts which increase the magnitudes of velocity and WSS by adding a secondary (circumferential) velocity component to the axial velocity. This can enhance fluid-wall mass transport and render the spatial distribution of WSS relatively uniform in curved conduits, and potentially at anastomoses .
Nevertheless, the graft non-planarity is often constrained by surgical considerations beyond hemodynamics (e.g., the stenosed artery, location of stenosis, etc.) .
Further efforts have been put to obtain a more favorable anastomosis by design of cuffed and patched anastomotic configurations. Miller et al.  introduced a vein cuff design that produced good patency in femoral-distal grafts (Figure 11). However, the utility of this technique is somewhat controversial. While some studies found that grafts implanted with vein cuffs resulted in decreased developments of IH [140, 141] and had better patency than those with a non-cuffed anastomosis , other investigations showed that (i) the use of a Miller cuff caused no difference in IH thickness , (ii) the improved patency was only in below-knee grafts (and not in above-knee popliteal bypasses) [138, 139, 144], and (iii) the use of a cuff has adverse effects on hemodynamic factors around the anastomosis, such as large variations in shear stress on the artery floor, low-momentum recirculation within the cuff, and prominent separation at the cuff toe [145–147]. It is hence suggested that the improved patency rates achieved with cuffed anastomoses are due not to a decrease in IH but to an increased anastomotic volume and the consequent ability to accommodate IH, before it causes significant stenosis [143, 148].
The Taylor vein patch technique  (Figure 12) has been found to decrease IH , diminish flow disturbances and undesirable flow separation at the toe of the anastomosis , and slightly reduce the WSSG in the anastomotic region . However, its improvement in hemodynamic factors is minor. Besides, patched and non-patched grafts have shown similar primary patency rates .
Linton patch was introduced as a technique in which the conduit is patched with a venous segment of about 40-50 mm long , as shown in Figure 13, and it was frequently used in femoral artery to facilitate construction of the proximal anastomosis of femoropopliteal bypasses. Linton patch technique could considerably increase the compliance at the junction. However, its flow patterns have been shown to be similar to those of conventional ETS anastomosis. The clinical patency of this technique has been reported to be 65-74% at 12-48 months post-operative .
Lei et al. , utilizing an iterative optimization procedure coupled with CFD simulations, further improved the geometric design of the Taylor patch to obtain smaller WSSGs. This improved design, whose anastomotic surface area was smaller than that of the Taylor patch, yielded a significant reduction in local time-averaged WSSG (ranging from two- to six-fold decrease, compared with standard and Taylor hooded configurations for a variety of flow splits between POS and DOS) both at the toe and on the floor. This reduction was due to the gradual S-shaped transition in wall curvature and cross-sectional area at the toe region, as shown in Figure 14.
The Tyrrell collar has been developed in attempts to incorporate the advantages of the Miller and Taylor anastomotic designs, by avoiding direct suturing of the graft and artery (which can cause high compliance mismatch in case a synthetic graft is used), and providing a more streamlined shape at the toe  (Figure 15). However, trials of Tyrrell collar venous anastomosis in arteriovenous grafts (AVGs) not only showed no improvements in graft patency , but also indicated that the use of the collar at the venous anastomosis of forearm loop AVGs resulted in early graft failure .
Longest and Kleinstreuer  numerically simulated the haemodynamics for a conventional ETS anastomosis (as the base case), the Venaflo™ graft, and an improved cuffed graft-end configuration for AVGs (Figure 16). The Venaflo™ graft demonstrated considerable improvements over the base case by enlarging the junction area and reducing the severity of disturbed flow patterns in predictive computer simulations. Considering the critical toe region, further improvements were achieved in the modified graft-end design by smoother wall curvatures and elimination of the graft bulges, which further reduced the maximum normalized WSSG to 6.4 from 18.1 for the Venaflo™ graft. However, results of clinical trials of the Venaflo™ graft have been controversial. Some studies have shown promising graft patency rates in the Venaflo™ grafts (58% versus 21% in the conventional standard grafts at 24 months) , while other investigations have demonstrated the 1-year patency rates of the Venaflo™ grafts to be inferior to those of non-cuffed ePTFE grafts (43% versus 47%) .
A streamlined anastomotic configuration in which the distal outlet segment (DOS) is aligned with the graft has been developed by Longest et al. . This configuration, shown in Figure 17, resulted in an advantageous reduction of the peak normalized WSSG values in the vicinity of the toe (to 1.7 from 11.8 in a conventional ETS model). However, particle-wall interactions remained significant throughout the anastomosis, which can result in platelet activation and may lead to IH.
O’Brien et al.  have designed a configuration to replace the anastomosis with a synthetic bifurcation connected in an ETE fashion with the proximal outlet segment (POS) and DOS (Figure 18). Their numerical simulations indicated that the smoothly curving bifurcation improves the WSS environment by reducing flow separation and stagnation. Although this prosthetic graft configuration has primarily been designed for femoral-popliteal bypasses, the concept may be relevant in other aspects of cardiovascular surgery. This prosthetic graft can be manufactured from clinically proven synthetic materials, does not require any additional training in its use, and combines attributes of ETS anastomoses with those of ETE anastomoses.
Chua et al.  designed a cuff-like sleeve for implantation at the distal anastomosis of CABGs as a connector between the graft and the host artery (Figure 19). Their computational simulation results suggested that the sleeve models with higher necks were preferred in terms of hemodynamics at the distal anastomosis.
In an attempt to alter the disturbed hemodynamic on the artery bed in the ETS anastomosis, O’Brien et al.  designed a flow-splitter to be placed into the junction of distal ETS anastomosis, as shown in Figure 20. This flow-splitter splits the flow profile entering the anastomosis into two channels and diverts the flow from artery bed toward the arterial side-walls. Although this flow-splitter could reduce the peaks of WSS and WSSG on the bed (by 36% and 49%, respectively) at particular phases (during deceleration) and also mitigate the flow separation at the toe, it caused large increases in WSS on both sides of the artery bed centerline which can result in high values of time-averaged WSSG over the cardiac cycle near the centerline on the arterial bed. Besides, implantation of this flow-splitter may be practical only if integrated in a synthetic graft suite, and not along with autologous grafts.
Walsh et al.  have designed a novel vascular grafting device with a bifurcating configuration (Figure 21a), in order to eliminate the flow impingement on the interior wall of the artery at the distal anastomosis. In this vascular device, the flow from the proximal anastomosis is bifurcated into two branches and these branch flows impinge upon each other at a central region of the lumen at the distal anastomosis. By avoiding arterial bed impingement, the possibility of disease formation is reduced. Besides, the opposing branch flows rapidly regain the normal hemodynamic behavior in the distal artery (Figure 21b). Another positive feature of this design is the mitigation of flow separation at the toe. This prosthetic vascular graft can be incorporated into the host artery by means of two ETE anastomoses (as shown by suture-lines 2 and 3 in Figure 21a) at the distal section and a side-to-end anastomosis at the proximal section (not shown here). Surgical feasibility of this design for treatment of peripheral arterial disease has been verified in vivo, by implantation of a PTFE graft into the aorta of a pig model (Figure 21c) . However, a major limitation of this graft is its geometrical complexity.
Coupled sequential anastomoses design
Based on the advantageous flow characteristics observed within the side-to-side (STS) anastomosis of typical sequential bypass grafts (i.e., a smoother flow with smaller spatial gradients of WSS than those in an ETS anastomosis ) and higher patency rates in STS anastomoses than in ETS anastomoses , Kabinejadian et al. [167, 168] developed a novel coupled STS-ETS sequential anastomoses bypass graft design, as shown in Figure 22. In this design, part of the graft flow is diverted into the coronary artery at the STS anastomosis, and when this flow in the coronary artery reaches the ETS anastomosis, it lifts up the flow coming from the graft and directs the graft flow smoothly into the coronary artery, which prevents impingement of blood flow on the arterial bed and eliminates the stagnation point and flow recirculation at the ETS anastomosis.
Computational simulations of blood flow through this novel design have shown improvements of HPs, especially at the heel and on the arterial bed of the ETS anastomosis. These improvements in distribution of HPs include an increase in the time-averaged WSS on the artery bed of the ETS anastomosis of the SQA (as compared to the conventional ETS anastomosis), reduction of the time-averaged WSSG at the heel and bed of the ETS component as well as at the toe and suture line of the STS component of the novel SQA (as compared to the conventional ETS and typical parallel STS anastomoses, respectively), and reduction of the OSI at the ETS anastomosis of the SQA at the heel region and on the artery wall and bed opposite to the heel (in comparison with the conventional ETS anastomosis). Besides, this design provides a spare route for the blood flow to the coronary artery in order to avoid re-operation in case of re-stenosis in either of the anastomoses. This design can be employed using autologous grafts without the need for any additional training.
Conclusions and future directions
The search for an ideal distal anastomotic configuration for coronary bypass grafting has led to numerous designs. Optimal anastomosis design must take into account practical issues such as surgical construction. An anastomotic design should be feasible to be implemented by surgeons in a reasonable time.
As reviewed, some of the designed anastomotic configurations which are feasible to be constructed during operation using autologous materials (including Miller cuff, Taylor patch, and Tyrrell collar) have not shown a remarkable enhancement in patency of bypass grafts. On the other hand, some configurations which have shown considerable improvement in HPs distribution can only be made by synthetic materials due to their complex geometry (such as Venaflo™ graft, bifurcating graft-end design, cuff-like sleeve, and bifurcating vascular graft); this has consequences of blood-exposed non-intimal surface and high compliance mismatch at the synthetic material-blood vessel interface.
Moreover, the anastomotic designs which have improved HPs distribution and are feasible to be surgically constructed by autologous grafts (such as the coupled sequential anastomoses design), have not gone through animal trials or clinical investigations yet to demonstrate their in vivo performance and patency rates. Hence, the dilemma of designing an optimal anastomosis, which can bring about considerable improvements in the flow regime, HPs distribution, and graft patency, still remains unsolved.
In conclusion, there are a few aspects to be considered in the design of an optimal CABG:
Compatibility of the graft with the arterial pressure and the supplied blood flow rate, to ensure a physiologic range of intramural stresses and hemodynamic forces in the graft itself. Arterial grafts, such as left internal mammary artery (LIMA), have demonstrated considerably higher patency rates than the most commonly used saphenous vein grafts . However, due to lack of arterial conduits, veins are currently used most commonly as grafts. With technological advances, the time required for production of matured implantable tissue-engineered grafts, which could fulfill the ideal characteristics present in the arteries, will be shortened  and they can replace the vein grafts in CABG.
Arterial compliance of the graft, to avoid compliance mismatch with the host artery at the anastomotic junction, to prevent escalation of intramural stresses in the artery and the graft, which can result in IH formation, especially on the suture-line. As discussed in this review, compliance mismatch between the graft and the host artery results in an increase of intramural stresses, which in turn promotes IH. Use of arterial conduits can (to some extent) address this issue too. However, as mentioned above, tissue-engineered grafts might be the future solution to this problem.
Hemodynamic performance driven design of anastomotic configuration of the distal anastomosis, to regulate the hemodynamic parameters and wall shear stress indices, in order to avoid triggering of the pathogenic factors of IH and thrombosis (e.g., platelet activation, long near-wall residence time, etc.). As reviewed, it is well established that HPs play an important role in the initiation and progression of atherosclerosis and IH. A hemodynamically optimized anastomotic configuration can provide moderate shear stress parameters and smooth blood flow without flow disturbances, to avoid triggering the associated atherogenic phenomena.
Minimal vascular injury, to minimize proliferation of SMCs as a wound healing response. Technological advances may further develop the suggested alternatives to sutures (e.g., biological glues, laser generated solders, etc.) to a practicable level for routine clinical use. Not only can such products minimize vascular injury, but also they can eliminate the para-anastomotic hyper-compliant zone and the associated elevating intramural stresses which are caused by the stiff sutures.
Patient-specific designs, to tailor the design considerations to each particular patient’s cardiovascular characteristics. Development of clinical imaging (e.g., magnetic resonance and computed tomography) enables a detailed patient-specific description of the actual hemodynamics and structural behavior of living tissues. Coupling of these data with engineering analyses is becoming a standard evaluation that is expected to become part of the clinical practice in diagnosis and surgical planning in advanced medical centers . This would optimize the design considerations and choice of graft for each particular patient, depending on the number, location, and severity of stenosis, etc.
DNG: He is currently involved with Southern Ozarks Alliance for Rural Development in Willow Springs MO, setting up the Ozark Rural University MO (USA), and advising universities in their Programs development. He is a pioneer in the fields of biomedical engineering, healthcare engineering, and management. He has published over 450 works in the fields of engineering science, biomedical engineering, medical science, and social sciences, as well as 29 books on biomedical engineering, engineering physiology, cardiovascular physics, orthopedic mechanics, medical and life physics, and spinal injury. The inventor of life-saving implant devices has also been the former Chief Academic Officer (Provost) & Registrar of the Parkway College of Health Sciences (Singapore) and Professor in the School of Chemical & Biomedical Engineering of Nanyang Technological University (Singapore).
FK: He is currently a postdoctoral research fellow at National University of Singapore (NUS). He has about 10 years of research experience in Biomedical Engineering, including analytical, experimental, and computational investigations on biofluid and cardiovascular mechanics and design and development of relevant medical devices.
Coronary arterial bypass grafting
Coronary artery disease
Computational fluid dynamic
Distal anastomotic intimal hyperplasia
Distal outlet segment
Internal mammary artery
Left anterior descending coronary artery
Left main coronary artery
Oscillatory shear index
Percutaneous coronary intervention
Proximal outlet segment
Right coronary artery
Smooth muscle cell
Saphenous vein graft
Wall shear stress
Wall shear stress gradient.
Molina JA, Heng BH: Global trends in cardiology and cardiothoracic surgery - an opportunity or a threat? Ann Acad Med Singapore 2009, 38: 541–545.
Lloyd-Jones D, Adams RJ, Brown TM, Carnethon M, Dai S, De Simone G, Ferguson TB, Ford E, Furie K, Gillespie C, et al.: Executive summary: heart disease and stroke statistics-2010 update: a report from the american heart association. Circulation 2010, 121: e46-e215.
Rihal CS, Raco DL, Gersh BJ, Yusuf S: Indications for coronary artery bypass surgery and percutaneous coronary intervention in chronic stable angina: review of the evidence and methodological considerations. Circulation 2003, 108: 2439–2445.
Smith SC Jr, Dove JT, Jacobs AK, Kennedy JW, Kereiakes D, Kern MJ, Kuntz RE, Popma JJ, Schaff HV, Williams DO, et al.: ACC/AHA guidelines for percutaneous coronary intervention (revision of the 1993 PTCA guidelines) - Executive summary: a report of the American College of Cardiology/American Heart Association Task Force on Practice Guidelines (Committee to Revise the 1993 Guidelines for Percutaneous Transluminal Coronary angioplasty) Endorsed by the society for Cardiac Angiography and Interventions. Circulation 2001, 103: 3019–3041.
Braunwald H: Heart Disease: A Textbook of Cardiovascular Medicine. 5th edition. Philadelphia, PA: Saunders; 1997.
Davies MG, Hagen PO: Pathobiology of intimal hyperplasia. Br J Surg 1994, 81: 1254–1269.
Canver CC: Conduit options in coronary artery bypass surgery. Chest 1995, 108: 1150–1155.
FitzGibbon GM, Kafka HP, Leach AJ, Keon WJ, Hooper GD, Burton JR: Coronary bypass graft fate and patient outcome: angiographic follow-up of 5,065 grafts related to survival and reoperation in 1,388 patients during 25 years. J Am Coll Cardiol 1996, 28: 616–626.
Whittemore AD, Clowes AW, Couch NP, Mannick JA: Secondary femoro-popliteal reconstruction. Ann Surg 1981, 193: 35–42.
Butany JW, David TE, Ojha M: Histological and morphometric analyses of early and late aortocoronary vein grafts and distal anastomoses. Can J Cardiol 1998, 14: 671–677.
Clowes AW, Reidy MA, Clowes MM: Kinetics of cellular proliferation after arterial injury. I. Smooth muscle growth in the absence of endothelium. Lab Invest 1983, 49: 327–333.
Liu MW, Roubin GS, King SB III: Restenosis after coronary angioplasty: potential biologic determinants and role of intimal hyperplasia. Circulation 1989, 79: 1374–1387.
Keynton RS, Evancho MM, Sims RL, Rodway NV, Gobin A, Rittgers SE: Intimal hyperplasia and wall shear in arterial bypass graft distal anastomoses: an in vivo model study. J Biomech Eng 2001, 123: 464–473.
White SS, Zarins CK, Giddens DP, Bassiouny H, Loth F, Jones SA, Glagov S: Hemodynamic patterns in two models of end-to-side vascular graft anastomoses: effects of pulsatility, flow division, Reynolds number, and hood length. J Biomech Eng 1993, 115: 104–111.
Ballyk PD, Walsh C, Butany J, Ojha M: Compliance mismatch may promote graft-artery intimal hyperplasia by altering suture-line stresses. J Biomech 1997, 31: 229–237.
Clowes AW, Kirkman TR, Clowes MM: Mechanisms of arterial graft failure. II. Chronic endothelial and smooth muscle cell proliferation in healing polytetrafluoroethylene prostheses. J Vasc Surg 1986, 3: 877–884.
Bassiouny HS, White S, Glagov S, Choi E, Giddens DP, Zarins CK: Anastomotic intimal hyperplasia: mechanical injury or flow induced. J Vasc Surg 1992, 15: 708–717.
Hofer M, Rappitsch G, Perktold K, Trubel W, Schima H: Numerical study of wall mechanics and fluid dynamics in end-to-side anastomoses and correlation to intimal hyperplasia. J Biomech 1996, 29: 1297–1308.
Sottiurai VS, Yao JST, Batson RC, Sue SL, Jones R, Nakamura YA: Distal anastomotic intimal hyperplasia: histopathologic character and biogenesis. Ann Vasc Surg 1989, 3: 26–33.
Ojha M, Cobbold RSC, Johnston KW: Influence of angle on wall shear stress distribution for an end-to-side anastomosis. J Vasc Surg 1994, 19: 1067–1073.
Brien TO, Walsh M, McGloughlin T: On reducing abnormal hemodynamics in the femoral end-to-side anastomosis: the influence of mechanical factors. Ann Biomed Eng 2005, 33: 310–322.
Fei DY, Thomas JD, Rittgers SE: The effect of angle and flow rate upon hemodynamics in distal vascular graft anastomoses: a numerical model study. J Biomech Eng 1994, 116: 331–336.
Hughes PE, How TV: Effects of geometry and flow division on flow structures in models of the distal end-to-side anastomosis. J Biomech 1996, 29: 855–872.
Jackson ZS, Ishibashi H, Gotlieb AI, Lowell Langille B: Effects of anastomotic angle on vascular tissue responses at end-to-side arterial grafts. J Vasc Surg 2001, 34: 300–307.
Pietrabissa R, Inzoli F, Fumero R: Simulation study of the fluid dynamics of aorto-coronary bypass. J Biomed Eng 1990, 12: 419–424.
Staalsen NH: The anastomosis angle does change the flow fields at vascular end-to-side anastomoses in vivo. J Vasc Surg 1995, 21: 460–471.
Keynton RS, Rittgers SE, Shu MCS: The effect of angle and flow rate upon hemodynamics in distal vascular graft anastomoses: an in vitro model study. J Biomech Eng 1991, 113: 458–463.
Cole JS, Watterson JK, O’Reilly MJG: Numerical investigation of the haemodynamics at a patched arterial bypass anastomosis. Med Eng Phys 2002, 24: 393–401.
Lei M, Archie JP, Kleinstreuer C: Computational design of a bypass graft that minimizes wall shear stress gradients in the region of the distal anastomosis. J Vasc Surg 1997, 25: 637–646.
Leuprecht A, Perktold K, Prosi M, Berk T, Trubel W, Schima H: Numerical study of hemodynamics and wall mechanics in distal end-to-side anastomoses of bypass grafts. J Biomech 2002, 35: 225–236.
Longest PW, Kleinstreuer C: Computational haemodynamics analysis and comparison study of arterio-venous grafts. J Med Eng Technol 2000, 24: 102–110.
Bonert M, Myers JG, Fremes S, Williams J, Ethier CR: A numerical study of blood flow in coronary artery bypass graft side-to-side anastomoses. Ann Biomed Eng 2002, 30: 599–611.
Qiao A, Liu Y: Influence of graft-host diameter ratio on the hemodynamics of CABG. Bio-Med Mater Eng 2006, 16: 189–201.
Zahab ZE, Divo E, Kassab A: Minimisation of the wall shear stress gradients in bypass grafts anastomoses using meshless CFD and genetic algorithms optimisation. Comput Methods Biomech Biomed Engin 2010, 13: 35–47.
Papaharilaou Y, Doorly DJ, Sherwin SJ: The influence of out-of-plane geometry on pulsatile flow within a distal end-to-side anastomosis. J Biomech 2002, 35: 1225–1239.
Sankaranarayanan M, Ghista DN, Chua LP, Tan YS, Kassab GS: Analysis of blood flow in an out-of-plane CABG model. Am J Physiol-Heart C 2006, 291: H283-H295.
Sherwin SJ, Shah O, Doorly DJ, Peiró J, Papaharilaou Y, Watkins N, Caro CG, Dumoulin CL: The influence of out-of-plane geometry on the flow within a distal end- to-side anastomosis. J Biomech Eng 2000, 122: 86–95.
Deplano V, Bertolotti C, Boiron O: Numerical simulations of unsteady flows in a stenosed coronary bypass graft. Med Biol Eng Comput 2001, 39: 488–499.
Kute SM, Vorp DA: The effect of proximal artery flow on the hemodynamics at the distal anastomosis of a vascular bypass graft: computational study. J Biomech Eng 2001, 123: 277–283.
Kabinejadian F, Chua LP, Ghista DN, Tan YS: CABG models flow simulation study on the effects of valve remnants in the venous graft. J Mech Med Biol 2010, 10: 593–609.
Bertolotti C, Deplano V, Fuseri J, Dupouy P: Numerical and experimental models of post-operative realistic flows in stenosed coronary bypasses. J Biomech 2001, 34: 1049–1064.
Hida S, Chikamori T, Hirayama T, Usui Y, Yanagisawa H, Morishima T, Ishimaru S, Yamashina A: Beneficial effect of coronary artery bypass grafting as assessed by quantitative gated single-photon emission computed tomography. Circ J 2003, 67: 499–504.
Shah PJ, Gordon I, Fuller J, Seevanayagam S, Rosalion A, Tatoulis J, Raman JS, Buxton BF: Factors affecting saphenous vein graft patency: clinical and angiographic study in 1402 symptomatic patients operated on between 1977 and 1999. J Thorac Cardiovasc Surg 2003, 126: 1972–1977.
Bulkley BH, Hutchins GM: Accelerated “atherosclerosis”. A morphologic study of 97 saphenous vein coronary artery bypass grafts. Circulation 1977, 55: 163–169.
Loth F, Fischer PF, Bassiouny HS: Blood flow in end-to-side anastomoses. Annu Rev Fluid Mech 2008, 40: 367–393.
Ross R: The pathogenesis of atherosclerosis: a perspective for the 1990s. Nature 1993, 362: 801–809.
Carrel A, Guthrie CC: Anastomosis of blood vessels by the patching method and transplantation of the kidney. JAMA 1906, 47: 1648–1651.
Davies PF: Flow-mediated endothelial mechanotransduction. Physiol Rev 1995, 75: 519–560.
Chervu A, Moore WS: An overview of intimal hyperplasia. Surg Gynecol Obstet 1990, 171: 433–447.
Virmani R, Kolodgie FD, Burke AP, Farb A, Schwartz SM: Lessons from sudden coronary death - a comprehensive morphological classification scheme for atherosclerotic lesions. Arterioscler Thromb Vasc Biol 2000, 20: 1262–1275.
Lemson MS, Tordoir JHM, Daemen MJAP, Kitslaar PJEHM: Intimal hyperplasia in vascular grafts. Eur J Vasc Endovasc Surg 2000, 19: 336–350.
Barakat AI: Blood flow and arterial endothelial dysfunction: mechanisms and implications. Comptes Rendus Physique 2013, 14: 479–496.
Pasterkamp G, De Kleijn DPV, Borst C: Arterial remodeling in atherosclerosis, restenosis and after alteration of blood flow: potential mechanisms and clinical implications. Cardiovasc Res 2000, 45: 843–852.
Patel SD, Waltham M, Wadoodi A, Burnand KG, Smith A: The role of endothelial cells and their progenitors in intimal hyperplasia. Ther Adv Cardiovasc Dis 2010, 4: 129–141.
Mitra AK, Gangahar DM, Agrawal DK: Cellular, molecular and immunological mechanisms in the pathophysiology of vein graft intimal hyperplasia. Immunol Cell Biol 2006, 84: 115–124.
Newby AC, Zaltsman AB: Molecular mechanisms in intimal hyperplasia. J Pathol 2000, 190: 300–309.
Cox JL, Chiasson DA, Gotlieb AI: Stranger in a strange land: the pathogenesis of saphenous vein graft stenosis with emphasis on structural and functional differences between veins and arteries. Prog Cardiovasc Dis 1991, 34: 45–68.
Ip JH, Fuster V, Badimon L, Badimon J, Taubman MB, Chesebro JH: Syndromes of accelerated atherosclerosis: role of vascular injury and smooth muscle cell proliferation. J Am Coll Cardiol 1990, 15: 1667–1687.
Allaire E, Clowes AW: Endothelial cell injury in cardiovascular surgery: the intimal hyperplastic response. Ann Thorac Surg 1997, 63: 582–591.
Bechler SL, Si Y, Yu Y, Ren J, Liu B, Lynn DM: Reduction of intimal hyperplasia in injured rat arteries promoted by catheter balloons coated with polyelectrolyte multilayers that contain plasmid DNA encoding PKCδ. Biomaterials 2013, 34: 226–236.
Goel SA, Guo LW, Liu B, Kent KC: Mechanisms of post-intervention arterial remodelling. Cardiovasc Res 2012, 96: 363–371.
Heckenkamp J, Lamuraglia GM: Intimal hyperplasia, arterial remodeling, and restenosis: an overview. Perspect Vasc Surg Endovasc Ther 1999, 11: 71–94.
Zalewski A, Shi Y: Vascular myofibroblasts: lessons from coronary repair and remodeling. Arterioscl Throm Vas 1997, 17: 417–422.
Dobrin P: Vascular mechanics. In Handbook of Physiology, Section 2: The Cardiovascular System, Volume III, Peripheral Circulation and Organ Blood Flow. Edited by: Shepherd JT, Abboud FM. Bethesda, MD: American Physiological Society; 1983:80–83.
Boutouyrie P, Laurent S, Girerd X, Benetos A, Lacolley P, Abergel E, Safar M: Common carotid artery stiffness and patterns of left ventricular hypertrophy in hypertensive patients. Hypertension 1995, 25: 651–659.
Abbott WM, Megerman J, Hasson JE, Litalien G, Warnock DF: Effect of compliance mismatch on vascular graft patency. J Vasc Surg 1987, 5: 376–382.
Trubel W, Moritz A, Schima H, Raderer F, Scherer R, Ullrich R, Losert U, Polterauer P: Compliance and formation of distal anastomotic intimal hyperplasia in Dacron mesh tube constricted veins used as arterial bypass grafts. ASAIO J 1994, 40: M273-M278.
Courtman DW PhD thesis. In The role of circumferential strain in vascular healing. The University of Toronto; 1994.
Carosi JA, Eskin SG, McIntire LV: Cyclical strain effects on production of vasoactive materials in cultured endothelial cells. J Cell Physiol 1992, 151: 29–36.
Letsou GV, Rosales O, Maitz S, Vogt A, Sumpio BE: Stimulation of adenylate cyclase activity in cultured endothelial cells subjected to cyclic stretch. J Cardiovasc Surg 1990, 31: 634–639.
Melbin J, Ho PC: Stress reduction by geometric compliance matching at vascular graft anastomoses. Ann Biomed Eng 1997, 25: 874–881.
Vijayan V, Shukla N, Johnson JL, Gadsdon P, Angelini GD, Smith FCT, Baird R, Jeremy JY: Long-term reduction of medial and intimal thickening in porcine saphenous vein grafts with a polyglactin biodegradable external sheath. J Vasc Surg 2004, 40: 1011–1019.
Desai M, Mirzay-Razzaz J, Von Delft D, Sarkar S, Hamilton G, Seifalian AM: Inhibition of neointimal formation and hyperplasia in vein grafts by external stent/sheath. Vasc Med 2010, 15: 287–297.
Okuhn SP, Connelly DP, Calakos N, Ferrell L, Man-Xiang P, Goldstone J: Does compliance mismatch alone cause neointimal hyperplasia? J Vasc Surg 1989, 9: 35–45.
Tiwari A, Cheng KS, Salacinski H, Hamilton G, Seifalian AM: Improving the patency of vascular bypass grafts: the role of suture materials and surgical techniques on reducing anastomotic compliance mismatch. Eur J Vasc Endovasc Surg 2003, 25: 287–295.
Suyker WJL, Borst C: Coronary connector devices: analysis of 1,469 anastomoses in 1,216 patients. Ann Thorac Surg 2008, 85: 1828–1836.
Gummert JF, Opfermann U, Jacobs S, Walther T, Kempfert J, Mohr FW, Falk V: Anastomotic devices for coronary artery bypass grafting: technological options and potential pitfalls. Comput Biol Med 2007, 37: 1384–1393.
Scheltes JS, Van Andel CJ, Pistecky PV, Borst C: Coronary anastomotic devices: blood-exposed non-intimal surface and coronary wall stress. J Thorac Cardiovasc Surg 2003, 126: 191–199.
Stewart SFC, Lyman DJ: Effects of a vascular graft/natural artery compliance mismatch on pulsatile flow. J Biomech 1992, 25: 297–310.
Stewart SFC, Lyman DJ: Effects of an artery/vascular graft compliance mismatch on protein transport: a numerical study. Ann Biomed Eng 2004, 32: 991–1006.
Steinman DA, Ethier CR: The effect of wall distensibility on flow in a two-dimensional end-to-side anastomosis. J Biomech Eng 1994, 116: 294–301.
Sottiurai VS, Lim Sue S, Feinberg Ii EL, Bringaze WL, Tran AT, Batson RC: Distal anastomotic intimal hyperplasia: biogenesis and etiology. Eur J Vasc Surg 1988, 2: 245–256.
Sottiurai VS: Distal anastomotic intimal hyperplasia: histocytomorphology, pathophysiology, etiology, and prevention. Int J Angiol 1999, 8: 1–10.
Ojha M: Spatial and temporal variations of wall shear stress within an end-to-side arterial anastomosis model. J Biomech 1993, 26: 1377–1388.
Fillinger MF, Reinitz ER, Schwartz RA, Resetarits DE, Paskanik AM, Bruch D, Bredenberg CE: Graft geometry and venous intimal-medial hyperplasia in arteriovenous loop grafts. J Vasc Surg 1990, 11: 556–566.
Giordana S, Sherwin SJ, Peiro J, Doorly DJ, Crane JS, Lee KE, Cheshire NJW, Caro CG: Local and global geometric influence on steady flow in distal anastomoses of peripheral bypass grafts. J Biomech Eng 2005, 127: 1087–1098.
Loth F, Jones SA, Zarins CK, Giddens DP, Nassar RF, Glagov S, Bassiouny HS: Relative contribution of wall shear stress and injury in experimental intimal thickening at PTFE end-to-side arterial anastomoses. J Biomech Eng 2002, 124: 44–51.
Rittgers SE, Karayannacos PE, Guy JF: Velocity distribution and intimal proliferation in autologous vein grafts in dogs. Circ Res 1978, 42: 792–801.
Kleinstreuer C, Nazemi M, Archie JP: Hemodynamics analysis of a stenosed carotid bifurcation and its plaque-mitigating design. J Biomech Eng 1991, 113: 330–335.
Fry DL: Acute vascular endothelial changes associated with increased blood velocity gradients. Circ Res 1968, 22: 165–197.
Fry DL: Certain histological and chemical responses of the vascular interface to acutely induced mechanical stress in the aorta of the dog. Circ Res 1969, 24: 93–108.
Caro CG, Fitz-Gerald JM, Schroter RC: Atheroma and arterial wall shear. Observation, correlation and proposal of a shear dependent mass transfer mechanism for atherogenesis. P Roy Soc Lond B Bio 1971, 177: 109–159.
Nazemi M, Kleinstreuer C, Archie JP, Sorrell FY: Fluid flow and plaque formation in an aortic bifurcation. J Biomech Eng-T Asme 1989, 111: 316–324.
Nazemi M, Kleinstreuer C, Archie JP Jr: Pulsatile two-dimensional flow and plaque formation in a carotid artery bifurcation. J Biomech 1990, 23: 1031–1037.
Shahcheraghi N, Dwyer HA, Cheer AY, Barakat AI, Rutaganira T: Unsteady and three-dimensional simulation of blood flow in the human aortic arch. J Biomech Eng-T Asme 2002, 124: 378–387.
Nerem RM: Vascular fluid mechanics, the arterial wall, and atherosclerosis. J Biomech Eng 1992, 114: 274–282.
Ishibashi H, Sunamura M, Karino T: Flow patterns and preferred sites of intimal thickening in end-to-end anastomosed vessels. Surgery 1995, 117: 409–420.
Sunamura M, Ishibashi H, Karino T: Flow patterns and preferred sites of intimal thickening in diameter-mismatched vein graft interpositions. Surgery 2007, 141: 764–776.
Morinaga K, Okadome K, Kuroki M: Effect of wall shear stress on intimal thickening of arterially transplanted autogenous veins in dogs. J Vasc Surg 1985, 2: 430–433.
Ku DN, Giddens DP, Zarins CK, Glagov S: Pulsatile flow and atherosclerosis in the human carotid bifurcation. Positive correlation between plaque location and low and oscillating shear stress. Arteriosclerosis 1985, 5: 293–302.
He X, Ku DN: Pulsatile flow in the human left coronary artery bifurcation: average conditions. J Biomech Eng 1996, 118: 74–82.
Li XM, Rittgers SE: Hemodynamic factors at the distal end-to-side anastomosis of a bypass graft with different POS:DOS flow ratios. J Biomech Eng 2001, 123: 270–276.
Passerini AG, Milsted A, Rittgers SE: Shear stress magnitude and directionality modulate growth factor gene expression in preconditioned vascular endothelial cells. J Vasc Surg 2003, 37: 182–190.
Zhang JM, Chua LP, Ghista DN, Yu SCM, Tan YS: Numerical investigation and identification of susceptible sites of atherosclerotic lesion formation in a complete coronary artery bypass model. Med Biol Eng Comput 2008, 46: 689–699.
DePaola N, Gimbrone MA Jr, Davies PF, Dewey CF Jr: Vascular endothelium responds to fluid shear stress gradients. Arterioscler Thromb 1992, 12: 1254–1257.
Haruguchi H, Teraoka S: Intimal hyperplasia and hemodynamic factors in arterial bypass and arteriovenous grafts: a review. J Artif Organs 2003, 6: 227–235.
Lei M, Kleinstreuer C, Truskey GA: Numerical investigation and prediction of atherogenic sites in branching arteries. J Biomech Eng 1995, 117: 350–357.
Lei M, Kleinstreuer C, Truskey GA: A focal stress gradient-dependent mass transfer mechanism for atherogenesis in branching arteries. Med Eng Phys 1996, 18: 326–332.
Hughes PE, How TV: Flow structures at the proximal side-to-end anastomosis. Influence of geometry and flow division. J Biomech Eng 1995, 117: 224–236.
Friedrich P, Reininger AJ: Occlusive thrombus formation on indwelling catheters: in vitro investigation and computational analysis. Thromb Haemost 1995, 73: 66–72.
Reininger AJ, Heinzmann U, Reininger CB, Friedrich P, Wurzinger LJ: Flow mediated fibrin thrombus formation in an endothelium-lined model of arterial branching. Thromb Res 1994, 74: 629–641.
Li XM, Rittgers SE: Hemodynamic factors at the distal end-to-side anastomosis of a bypass graft with different POS:DOS ratios. American Society of Mechanical Engineers, Bioengineering Division (Publication) BED 1999, 42: 225–226.
Bates CJ, O’Doherty DM, Williams D: Flow instabilities in a graft anastomosis: a study of the instantaneous velocity fields. P I Mech Eng H 2001, 215: 579–587.
Binns RL, Ku DN, Stewart MT, Ansley JP, Coyle KA: Optimal graft diameter: effect of wall shear stress on vascular healing. J Vasc Surg 1989, 10: 326–337.
Idu MM, Buth J, Hop WCJ, Cuypers P, Van De Pavoordt EDWM, Tordoir JMH: Factors influencing the development of vein-graft stenosis and their significance for clinical management. Eur J Vasc Endovasc Surg 1999, 17: 15–21.
Varty K, London NJM, Brennan JA, Ratliff DA, Bell PRF: Infragenicular in situ vein bypass graft occlusion: a multivariate risk factor analysis. Eur J Vasc Surg 1993, 7: 567–571.
Schanzer A, Hevelone N, Owens CD, Belkin M, Bandyk DF, Clowes AW, Moneta GL, Conte MS: Technical factors affecting autogenous vein graft failure: observations from a large multicenter trial. J Vasc Surg 2007, 46: 1180–1190. discussion 1190
Towne JB, Schmitt DD, Seabrook GR, Bandyk DF: The effect of vein diameter on patency of in situ grafts. J Cardiovasc Surg 1991, 32: 192–196.
Yasuura K, Takagi Y, Ohara Y, Takami Y, Matsuura A, Okamoto H: Theoretical analysis of right gastroepiploic artery grafting to right coronary artery. Ann Thorac Surg 2000, 69: 728–731.
Bezon E, Choplain JN, Maguid YA, Aziz AA, Barra JA: Failure of internal thoracic artery grafts: conclusions from coronary angiography mid-term follow-up. Ann Thorac Surg 2003, 76: 754–759.
Botman CJ, Schonberger J, Koolen S, Penn O, Botman H, Dib N, Eeckhout E, Pijls N: Does stenosis severity of native vessels influence bypass graft patency? A prospective fractional flow reserve-guided study. Ann Thorac Surg 2007, 83: 2093–2097.
Nakajima H, Kobayashi J, Toda K, Fujita T, Shimahara Y, Kasahara Y, Kitamura S: A 10-year angiographic follow-up of competitive flow in sequential and composite arterial grafts. Eur J Cardio-Thorac 2011, 40: 399–404.
Nordgaard H, Nordhaug D, Kirkeby-Garstad I, Løvstakken L, Vitale N, Haaverstad R: Different graft flow patterns due to competitive flow or stenosis in the coronary anastomosis assessed by transit-time flowmetry in a porcine model. Eur J Cardio-Thorac 2009, 36: 137–142.
Pagni S, Storey J, Ballen J, Montgomery W, Chiang BY, Etoch S, Spence PA: ITA versus SVG: a comparison of instantaneous pressure and flow dynamics during competitive flow. Eur J Cardio-Thorac 1997, 11: 1086–1092.
Sabik JF III, Lytle BW, Blackstone EH, Khan M, Houghtaling PL, Cosgrove DM, Shumway SJ, Tatoulis J, Bednarski P: Does competitive flow reduce internal thoracic artery graft patency? Ann Thorac Surg 2003, 76: 1490–1497.
Speziale G: Competitive Flow and Steal Phenomenon in Coronary Surgery. In Intraoperative Graft Patency Verification in Cardiac and Vascular Surgery. Edited by: D'Ancona G, Karamanoukian HL, Ricci M, Salerno TA, Bergsland J. New York: Futura Publishing; 2001.
Villareal RP, Mathur VS: The string phenomenon: an important cause of internal mammary artery graft failure. Tex Heart Inst J 2000, 27: 346–349.
Wiesner TF, Levesque MJ, Rooz E, Nerem RM: Epicardial coronary blood flow including the presence of stenoses and aorto-coronary bypasses II: experimental comparison and parametric investigations. J Biomech Eng 1988, 110: 144–149.
Sabik JF III, Lytle BW, Blackstone EH, Houghtaling PL, Cosgrove DM: Comparison of saphenous vein and internal thoracic artery graft patency by coronary system. Ann Thorac Surg 2005, 79: 544–551.
Gaudino M, Alessandrini F, Nasso G, Bruno P, Manzoli A, Possati G: Severity of coronary artery stenosis at preoperative angiography and midterm mammary graft status. Ann Thorac Surg 2002, 74: 119–121.
Hirotani T, Kameda T, Shirota S, Nakao Y: An evaluation of the intraoperative transit time measurements of coronary bypass flow. Eur J Cardio-Thorac 2001, 19: 848–852.
Kawasuji M, Sakakibara N, Takemura H, Tedoriya T, Ushijima T, Watanabe Y: Is internal thoracic artery grafting suitable for a moderately stenotic coronary artery? J Thorac Cardiovasc Surg 1996, 112: 253–259.
Bertolotti C, Deplano V: Three-dimensional numerical simulations of flow through a stenosed coronary bypass. J Biomech 2000, 33: 1011–1022.
Chen J, Lu XY, Wang W: Non-Newtonian effects of blood flow on hemodynamics in distal vascular graft anastomoses. J Biomech 2006, 39: 1983–1995.
Su CM, Lee D, Tran-Son-Tay R, Shyy W: Fluid flow structure in arterial bypass anastomosis. J Biomech Eng 2005, 127: 611–618.
Caro CG, Cheshire NJ, Watkins N: Preliminary comparative study of small amplitude helical and conventional ePTFE arteriovenous shunts in pigs. J R Soc Interface 2005, 2: 261–266.
Huijbregts HJTAM, Blankestijn PJ, Caro CG, Cheshire NJW, Hoedt MTC, Tutein Nolthenius RP, Moll FL: A helical PTFE arteriovenous access graft to swirl flow across the distal anastomosis: results of a preliminary clinical study. Eur J Vasc Endovasc Surg 2007, 33: 472–475.
Miller JH, Foreman RK, Ferguson L, Faris I: Interposition vein cuff for anastomosis of prosthesis to small artery. Aust N Z J Surg 1984, 54: 283–285.
Stonebridge PA, Prescott RJ, Ruckley CV: Randomized trial comparing infrainguinal polytetrafluoroethylene bypass grafting with and without vein interposition cuff at the distal anastomosis. J Vasc Surg 1997, 26: 543–550.
Kissin M, Kansal N, Pappas PJ, DeFouw DO, Durán WN, Hobson Ii RW: Vein interposition cuffs decrease the intimal hyperplastic response of polytetrafluoroethylene bypass grafts. J Vasc Surg 2000, 31: 69–83.
Suggs WD, Hendriques HF, DePalma RG: Vein cuff interposition prevents juxta-anastomotic neointimal hyperplasia. Ann Surg 1988, 207: 717–723.
Brumby SA, Petrucco MF, Walsh JA, Bond MJ: A retrospective analysis of infra-inguinal arterial reconstruction: three year patency rates. Aust N Z J Surg 1992, 62: 256–260.
Norberto JJ, Sidawy AN, Trad KS, Jones BA, Neville RF, Najjar SF, Sidawy MK, DePalma RG: The protective effect of vein cuffed anastomoses is not mechanical in origin. J Vasc Surg 1995, 21: 558–566.
Raptis S, Miller JH: Influence of a vein cuff on polytetrafluoroethylene grafts for primary femoropopliteal bypass. Br J Surg 1995, 82: 487–491.
Cole JS, Wijesinghe LD, Watterson JK, Scott DJA: Computational and experimental simulations of the haemodynamics at cuffed arterial bypass graft anastomoses. P I Mech Eng H 2002, 216: 135–143.
Henry FS, Küpper C, Lewington NP: Simulation of flow through a Miller cuff bypass graft. Comput Methods Biomech Biomed Engin 2002, 5: 207–217.
Longest PW, Kleinstreuer C, Archie JP Jr: Particle hemodynamics analysis of Miller cuff arterial anastomosis. J Vasc Surg 2003, 38: 1353–1362.
Wijesinghe LD, Mahmood T, Scott DJA: Axial flow fields in cuffed end-to-side anastomoses: effect of angle and disease progression. Eur J Vasc Endovasc Surg 1999, 18: 240–244.
Taylor RS, Loh A, McFarland RJ, Cox M, Chester JF: Improved technique for polytetrafluoroethylene bypass grafting: long-term results using anastomotic vein patches. Br J Surg 1992, 79: 348–354.
Gentile AT, Mills JL, Gooden MA, Hagerty RD, Berman SS, Hughes JD, Kleinert LB, Williams SK: Vein patching reduces neointimal thickening associated with prosthetic graft implantation. Am J Surg 1998, 176: 601–607.
Yeung KK, Mills JL Sr, Hughes JD, Berman SS, Gentile AT, Westerband A: Improved patency of infrainguinal polytetrafluoroethylene bypass grafts using a distal Taylor vein patch. Am J Surg 2001, 182: 578–583.
Linton RR, Darling RC: Autogenous saphenous vein bypass grafts in femoropopliteal obliterative arterial disease. Surgery 1962, 51: 62–73.
Batson RC, Sottiurai VS, Craighead CC: Linton patch angioplasty. An adjunct to distal bypass with polytetrafluoroethylene grafts. Ann Surg 1984, 199: 684–693.
Tyrrell MR, Wolfe JHN: New prosthetic venous collar anastomotic technique: combining the best of other procedures. Br J Surg 1991, 78: 1016–1017.
Lemson MS, Tordoir JHM, Van Det RJ, Welten RJTJ, Burger H, Estourgie RJA, Stroecken HJG, Leunissen KML: Effects of a venous cuff at the venous anastomosis of polytetrafluoroethylene grafts for hemodialysis vascular access. J Vasc Surg 2000, 32: 1155–1163.
Gagne PJ, Martinez J, DeMassi R, Gregory R, Parent FN, Gayle R, Meier Iii GH, Philput C: The effect of a venous anastomosis Tyrell vein collar on the primary patency of arteriovenous grafts in patients undergoing hemodialysis. J Vasc Surg 2000, 32: 1149–1154.
Sorom AJ, Hughes CB, McCarthy JT, Jenson BM, Prieto M, Panneton JM, Sterioff S, Stegall MD, Nyberg SL: Prospective, randomized evaluation of a cuffed expanded polytetrafluoroethylene graft for hemodialysis vascular access. Surgery 2002, 132: 135–140.
Lumsden AB, Weaver FA, Hood DB: Prospective multi-center evaluation of VENAFLO ePTFE as compared to Impra ePTFE vascular graft in hemodialysis applications. In Vascular Access for Hemodialysis. 4th edition. Edited by: Henry ML. Chicago: Precept Press; 1997:242–249.
Longest PW, Kleinstreuer C, Deanda A: Numerical simulation of wall shear stress and particle-based hemodynamic parameters in pre-cuffed and streamlined end-to-side anastomoses. Ann Biomed Eng 2005, 33: 1752–1766.
O’Brien TP, Grace P, Walsh M, Burke P, McGloughlin T: Computational investigations of a new prosthetic femoral-popliteal bypass graft design. J Vasc Surg 2005, 42: 1169–1175.
Chua LP, Tong JH, Zhou T: Numerical simulation of steady flows in designed sleeve models at distal anastomoses. Int Commun Heat Mass 2005, 32: 707–714.
O’Brien T, Walsh M, McGloughlin T: Altering end-to-side anastomosis junction hemodynamics: the effects of flow-splitting. Med Eng Phys 2006, 28: 727–733.
Walsh MT, McGloughlin TM, Grace P: A Vascular Graft. 2010. US7651526B2
O’Brien TP, Walsh MT, Kavanagh EG, Finn SP, Grace PA, McGloughlin TM: Surgical feasibility study of a novel polytetrafluoroethylene graft design for the treatment of peripheral arterial disease. Ann Vasc Surg 2007, 21: 611–617.
Sankaranarayanan M, Ghista DN, Chua LP, Tan YS, Sundaravadivelu K, Kassab GS: Blood Flow in an Out-of-Plane Aorto-left Coronary Sequential Bypass Graft. In Computational Cardiovascular Mechanics: Modeling and Applications in Heart Failure. Edited by: Guccione JM, Kassab GS, Ratcliffe M. New York: Springer US; 2010:277–295.
Vural KM, Sener E, Tasdemir O: Long-term patency of sequential and individual saphenous vein coronary bypass grafts. Eur J Cardiothorac Surg 2001, 19: 140–144.
Kabinejadian F, Chua LP, Ghista DN, Sankaranarayanan M, Tan YS: A novel coronary artery bypass graft design of sequential anastomoses. Ann Biomed Eng 2010, 38: 3135–3150.
Kabinejadian F, Ghista DN: Compliant model of a coupled sequential coronary arterial bypass graft: effects of vessel wall elasticity and non-Newtonian rheology on blood flow regime and hemodynamic parameters distribution. Med Eng Phys 2012, 34: 860–872.
Goldman S, Zadina K, Moritz T, Ovitt T, Sethi G, Copeland JG, Thottapurathu L, Krasnicka B, Ellis N, Anderson RJ, Henderson W: Long-term patency of saphenous vein and left internal mammary artery grafts after coronary artery bypass surgery: results from a Department of Veterans Affairs Cooperative Study. J Am Coll Cardiol 2004, 44: 2149–2156.
Sarkar S, Schmitz-Rixen T, Hamilton G, Seifalian AM: Achieving the ideal properties for vascular bypass grafts using a tissue engineered approach: a review. Med Biol Eng Comput 2007, 45: 327–336.
Migliavacca F, Dubini G: Computational modeling of vascular anastomoses. Biomech Model Mechanobiol 2005, 3: 235–250.
The authors declare no competing interest with regards to this invited review.
FK: collection, organizing, and review of the literature; preparing the manuscript. DNG: manuscript review, modification, editing, and revision. Both authors read and approved the final manuscript.
Authors’ original submitted files for images
Below are the links to the authors’ original submitted files for images.
About this article
Cite this article
Ghista, D.N., Kabinejadian, F. Coronary artery bypass grafting hemodynamics and anastomosis design: a biomedical engineering review. BioMed Eng OnLine 12, 129 (2013). https://doi.org/10.1186/1475-925X-12-129
- Coronary artery bypass graft (CABG)
- Wall shear stress
- Intimal hyperplasia