- Letter to the Editor
- Open Access
Time lag between oscillatory pressure and flow affecting accuracy of forced oscillation technique
© Ohishi and Kurosawa; licensee BioMed Central Ltd. 2011
- Received: 13 May 2011
- Accepted: 29 July 2011
- Published: 29 July 2011
The forced oscillation technique (FOT) is a simple method for assessing the oscillatory mechanics of the respiratory system. The oscillatory properties, respiratory system resistance (Rrs) and reactance (Xrs), are calculated from the oscillatory pressure/flow relationship. Although the FOT has been a well-established technique, some detailed experimental conditions would be different among institutions.
We evaluated whether time lags produced by the experimental conditions such as different positions of the sensors can affect the accuracy of the FOT. If the position of the pressure sensor is different from the flow sensor, a time lag may occur in the measurements. The effect of the time lag was studied by numerical analysis.
Rrs was estimated to be increased and Xrs decreased with an increase in the time lag, especially at a high oscillatory frequency of the medium-frequency range (5-35 Hz). At the high-frequency range (10-500 Hz), Rrs and Xrs were strikingly different in the values of the time lag.
A time lag between the oscillatory pressure and flow may be involved in the accuracy of the FOT, suggesting that it needs to be minimized or compensated for with signal processing. Researchers should pay attention to such detailed experimental conditions of the FOT apparatus.
- Respiratory System
- Oscillatory Pressure
- Oscillatory Mechanic
- High Oscillatory Frequency
- Simple Linear Model
The forced oscillation technique (FOT) is a simple method for assessing the oscillatory mechanics of the respiratory system, which has provided important findings in respiratory physiology [1, 2]. Currently, the FOT using multi-frequency signals such as pseudo random noise or impulse signals is widely used in clinical research [3–6]. Although the FOT has been a well-established technique, some detailed experimental conditions would be different among institutions. To elucidate the effects of the experimental conditions may be important for more accurate FOT measurements. The purpose of this study was to evaluate whether time lags produced by the experimental conditions such as different positions of the sensors can affect the accuracy of the FOT.
The oscillatory properties, Rrs as the resistive component and Xrs as the reactive one, are calculated by a spectral relationship between the oscillatory pressure and flow. Generally, the oscillatory pressure and flow are measured simultaneously by sensors such as a pressure transducer or pneumotachograph near the subject's mouth, which are analyzed in the signal processing by Fourier transform . In some previous papers, the sensors in the diagram were at different distances from the subject's mouth [2–4]. This difference would cause a time lag in the oscillatory pressure and flow in the signal processing. To evaluate whether the time lag could affect the measurement of the FOT, this effect was examined by numerical analysis as follows.
In conclusion, the time lag between the oscillatory pressure and flow may affect the accuracy of the FOT. This result suggests that the time lag needs to be minimized or compensated for in the signal processing for the optimal FOT measurement. Researchers should pay attention to such detailed experimental conditions of the FOT apparatus.
Acknowledgements and Funding
This study was supported by Grant-in-Aid for Japanese Society for the Promotion of Science Fellows.
- DuBois AB, Brody AW, Lewis DH, Burgess BF Jr: Oscillation mechanics of lung and chest in man. J Appl Physiol 1956, 8: 587–594.Google Scholar
- Oostveen E, MacLeod D, Lorino H, Farre R, Hantos Z, Desager K, Marchal F, ERS Task Force on Respiratory Impedance Measurements: The forced oscillation technique in clinical practice: methodology, recommendations and future developments. Eur Respir J 2003, 22: 1026–1041. 10.1183/09031936.03.00089403View ArticleGoogle Scholar
- Frey U: Forced oscillation technique in infants and young children. Paediatr Respir Rev 2005, 6: 246–254. 10.1016/j.prrv.2005.09.010View ArticleGoogle Scholar
- Beydon N, Davis SD, Lombardi E, Allen JL, Arets HG, Aurora P, Bisgaard H, Davis GM, Ducharme FM, Eigen H, Gappa M, Gaultier C, Gustafsson PM, Hall GL, Hantos Z, Healy MJ, Jones MH, Klug B, Lødrup Carlsen KC, McKenzie SA, Marchal F, Mayer OH, Merkus PJ, Morris MG, Oostveen E, Pillow JJ, Seddon PC, Silverman M, Sly PD, Stocks J, Tepper RS, Vilozni D, Wilson NM, American Thoracic Society/European Respiratory Society Working Group on Infant and Young Children Pulmonary Function Testing: An official American Thoracic Society/European Respiratory Society statement: pulmonary function testing in preschool children. Am J Respir Crit Care Med 2007, 175: 1304–1345. 10.1164/rccm.200605-642STView ArticleGoogle Scholar
- Faria AC, Lopes AJ, Jansen JM, Melo PL: Evaluating the forced oscillation technique in the detection of early smoking-induced respiratory changes. Biomed Eng Online 2009., 8(22): 10.1186/1475-925X-8-22Google Scholar
- Marchal F, Hall GL: Forced oscillation technique. Eur Respir Mon 2010, 47: 121–136.Google Scholar
- Michaelson ED, Grassman ED, Peters WR: Pulmonary mechanics by spectral analysis of forced random noise. J Clin Invest 1975, 56: 1210–1230. 10.1172/JCI108198View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.